(2734,2732,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S MLN NA94V-0 3207','4030359','2021-07-06','Test.','103','','',3,1,1,11),(2735,2733,2400,NULL,NULL,NULL,21,51,52,'OPTA1','253L 1413 2176 PS','2021-07-07','4-20 Output faulted\r\n100R, 150R => 2 II 300R','7 548#','','VACON',10,3,1,11),(2736,2734,2400,NULL,NULL,NULL,21,51,52,'OPTA1','253L 1545 4506 AS','2021-07-07','AT24C02 - BUS are shorted\r\nProgramming the new one.','7 548#','','VACON',10,3,1,11),(2737,2735,12000,NULL,NULL,NULL,21,396,103,'00050','1108','2021-07-07','Replacing part:\r\nRS number 714-2439 (673-7742).\r\n638:\r\nLow drop module A-AD82A6 V1/1 (SMD LOUB) faulted.\r\nChanged on LM2940-5\r\nFlat connector changed.\r\n1185: new cable installed.\r\n1255: new connector installed.\r\n1456: new connector installed.\r\n1456: the new IDC connector installed. Flat cable int.\r\n2735: LCD connector broken. Flat cable repair.','65','','',5,3,1,11),(2738,2736,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARMH-0007','2021-07-07','Broken PCB LCD plug.\r\nReplacing the LCD.','','','',8,4,1,11),(2739,2737,2400,NULL,NULL,NULL,21,51,52,'OPTA1','253L 1425 0847 PS','2021-07-08','4-20 faulted\r\n100R, 150R => 2 II 300R','7 548#','','VACON',10,3,1,11),(2740,2738,2400,NULL,NULL,NULL,21,51,52,'OPTA1','253L 919 2406 SUZ','2021-07-08','PCB flashed.','7 548#','','VACON',10,3,1,11),(2741,2739,18000,NULL,NULL,NULL,21,51,13,'NXS00615A2H1SSVA1A2000000','10050727','2021-07-11','Test only','7 548#','','bin',20,20,1,11),(2742,2740,6400,NULL,NULL,NULL,21,263,19,'PC11-F30B-0000','00006013','2021-07-11','Corrosion on CPU and POWER PCB.\r\nInt. temp. sensor 2024-T4\r\nInt. 2 Coolers.','','SCR pdf','',25,12,1,11),(2743,2741,6400,NULL,NULL,NULL,21,263,19,'PC11-F30B-0000','00006017','2021-07-11','Int. 2 Coolers.','','SCR pdf','',10,5,1,11),(2744,2742,4800,NULL,NULL,NULL,21,683,48,'13079','04409','2021-07-12','After heating the power supply stopped at output 24V\r\nInstead TRACO POWER TMS 25124 used RS LHE25-20B24 (181-2130)\r\nTest - ok.','','','',8,4,9,11),(2745,2743,3600,NULL,NULL,NULL,21,500,6,'Gavish 085','299','2021-07-12','299: 485 terminal replaced.\r\nFull test provided.\r\n2743: New terminals installing.\r\nIncorrect nominal and polarity of capacitors used.\r\nRepair.','6','','',9,4,1,11),(2746,2744,12000,NULL,NULL,NULL,21,118,18,'MF/E101374101SS5ER3321111','V/10384/20/1','2021-07-13','1093: Capacitors.\r\n_________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\n\r\n1730: PLCC Socket.\r\n2744: test only','','','',10,2,1,11),(2747,2745,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240G110/2','5909013','2021-07-14','2745: FAN flow corrosion.\r\nInt. print between PCB and ISA-connector (Flashed).\r\nTested. Wile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',40,20,1,11),(2748,2746,2700,NULL,NULL,NULL,21,457,139,'IR140Y-4','0011524723','2021-07-18','220uF 63V\r\nCleaning','133','','',12,8,1,11),(2749,2747,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240G110/2','5908959','2021-07-20','FAN flow corrosion.\r\nInt. print between PCB and ISA-connector (Flashed).\r\nSMD: 2k2, 4148, ZD15V, UC3855.\r\nSS-Resistor 33 Ohm 9W.\r\nTested. Wile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',60,40,1,11),(2750,2748,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARKD-0734','2021-07-22','1868: Corrosion,\r\ncleaning.\r\n2748: Problem of heating the sensor.\r\nReplacing all boards besides CPU-board.','144','','',6,3,1,11),(2751,2749,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240G108/2','6241226','2021-07-22','FAN flow corrosion.\r\nInt. print between PCB and ISA-connector (Flashed).\r\nTested. Wile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',60,40,1,11),(2752,2750,14152,NULL,NULL,NULL,21,35,4,'SGDV-260D11A','D0095B754910001','2021-07-25','1710: 0/996/682/930\r\nModule: PM100RLA120 -ok\r\nFlash on capacitors PCB.\r\nCOOLER 24V, LED, FUSE 63A 660V, PCB.\r\nCorrosion.\r\nCleaning.\r\n2750: COOLER 24V: NMB-MAT MODEL 2410ML-05W-B69 (with digital TACHO) \r\nreplaced to:\r\nRS: 335-7012 TUDC24B4 - without wire TACHO \r\nbut with internal TACHO circuit and 100% compatible.\r\nRS: 191-9277 - pin to pin 100% compatible. \r\nwith 2410ML-05W-B69 (only no pin to pin). \r\nCorrosion.\r\nCleaning.','101','','',20,8,1,11),(2753,2751,4600,NULL,NULL,NULL,21,161,7,'AS-B838-032','0018700','2021-07-26','Configuration:\r\n\r\nOOOO-+00000000-+0000 OOOO-+00000000-+0000\r\n\r\n1768: ch_6 IRFD9120\r\n1861: test ok\r\n2751: ch_22: IRFD9120','','','',9,5,1,11),(2754,2752,14152,NULL,NULL,NULL,21,35,4,'SGDV-260D11A','D009YI218610002','2021-07-26','1706:\r\n0/996/682/930\r\nAXIS-2 3V2\r\nModule: PM100RLA120 -ok\r\nCorrosion!\r\nFlash on power PCB.\r\nLED DC BUS.\r\n2752: COOLER 24V: NMB-MAT MODEL 2410ML-05W-B69 (with digital TACHO) \r\n3 units are replaced to:\r\nRS: 335-7012 TUDC24B4 - without wire TACHO \r\nbut with internal TACHO circuit and 100% compatible.\r\nRS: 191-9277 - pin to pin 100% compatible. \r\nwith 2410ML-05W-B69 (only no pin to pin). \r\nCorrosion.\r\nCleaning.','101','','',20,8,1,11),(2755,2753,14152,NULL,NULL,NULL,21,35,4,'SGDV-260D11A','D0095B754910003','2021-07-27','1709: \r\n0/996/682/930\r\nModule: PM100RLA120 -ok\r\nAXIS-1 2V2\r\nCorrosion!\r\nFlash on power PCB.\r\nFast fuse 63A => 3*25A fast.\r\n2753: COOLER 24V: NMB-MAT MODEL 2410ML-05W-B69 \r\n(with digital TACHO) \r\n3 units are replaced to:\r\nRS: 335-7012 TUDC24B4 - without wire TACHO \r\nbut with internal TACHO circuit and 100% compatible.\r\nRS: 191-9277 - pin to pin 100% compatible. \r\nwith 2410ML-05W-B69 (only no pin to pin). \r\nCorrosion.\r\nCleaning.','101','','',20,8,1,11),(2756,2754,14152,NULL,NULL,NULL,21,35,4,'SGDV-260D11A','D0095B754910005','2021-07-28','2700: 0/996/682/930\r\nModule: PM100RLA120 -ok\r\nCOOLER 24V: NMB-MAT MODEL 2410ML-05W-B69 (with digital TACHO) \r\nreplaced to:\r\nRS: 335-7012 TUDC24B4 - without wire TACHO \r\nbut with internal TACHO circuit and 100% compatible.\r\nRS: 191-9277 - pin to pin 100% compatible.\r\nwith 2410ML-05W-B69 (only no pin to pin). \r\nCorrosion.\r\nCleaning.\r\n2754: COOLER 24V: NMB-MAT MODEL 2410ML-05W-B69 (with digital TACHO) \r\nreplaced to:\r\nRS: 335-7012 TUDC24B4 - without wire TACHO \r\nbut with internal TACHO circuit and 100% compatible \r\nwith 2410ML-05W-B69 (only no pin to pin).','101','','',20,5,1,11),(2757,2755,2000,NULL,NULL,NULL,21,161,22,'AS-B809-016','0024671','2021-07-28','735: Neon lamp -3.\r\nFull test.\r\n2755: Neon lamp -7.','8','','',7,4,5,11),(2758,2756,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301575','2021-08-01','Capacitors.\r\nR1, R2: 100k 2W\r\nR85: 39R 2W\r\nTesting, - ok.','123','','',15,6,1,11),(2759,2757,8200,NULL,NULL,NULL,21,519,18,'8732ESR2A1NAM4C1','0310682','2021-08-02','PS: shorted Ultra Fust Diode on output voltage.\r\nBecause shorting was faulted PWM controller (OPA425R).\r\nReplacing the new one PS board.\r\nTesting.\r\nok','','','',10,6,2,11),(2760,2758,1400,NULL,NULL,NULL,21,687,33,'M5.4443','400040488021110L2012','2021-08-02','2688: Total corrosion\r\nR110, R111: 4M3\r\n1N5408 -4.\r\n2758: 100uF 400V','','','',10,5,1,11),(2762,2760,3285,NULL,NULL,NULL,21,690,6,'Anybus Communicator ModbusPlus S','00026929','2021-08-02','No problem founded.\r\nReplacing the diode S1J to LL4007G','','','',5,3,1,11),(2761,2759,2700,NULL,NULL,NULL,21,457,139,'IR140Y-4','960421630','2021-08-02','Transformer 230VAC/24VAC 2.3VA => 2*12VAC (121-3829)\r\n100uF 63V, DIP_SW, Terminal, Relay 24VDC 2ch/o, \r\nRES_SW: 479-1542\r\nR201.2: 10k (flashed)\r\nCleaning','133','','',12,8,1,11),(2763,2761,2700,NULL,NULL,NULL,21,457,139,'IR140Y-4','960421562','2021-08-04','DIP_SW, Terminal, Relay 24VDC 2ch/o, \r\nBroken knob of trimmer.\r\nCleaning','133','','',12,8,1,11),(2764,2762,5200,NULL,NULL,NULL,21,498,24,'AFM-16','2233253','2021-08-05','2762: Capacitors, 56R -2.','9','','bin',15,9,2,11),(2765,2763,9500,NULL,NULL,NULL,21,116,76,'Lynxbatch','634','2021-08-05','634: Communication test, - ok.\r\nMaking circuit.\r\n______________________________\r\n\r\n8.12.2015 was short with HV and CPU board.\r\nLT1170HVCT => LM2577-ADJ\r\n2478: Li BAT 3.7V\r\n2763: DROM alarm\r\nProgramming the old 93C56N','47','','bin',10,2,1,11),(2766,2764,2000,NULL,NULL,NULL,21,469,76,'PAXS0010','15102','2021-08-08','2162: Corrosion. \r\nCleaning.\r\nTested.\r\n2764: Corrosion. \r\nCleaning.','40','','RedLion',8,5,1,11),(2767,2765,5000,NULL,NULL,NULL,21,612,85,'TMS-M','PM08-70-172','2021-08-09','Regeneration of sensor.','77, 50','CMC','',24,24,1,11),(2768,2766,12500,NULL,NULL,NULL,21,572,76,'FIP 0401 FBW INTECONT PLUS','G0003991','2021-08-11','482: Corrozion.\r\n1uF - broken (circuit LM337T)\r\nSaving data.\r\nCD4046 -2 (impossible test).\r\nMaking circuit.\r\nCleaning.\r\nNo problem found.\r\n2032: No problem found.\r\nX2 - new socket installed.\r\n!!! LC give pulses from controller instead voltage.\r\n2302: BTZ03C36 (V8, V10) broken PIN\'s corrosion.\r\nIonistor 0.47F 5.5V\r\n2766: no problem','44','meir','bin',10,4,1,11),(2769,2767,4500,NULL,NULL,NULL,21,116,98,'M400/2H','1233167024','2021-08-12','Reset sw.\r\nCleaning','72','','',12,1,1,11),(2770,2768,5000,NULL,NULL,NULL,21,612,85,'TMS-M/P/100','NM05-70-155','2021-08-15','Regeneration of sensor.','77, 50','CMC','',24,24,1,11),(2771,2769,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240G110/2','','2021-08-15','FAN flow corrosion.\r\nInt. PIN ISA socket.\r\nUC3855BDW - shorted.\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',40,10,1,11),(2772,2770,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240G110/2','6241230','2021-08-16','FAN flow corrosion.\r\nInt. PIN ISA socket.\r\nUC3855BDW - shorted. IPI60R099CP, Fuse: T20A\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',40,10,1,11),(2773,2771,4300,NULL,NULL,NULL,21,552,32,'multi9 RCI 21181','9806','2021-08-17','Capacitors.\r\nBC547.\r\nSYR001.','30 155','','',10,5,1,11),(2774,2772,5000,NULL,NULL,NULL,21,612,85,'TMS-M','PM08-70-172','2021-08-17','Regeneration of sensor.','77, 50','CMC','',24,24,1,11),(2775,2773,6400,NULL,NULL,NULL,21,263,19,'PC11-F30B-0000','00006018','2021-08-18','1373: Fire on input terminal.\r\nCooler 24V.\r\n2773: Cooler 24V, CPU -faulted because corrosion.\r\nCleaning.','27','SCR pdf','',20,7,1,11),(2776,2774,2200,NULL,NULL,NULL,213,692,35,'SA-19D','QA201207018','2021-08-23','Controller for actuator\r\n15uF 400V => 10uF 400V\r\nFuse 2A','','','',5,3,9,11),(2777,2775,7300,NULL,NULL,NULL,213,691,140,'SAL-0121(11)-C','2008.07.23','2021-08-23','Board for SA05 servomotor.\r\n\r\n150nF 275VAC\r\nF2A\r\nPCB repairing after after heating 13k(?) 5W.','','','',10,5,1,11),(2778,2776,3170,NULL,NULL,NULL,213,654,3,'FPN5228A','10293','2021-08-29','Price of new one: $904\r\nC26, C27: 560uF 200V \r\nC10, C13, C31: 3300uF 25V\r\nR1, R86: 27k 2W.\r\nR62, R60, R21: 220R 2W\r\nOptocouplers CNY17-3  -3.\r\nUC3842\r\nC41: 1uF 50V, C23: 47uF 25V, C25: 56uF 35V, C19: 220uF 35V.\r\nOutput voltage: 13.7V','107','','',20,15,1,11),(2779,2777,3170,NULL,NULL,NULL,213,654,3,'FPN5228A','17285','2021-08-29','Price of new one: $904\r\nC26, C27: 560uF 200V \r\nC10, C13, C31: 3300uF 25V\r\nR1, R86: 27k 2W.\r\nR62, R60, R21: 220R 2W\r\nOptocouplers CNY17-3  -3.\r\nUC3842\r\nC41: 1uF 50V, C23: 47uF 25V, C25: 56uF 35V, C19: 220uF 35V.\r\nOutput voltage: 13.7V','107','','',20,15,1,11),(2780,2778,3170,NULL,NULL,NULL,213,654,3,'FPN5522AA','2778','2021-08-30','Price of new one: $904\r\nUnit after non qualified repair and over heating.\r\n   \r\n300uF 200V -2; 470uF 25V\r\n1.5KE200CA to 6KE200  -2.\r\nPrint repair.  \r\nOptocouplers CNY17-3  -3.\r\nFuse T6A => F10A, NTC 10R => NTC 4R; LED.\r\nOutput voltage: 14V.','107','','',20,15,1,11),(2781,2779,2400,NULL,NULL,NULL,21,457,139,'IR140Y-4','960421607','2021-08-30','100uF 63V\r\nTactile switch.\r\nFull testing.','133','','',10,5,1,11),(2782,2780,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240G110/2','5909029','2021-09-01','FAN flow corrosion.\r\nInt. PIN ISA socket.\r\n470k SMD open. \r\nUC3855BDW - shorted. IPW60R045CP (752-8397) -2, \r\nUsed the new silicone paste for PFC and Output Side PS\r\nFuse: T20A\r\n\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',40,30,1,11),(2783,2781,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240G110/2','6241227','2021-09-01','FAN flow corrosion.\r\nInt. PIN ISA socket.\r\n470k SMD open. \r\nUC3855BDW - shorted.  \r\nUsed the new silicone paste for PFC and Output Side PS\r\n\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',40,30,1,11),(2784,2782,4700,NULL,NULL,NULL,21,530,3,'140CPS22400','20000317','2021-09-01','100nF 63V - faulted, was with high leakage:\r\nThe are is near control MOSFET transformer.','134','','',50,50,2,11),(2785,2783,3755,NULL,NULL,NULL,21,533,17,'MODICON TM221CE40R','IB184600137','2021-09-05','Regeneration.','','','',10,6,1,11),(2786,2784,6400,NULL,NULL,NULL,21,263,19,'PC11-F30B-0000','00006017','2021-09-09','2741: Int. 2 Coolers.\r\n2784: CPU corrosion, cleaning.','','SCR pdf','',10,5,1,11),(2787,2785,6400,NULL,NULL,NULL,21,263,19,'PC11-F30B-0000','00006016','2021-09-09','2785: Int. 2 Coolers.\r\nCPU corrosion, cleaning.','','SCR pdf','',10,5,1,11),(2788,2786,8200,NULL,NULL,NULL,21,519,18,'8732ESR2A1NOM4C1','0292506','2021-09-12','Testing.\r\nok','','','',10,6,2,11),(2789,2787,4700,NULL,NULL,NULL,21,519,41,'3081','743','2021-09-13','743:\r\nCorrosion.\r\nAAA, green LED.\r\nCleaning.\r\n1116: \r\nCorrosion,\r\nAAA.\r\n2787: \r\nR2 100k (int)\r\nCorrosion,\r\nAAA.','','','',5,3,5,11),(2790,2788,14100,NULL,NULL,NULL,21,531,98,'3081','A00-36344','2021-09-13','1760: Cleaning. \r\nAdd screws, cables, inputs.\r\nResistor 2 GOhm - broken.\r\nTest without calibration.\r\nTest after calibration.\r\n2788: Add screws.\r\nCalibration.','35','book 3081','',20,8,1,11),(2791,2789,12000,NULL,NULL,NULL,21,507,76,'AD-4322A','C7916067','2021-09-14','60: Interrupted power plug.\r\nCapacitors.\r\nTest.\r\nBridge color wire load cell connection:\r\n1, 2 - green exc+ (+12V),\r\n3, 4 - black exc- (0V),\r\n5 - white sig+,\r\n6 - red sig -.\r\n\r\n2789: Complaint about accuracy of weight. \r\nWas replaced LC-input module from another unit, \r\nfew capacitors 33uF, and cleaning.\r\n\r\nStay problem with VFD-controller because artefacts on display (DP)\r\nChip: HD6433712B90H','','PDF/AD-4322','BIN/Magnesium/ad4322/27512',30,20,1,11),(2792,2790,2230,NULL,NULL,NULL,21,118,8,'SPTU 240 R1','9258','2021-09-29','Capacitors.','','','',7,3,2,11),(2793,2791,30000,NULL,NULL,NULL,21,530,35,'140CPU67261','21113012331','2021-09-29','Corrosion under influence water.\r\nCleaning.\r\nTest - ok.','','','',30,10,2,11),(2794,2792,4700,NULL,NULL,NULL,21,530,3,'140CPS12420','330004840932902800','2021-09-29','100nF 63V - faulted, was with high leakage:\r\nThe are is near control MOSFET transformer.','134','','',50,50,2,11),(2795,2793,3700,NULL,NULL,NULL,21,678,5,'DTV-V109 99992109','2U2','2021-09-29','100uF 25V (was not tested)','','','',3,2,1,11),(2796,2794,14600,NULL,NULL,NULL,21,530,3,'AS-P840-000','5054','2021-09-30','2563: Capacitors\r\nD104: MBR4050PT => STPS61150CW\r\n\r\n2794: D101: MBR4050PT => STPS61150CW','123','','',25,12,1,11),(2797,2795,2800,NULL,NULL,NULL,21,653,18,'NL-3313LC12','A20010963','2021-10-03','Flash on terminal','','','',10,5,1,11),(2798,2796,14100,NULL,NULL,NULL,21,531,98,'3081 PH/0RP','A99-18480','2021-10-03','1361: Cleaning the corrosion on terminal \r\nand making the new connection.\r\nNew screws are installed.\r\nTest, - ok.\r\n2796: IR not working.\r\nReplacing the another one. \r\nCalibration.\r\nTest, - ok.\r\n\r\n\r\n\r\n\r\n\r\n502: new problem found: leakadge between input pin\'s.\r\n1063: new screws are installed.','35','3081 book','',10,5,1,11),(2799,2797,9500,NULL,NULL,NULL,21,116,76,'Lynxbatch','634','2021-10-04','634: Communication test, - ok.\r\nMaking circuit.\r\n______________________________\r\n\r\n8.12.2015 was short with HV and CPU board.\r\nLT1170HVCT => LM2577-ADJ\r\n2478: Li BAT 3.7V\r\n2763: DROM alarm\r\n2797: Programming the new U13 from new another DUMP.\r\n93C56N stay old program.','47','','bin',30,12,1,11),(2800,2798,18000,NULL,NULL,NULL,21,51,13,'NXS01055A5H0SSVA1A2000000','12914211','2021-10-05','1659: VFD without problem but without BRAKE function.\r\n2798: flashed because shorted motor.\r\nReplacing the SS-board.\r\nModule SKKH92/16E -2.\r\nCleaning and testing.','7','','bin',20,10,1,11),(2801,2799,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','5186-95','2021-10-06','Capacitors,\r\nrelays 12V -6.','53','','SOLCON\\MRP2000_D\\X2816.bin',30,10,1,11),(2802,2800,8200,NULL,NULL,NULL,21,51,13,'2.2CXL4G5I1','3307593','2021-10-06','65:\r\nReplacing and programming EEPROM\r\nAT28C64B.\r\nLCD panel repairing.\r\nReplacing cooler 24VDC on 2 coolers 12VDC.\r\n\r\n912: Short FAN 24V\r\n1020:\r\nFAN 24V, \r\nMaking new cables.\r\n2800: FAN 24V, Flash on the input filter, 1uF 250VAC','7 548#','','bin',20,10,1,11),(2803,2801,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARMH-0044','2021-10-07','Cleaning.\r\nTest - ok.','144','','',6,3,1,11),(2804,2802,8800,NULL,NULL,NULL,21,51,13,'NXS00075A5H1SSV','10159379','2021-10-07','40: Replacing power terminal\r\nfor brake resistor.\r\nReplacing screws for cover.\r\nSystem error 8.\r\nStill problem.\r\nAltera PLD - bad.\r\nCPU board must be changed.\r\n1657: CPU board was changed.\r\nOthers: Central cooler, AT24C02 bad in board OPTA_1.\r\n2712: because short in the brake resistor start resistor flashed.\r\n2802: brake resistor start resistor flashed.','7 548#','','',10,5,2,11),(2805,2803,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301612','2021-10-10','Capacitors.\r\nR1, R2: 100k 2W\r\nR85: 39R 2W\r\nTesting, - ok.','123','','',15,6,1,11),(2806,2804,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','304580','2021-10-10','Capacitors.\r\nR1, R2: 100k 2W\r\nR85: 39R 2W\r\nTesting, - ok.','123','','',15,6,1,11),(2807,2805,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','5187-95','2021-10-11','Capacitors,\r\nrelays 12V -6.','53','','SOLCON\\MRP2000_D\\X2816.bin',30,10,1,11),(2808,2806,4600,NULL,NULL,NULL,21,161,22,'AS-B827-032','0078394','2021-10-11','Cleaning\r\nFull test provided.','41','','',15,8,1,11),(2809,2807,3000,NULL,NULL,NULL,21,504,128,'WAS4 PRO DC/DC','2807','2021-10-12','Making the new project for 130R to 4-20mA converting.','135','','',25,15,1,11),(2810,2808,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','309240','2021-10-12','Capacitors.\r\nR85: 39R 2W\r\nTesting, - ok.','123','','',15,6,1,11),(2811,2809,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301590','2021-10-13','Capacitors.\r\nR1, R2: 100k 2W\r\nR85: 39R 2W\r\nTesting, - ok.','123','','',15,6,1,11),(2812,2810,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240G110/2','6241223','2021-10-14','FAN flow corrosion.\r\nInt. print.\r\n470k SMD open. \r\nUsed the new silicone paste for PFC and output voltage side PS\r\n\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',40,10,1,11),(2813,2811,8000,NULL,NULL,NULL,21,693,141,'VEGAPULS64 PS64.AXFGIJHXKMAXX','34042232','2021-10-14','Corrosion.','','','',3,1,1,11),(2814,2812,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','5186-95','2021-10-17','2799: Capacitors,\r\nrelays 12V -6.\r\n2812: new LCD used \r\n(532-6486 pin to pin compatible instead +/- backlight)','53','','SOLCON\\MRP2000_D\\X2816.bin',30,10,1,11),(2815,2813,1200,NULL,NULL,NULL,21,648,35,'JDR10','26800','2021-10-17','Capacitors','','','',2,1,1,11),(2816,2814,1200,NULL,NULL,NULL,21,648,35,'JDR10','26808','2021-10-18','Capacitors','','','',2,1,1,11),(2817,2815,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240G110/2','6242409','2021-10-18','FAN flow corrosion.\r\nInt. print.\r\n470k, 6k8, 4k3 (4k7 II 51k) SMD are  open. \r\nUsed the new silicone paste for PFC and output voltage side PS\r\n\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',50,20,1,11),(2818,2816,12000,NULL,NULL,NULL,21,118,18,'MF/E400372101SS5ER3321111','V/10384/19/2','2021-10-20','1853: _________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\n2816: test - ok.','','','',10,6,1,11),(2819,2817,900,NULL,NULL,NULL,21,504,128,'WAS5 CCC 2OLP','2817','2021-10-20','Terminal.','','','',4,2,9,11),(2820,2818,8000,NULL,NULL,NULL,21,639,35,'FRS10','48939HS1','2021-10-26','1605: Capacitors.\r\n2818: cleaning, testing.','26','','',6,3,1,11),(2821,2819,4300,NULL,NULL,NULL,21,552,32,'multi9 RCI 21181','9547','2021-10-26','2819: Label: K12\r\n\r\nCapacitors.\r\nBC547 => BC639.\r\nSYR001.\r\n22R\r\nTest 1ADC','30 155','','',10,5,1,11),(2822,2820,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5','6035-96','2021-10-28','Capacitors,\r\nrelays 12V -6.\r\n2820: problem with accuracy of the voltage monitoring.\r\nProgramming the X2816 and M27C512 the data from folder MPR2000\r\nOld sticker on EPROM is MPC080196 and old DUMP was saving to folder\r\nMPR2000_5.\r\nIn analog board was replacing LM324  -7.','53','','SOLCON\\MRP2000_D\\X2816.bin',30,10,1,11),(2823,2821,22000,NULL,NULL,NULL,21,694,13,'CMI 6326-02','2821','2021-10-31','Alarm OT\r\nPre-driver board (MAPS-3 MD-B3014E 0-48680-213) are faulted.\r\nAll capacitors bad with electrolytic nozzle.\r\nInt. print from lines of PS.\r\n20R 5W -2, 1k5 0.5W -4, UF4005 -5, 100uF 400V.','','','',20,10,1,11),(2824,2822,1800,NULL,NULL,NULL,21,498,43,'ACT-2-I','519818','2021-11-02','0-5A => 4-20mA\r\n\r\nU1: LT1013 -ok\r\nU2: CA3130 -bad\r\n\r\nCalibrated:\r\nI(mA)=I(in)*16/5+4','136','','',12,10,1,11),(2825,2823,18000,NULL,NULL,NULL,21,548,6,'HART 275','1364','2021-11-07','295: Repair input DC terminal.\r\n1364: Installed the new battery pack:\r\ninstead 1.2V 700mA Ni-Cd SANYO N-700AAC => RKB 49AA 1500 NiMH 1.2V\r\nbattery from remote control RF\r\n2823: Charging after very low battery level.','','','',25,15,1,11),(2826,2824,2400,NULL,NULL,NULL,21,51,52,'OPTA1','253J 6115368 S-KOS','2021-11-07','24C02 - are shorted  (the new one programming)','7 548#','','VACON',10,2,1,11),(2827,2825,2400,NULL,NULL,NULL,21,51,52,'OPTA1','253J 6055828 S-KOS','2021-11-07','24C02 - are shorted  (the new one programming)\r\nAll DIN-connectors and jumpers changed.\r\nTesting.','7 548#','','VACON',10,2,1,11),(2828,2826,2400,NULL,NULL,NULL,213,51,52,'OPTA1','253L 1107 0649 PS','2021-11-07','24C02 - bad  (the new one programming)\r\nTesting.','7 548#','','VACON',10,2,1,11),(2829,2827,2400,NULL,NULL,NULL,21,51,52,'OPTA1','253L 1107 0724 PS','2021-11-07','24C02 - flashed bad  (the new one programming)\r\n2 FET\'s on I2C BUS flashed, aborted from circuit, installed  2 shorts\r\ninstead FET-channel.\r\nTesting.','7 548#','','VACON',10,2,1,11),(2830,2828,2400,NULL,NULL,NULL,21,51,52,'OPTA1','253L 1243 1006 PS','2021-11-07','2828: 24C02 - flashed bad  (the new one programming)\r\nIDC-connectors and jumpers changed. .\r\nTesting.','7 548#','','VACON',10,2,1,11),(2831,2829,30000,NULL,NULL,NULL,21,560,6,'HART 375 Field Communicator','12072847','2021-11-08','390:  Touch screen AMT98234 replaced.\r\n1016: Braked DC plug (RS: 526-0825).\r\n1117: Making new battery block 1.2/1500mA*5 NiMH\r\n1277: Test only.\r\n1540: system card error.Bad placing becase shock during after fall.\r\n1576: recovery system provided. Flashing OS from internal dump.\r\nMaking the new cable.\r\n1628: Installed the new power socket. Soldering the MB, connectors\r\nSMD components. Re-Flashing EEPROM.\r\n1967: total fault. Changed battery pack without battery, \r\n- only board with closed module.\r\n2829: Very low battery level. PS-plug changed. \r\nCleaning and charging. \r\nTesting.','60','','',40,30,1,11),(2832,2830,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM 01','1896515538','2021-11-08','Capacitors\r\nFlash tantalum 15uF 25V => 22uF 35V \r\nProgramming the new EEPROM by 0P revision','32','','bin',10,3,1,11),(2833,2831,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM 01','1895192097','2021-11-08','Capacitors\r\nProgramming the new EEPROM by 0L revision','32','','bin',10,3,1,11),(2834,2832,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0R FBM 1','18981130972','2021-11-09','Capacitors\r\nBad EEPROM: NQ28C64A => AT28C64B\r\nProgramming rev 0R','32','','bin',10,3,1,11),(2835,2833,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515466','2021-11-09','2436: ad capacitors.\r\nBad tantalum 15uF 35V => 22uF 35V.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P\r\n2833: 49R9 - flashed (ch_7)','32','','',10,5,1,11),(2836,2834,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896505284','2021-11-09','Bad tantalum 15uF 35V => 22uF 35V -2.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2837,2835,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896464479','2021-11-09','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2838,2836,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896495037','2021-11-09','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2839,2837,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515500','2021-11-09','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2840,2838,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896464415','2021-11-10','2838: Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2841,2839,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896495109','2021-11-10','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2842,2840,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896464478','2021-11-10','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2843,2841,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896495069','2021-11-10','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2844,2842,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515539','2021-11-10','2842: Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2845,2843,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515526','2021-11-11','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2846,2844,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896505243','2021-11-11','2844: Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2847,2845,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515469','2021-11-11','2845: Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2848,2846,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515527','2021-11-11','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2849,2847,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1897035900','2021-11-11','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2850,2848,8500,NULL,NULL,NULL,21,564,98,'MSP110-11-C','SW041103/04MW','2021-11-15','Capacitors.\r\nTested.','','','Mobrey',10,4,1,11),(2851,2849,3000,NULL,NULL,NULL,21,539,54,'Fluke 52','38620556WS','2021-11-15','2849: Making TC-terminal.','','','',5,4,5,11),(2852,2850,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896505242','2021-11-17','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2853,2851,80450,NULL,NULL,NULL,21,390,98,'1700R12ABAEZZZ','3167736','2021-11-28','PS-board','137','','',10,3,1,11),(2854,2852,2200,NULL,NULL,NULL,21,533,3,'ABL 1REM24062','00002445','2021-11-30','All capacitors\r\nTest.','','','',10,5,1,11),(2855,2853,80450,NULL,NULL,NULL,21,390,98,'1700R12ABAEZZZ','3105617','2021-11-30','PS-board\r\nMaking the circuit','137','','',10,20,1,11),(2856,2854,8000,NULL,NULL,NULL,21,695,11,'FHC50','FHC50N1343006','2021-12-05','ADM2483 flashed from A_ch outside','138','','pdf',10,2,1,11),(2857,2855,8000,NULL,NULL,NULL,21,695,11,'FHC50','FHC50N132040','2021-12-05','? problem with flow.\r\nMaking the circuit, providing the simulation AI/AO.\r\nNo problem founded.\r\nReplacing LM324 -2. DG419 stay original without replacing.','138','','pdf',40,25,1,11),(2858,2856,2230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','272248','2021-12-06','Capacitors.\r\nRelays.','','','',7,3,2,11),(2859,2857,15000,NULL,NULL,NULL,21,485,87,'Esteem 96C','9C 0905','2021-12-07','Corrosion.\r\nLED red, all capacitors, DIP-sw 8ch.\r\nReplacing RF BNC PCB to bayonet (759-5271).\r\n\r\nDIP-switch configuration:\r\nON: 1,4,5,6,7\r\nOFF: 2,3,8','','','bin',30,20,1,11),(2860,2858,12000,NULL,NULL,NULL,21,118,18,'MF/E201366111ss','V/89041/7/1','2021-12-09','_____________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\n2277: C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\n2711: test only.','','','',10,1,1,11),(2861,2859,5200,NULL,NULL,NULL,21,498,24,'AFM-16','630543','2021-12-12','Yellow LED, 56R2 -2, all capacitors.','9','','',7,5,9,11),(2862,2860,5500,NULL,NULL,NULL,213,58,17,'Micro Logix 1400  1766-L32AWA PN','6L32AWA B22060907','2021-12-12','SS-capacitor 47uF 50V','','','',5,2,1,11),(2863,2861,2600,NULL,NULL,NULL,21,696,35,'AI936 Nr: 0001168','9069 55','2021-12-13','Capacitors','','','',4,2,1,11),(2864,2862,2230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','310383','2021-12-13','Capacitors.\r\nRelays.','','','',7,3,2,11),(2865,2863,2230,NULL,NULL,NULL,21,118,8,'SPTU 240 R1','10296','2021-12-13','Capacitors.','','','',7,3,2,11),(2866,2864,2230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','274050','2021-12-20','Electrican Laboratory.\r\nRelay.\r\nCapacitors.','','','',7,3,2,11),(2867,2865,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896505288','2021-12-20','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2868,2866,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515531','2021-12-20','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2869,2867,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0R FBM 1','SF98156933','2021-12-21','Capacitors\r\nBad EEPROM: NQ28C64A => AT28C64B\r\nProgramming rev 0R','32','','bin',10,3,1,11),(2870,2868,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0N FBM 1','1895263031','2021-12-21','Capacitors\r\nBad EEPROM: NQ28C64A => AT28C64B\r\nProgramming rev 0N','32','','bin',10,3,1,11),(2871,2869,1450,NULL,NULL,NULL,21,697,3,'TP-100B','RB18034834','2021-12-21','Capacitors','','','',5,2,1,11),(2872,2870,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0N FBM 1','1890340797','2021-12-22','Capacitors\r\nBad EEPROM: NQ28C64A => AT28C64B\r\nProgramming rev 0N','32','','bin',10,3,1,11),(2873,2871,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM 01','1895192134','2021-12-22','Capacitors\r\nProgramming the new EEPROM by 0L revision','32','','bin',10,3,1,11),(2874,2872,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM 01','1895191972','2021-12-22','Capacitors\r\nProgramming the new EEPROM by 0L revision','32','','bin',10,3,1,11),(2875,2873,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM 01','1895192094','2021-12-22','Capacitors\r\nProgramming the new EEPROM by 0L revision','32','','bin',10,3,1,11),(2876,2874,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','0044504','2021-12-23','Configuration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1\r\n        \r\nTested.','','','',9,5,1,11),(2877,2875,2000,NULL,NULL,NULL,21,161,22,'AS-B809-016','0027437','2021-12-23','1138: Neon lamp -8.\r\nFull test.\r\n\r\n2875: 22M -16.\r\nFull test.','8','','',7,4,2,11),(2878,2876,2400,NULL,NULL,NULL,21,58,6,'Stratix 2000','SV1FG0WF','2021-12-26','Replacing the resonator 25mHz','','','',10,5,9,11),(2879,2877,14100,NULL,NULL,NULL,21,531,98,'3081','AIT3012/2','2021-12-27','278:\r\nCleaning the corrosion on terminal.\r\nRepair screws.\r\nTest, - ok.\r\n______________________________\r\n501:\r\nReplacing CPU\r\n______________________________\r\n707: Repair input terminal, cleaning.\r\n_____________________________________\r\n2877: New screws installed. Cleaning and calibration.','35','','',10,3,1,11),(2880,2878,6700,NULL,NULL,NULL,21,116,98,'pH 2100/2','54268/1533600/MT','2021-12-27','Cleaning.\r\nReplacing the front case panel.\r\nCalibration (1100).','114','','',15,5,1,11),(2881,2879,2400,NULL,NULL,NULL,21,499,12,'Dosimat 725','13251','2021-12-28','Corrosion.\r\nPotentiometer 400k repair.\r\n47uF 25V','22','','',8,4,1,11),(2882,2880,6700,NULL,NULL,NULL,21,116,98,'pH 2100/2','54268/1533223/MT','2021-12-28','Cleaning.\r\nCalibration (1100).','114','','',15,5,1,11),(2883,2881,6700,NULL,NULL,NULL,21,116,98,'pH 2100/2','54268/1533600/MT','2021-12-29','Cleaning.\r\nCalibration (1100).','114','','',15,5,1,11),(2884,2882,240,NULL,NULL,NULL,21,698,108,'Yupiter 623L OSRAM LED 5W','2882','2021-12-29','Changing input voltage from 230VAC to 48VAC.\r\nAborting the internal UPS.\r\n750...800 Ohm 2W + bridge (instead 100mA used 50mA LED current).','','','',3,1,1,11),(2885,2883,240,NULL,NULL,NULL,21,698,108,'Yupiter 623L OSRAM LED 5W','2883','2021-12-29','Changing input voltage from 230VAC to 48VAC.\r\nAborting the internal UPS.\r\n750...800 Ohm 2W + bridge (instead 100mA used 50mA LED current).','','','',3,1,1,11),(2886,2884,240,NULL,NULL,NULL,21,698,108,'Yupiter 623L OSRAM LED 5W','2884','2021-12-29','Changing input voltage from 230VAC to 48VAC.\r\nAborting the internal UPS.\r\n750...800 Ohm 2W + bridge (instead 100mA used 50mA LED current).','','','',3,1,1,11),(2887,2885,240,NULL,NULL,NULL,21,698,108,'Yupiter 623L OSRAM LED 5W','2885','2021-12-29','Changing input voltage from 230VAC to 48VAC.\r\nAborting the internal UPS.\r\n750...800 Ohm 2W + bridge (instead 100mA used 50mA LED current).','','','',3,1,1,11),(2888,2886,240,NULL,NULL,NULL,21,698,108,'Yupiter 623L OSRAM LED 5W','2886','2021-12-29','Changing input voltage from 230VAC to 48VAC.\r\nAborting the internal UPS.\r\n750...800 Ohm 2W + bridge (instead 100mA used 50mA LED current).','','','',3,1,1,11),(2889,2887,240,NULL,NULL,NULL,21,698,108,'Yupiter 623L OSRAM LED 5W','2887','2021-12-29','Changing input voltage from 230VAC to 48VAC.\r\nAborting the internal UPS.\r\n750...800 Ohm 2W + bridge (instead 100mA used 50mA LED current).','','','',3,1,1,11),(2890,2888,240,NULL,NULL,NULL,21,698,108,'Yupiter 623L OSRAM LED 5W','2888','2021-12-29','Changing input voltage from 230VAC to 48VAC.\r\nAborting the internal UPS.\r\n750...800 Ohm 2W + bridge (instead 100mA used 50mA LED current).','','','',3,1,1,11),(2891,2889,240,NULL,NULL,NULL,21,698,108,'Yupiter 623L OSRAM LED 5W','2889','2021-12-29','Changing input voltage from 230VAC to 48VAC.\r\nAborting the internal UPS.\r\n750...800 Ohm 2W + bridge (instead 100mA used 50mA LED current).','','','',3,1,1,11),(2892,2890,4200,NULL,NULL,NULL,21,326,3,'DLP240-24-1','0ED-451CI8-0096W2304','2021-12-30','Corrosion.\r\nCapacitors.\r\nCase repair.','','','',8,4,9,11),(2893,2891,4800,NULL,NULL,NULL,21,589,67,'S01E00','PC0MRMDT','2021-12-30','Cleaning MB and USB.\r\nFront side USB stay with problem of contacts.\r\nNew CPU grease applied.','','','',5,3,1,11),(2894,2892,17500,NULL,NULL,NULL,21,530,1,'CPU 434 12A','31012630340','2021-12-30','Cleaning','','','',12,4,9,11),(2895,2893,5200,NULL,NULL,NULL,21,498,24,'AFM-16','630550','2022-01-04','2018: Corrosion!\r\nConnectors, capacitors, EPROM repair, sockets.\r\n2893: no problem founded.','9','','bin',10,2,2,11),(2896,2894,5200,NULL,NULL,NULL,21,498,24,'AFM-16','2233253','2022-01-04','2762: Capacitors, 56R -2.\r\n2894: VN10KM => BS170','9','','bin',15,9,2,11),(2897,2895,15000,NULL,NULL,NULL,21,555,19,'10007 26/99','45143/04RS','2022-01-06','371: Lost phase alarm.\r\n\r\n220uF 35V -3,\r\n10uF 50V -8.\r\nInrerupted pin tranfformer (cool soldering).\r\n\r\n2895: Lost phase alarm.\r\n220uF 35V -3,\r\n10uF 50V -8.\r\nSFH617','141','','',15,3,1,11),(2898,2896,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515521','2022-01-09','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2899,2897,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0R FBM 1','SF98177139','2022-01-09','Capacitors\r\nBad EEPROM: NQ28C64A => AT28C64B\r\nProgramming rev 0R','32','','bin',10,3,1,11),(2900,2898,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0R FBM 1','SF98156959','2022-01-10','Capacitors\r\nBad EEPROM: NQ28C64A => AT28C64B\r\nProgramming rev 0R','32','','bin',10,3,1,11),(2901,2899,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0R FBM 1','SF98156950','2022-01-10','Capacitors\r\nBad EEPROM: NQ28C64A => AT28C64B\r\nProgramming rev 0R','32','','bin',10,3,1,11),(2902,2900,870,NULL,NULL,NULL,21,518,11,'201','A151452','2022-01-11','1508: Tact switch, jampers, terminal.\r\n\r\n4mA => 0\r\n20mA => 100\r\n\r\n2900: Tact switch, jampers, terminal.\r\n\r\n4mA => 0\r\n20mA => 100','pdf','','bin',6,4,2,11),(2903,2901,2200,NULL,NULL,NULL,21,631,60,'ST-1005-8/2-2/1','1305300988','2022-01-12','BY228 -1 (1500V 3A)\r\nVaristor K385 -4 => V510LA40AP -4.\r\nPCB flashed.','','','',4,1,1,11),(2904,2902,15000,NULL,NULL,NULL,213,99,66,'bridge 80PF80 80PFR80','2902','2022-01-16','Replacing the new rectifiers  80PF80 80PFR80 and varistor','','','',20,10,1,11),(2905,2903,9500,NULL,NULL,NULL,21,329,11,'MPR2000/D-H-5R-2-M-S','5179-95','2022-01-17','1053: Capacitors, F500 MA,\r\nFlash on PCB,\r\nRelay 12V -6.\r\n\r\n2903: PS: 2200uF 16V => 2200uF 25V, Relay 12V -2, MC7915','53','','SOLCON\\MRP2000_D\\X2816.bin',20,10,1,11),(2906,2904,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','0026434','2022-01-18','436: Full test.\r\nNeon lamp -8.\r\n2904: Neon lamp -3,\r\n22M -12.','8','','',7,4,1,11),(2907,2905,4600,NULL,NULL,NULL,21,161,22,'AS-B827-032','0080275','2022-01-19','1309: Cleaning.\r\nTest.\r\n2905: corrosion (LM339)\r\nTesting.','41','','',15,5,2,11),(2908,2906,6920,NULL,NULL,NULL,21,272,6,'TSXCUSBMBP','21072201141','2022-01-19','PLCC socket was broken.\r\nAborting the all PLCC sockets.\r\nThe dump of AM29F040B was saved.\r\nCleaning and testing with M-BUS (PLDD not running without PC)','86','','',10,6,1,11),(2909,2907,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0R FBM 1','SF98156965','2022-01-23','Capacitors\r\nEEPROM: NQ28C64A => AT28C64B\r\nProgramming rev 0R','32','','bin',10,3,1,11),(2910,2908,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896495070','2022-01-23','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2911,2909,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896494960','2022-01-23','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2912,2910,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896495050','2022-01-24','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2913,2911,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515509','2022-01-24','2911: Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(2914,2912,14300,NULL,NULL,NULL,21,396,87,'E-ED03A0 V1/2','0358651992','2022-01-24','Relay NAIS SF2-DC 12V.','24','','',3,1,1,11),(2915,2913,7300,NULL,NULL,NULL,21,58,122,'BULLETIN 1409 ARCING GROUND FAUL','2913','2022-01-25','Interrupted the pin of transistor.\r\nTotal corrosion\r\nCleaning\r\n(!120VAC)','','','',20,10,2,11),(2916,2914,6970,NULL,NULL,NULL,21,434,3,'EWS600-24','731Z02-0024W','2022-01-30','1124: FAN 12V\r\n2052: FAN, capacitors.\r\n2914: FAN.','62','','',6,4,2,11),(2917,2915,20000,NULL,NULL,NULL,21,607,98,'Cerabar S PMC71-5CQ82/101','RA10300109C','2022-02-01','Making the mobile version','139','','',60,40,1,11),(2918,2916,14000,NULL,NULL,NULL,21,236,13,'VFD110E43A 11kW/15HP','110E43A7W3460045','2022-02-01','Int. SMD resistor 1k (current sense circuit UC3842)','140','','',30,10,1,11),(2919,2917,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM 01','1895192092','2022-02-02','Capacitors\r\nProgramming the new EEPROM by 0L revision','32','','bin',10,3,1,11),(2920,2918,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM 01','1895192043','2022-02-02','Capacitors\r\nProgramming the new EEPROM by 0L revision','32','','bin',10,3,1,11),(2921,2919,11600,NULL,NULL,NULL,21,700,65,'DR 6/A','60349','2022-02-03','Repair the knobs\r\nWire resistor 1 OHm 0.1% -2 => 1 OHm 1% -2 (was flashed)','','','',10,6,1,11),(2922,2920,1500,NULL,NULL,NULL,21,175,3,'DPP240-24-1','R080663-960130080737','2022-02-06','SS 100uF 35V','','','',10,5,1,11),(2923,2921,1500,NULL,NULL,NULL,21,599,8,'MSR 020','M2-U3','2022-02-06','100uF 50V, 330uF 63V => 330uF 50V 125C\r\nRelay 24V -2.\r\nTr-r: 230V/24V/3W.','','','',5,4,1,11),(2924,2922,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','0039308','2022-02-07','10M  -9\r\n\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1\r\n        \r\nTested.','','','',9,5,1,11),(2925,2923,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM 01','1895192115','2022-02-07','Capacitors\r\nProgramming the new EEPROM by 0L revision','32','','bin',10,3,1,11),(2926,2924,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM 01','1895192062','2022-02-07','Capacitors\r\nProgramming the new EEPROM by 0L revision','32','','bin',10,3,1,11),(2927,2925,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM 01','1895192109','2022-02-08','Capacitors\r\nProgramming the new EEPROM by 0L revision','32','','bin',10,3,1,11),(2928,2926,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM 01','1895192022','2022-02-08','Capacitors\r\nProgramming the new EEPROM by 0L revision','32','','bin',10,3,1,11),(2929,2927,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM 01','1895192070','2022-02-08','Capacitors\r\nProgramming the new EEPROM by 0L revision','32','','bin',10,3,1,11),(2930,2928,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM 01','1895192093','2022-02-10','Capacitors\r\nProgramming the new EEPROM by 0L revision','32','','bin',10,3,1,11),(2931,2929,13943,NULL,NULL,NULL,21,664,101,'DSF-2010-MB P/N: 2653-271-06','500977','2022-02-13','The input wire +24V was cut and others wires was \r\ndefective. \r\nCapacitor C113 10uF was outside from PCB.\r\nBecause current output was 4.43mA system was defaulted.\r\n\r\nTested with PC, - ok.\r\nPassword: 123456\r\nAddress 4','116','','',10,6,1,11),(2932,2930,10500,NULL,NULL,NULL,213,533,95,'140DAO84000','WX172001135','2022-02-13','K275 SMD varistor short.\r\nReplaced to S14K275','','','',5,1,1,11),(2933,2931,11160,NULL,NULL,NULL,213,319,142,'LB4401','165348-0556','2022-02-15','Int. transformer PIN.\r\nFuse holder was broken.','','','',5,2,1,11),(2934,2932,11160,NULL,NULL,NULL,213,319,142,'LB444-01','60023272-6968','2022-02-16','LT1170HVCT => LT1170CT.\r\nT800mA => T1A, capacitors.','','','',15,6,1,11),(2935,2933,6700,NULL,NULL,NULL,213,116,120,'M300 Multi 2 Channel 1/2DIN','1233258027','2022-02-16','All capacitors was changed (PS+CPU).','','','',5,2,1,11),(2936,2934,3170,NULL,NULL,NULL,213,654,3,'FPN5228A','11360','2022-02-16','Price of new one: $904\r\nUnit after non qualified repair and over heating.\r\nC10, C13, C31: 3300uF 25V\r\nR1, R86: 27k 2W.\r\nR62, R60, R21: 220R 2W\r\nOptocouplers CNY17-3  -3.\r\nC41: 1uF 50V, C23: 47uF 25V, C25: 56uF 35V => 100uF 50V,\r\nC19: 220uF 35V.\r\nOutput voltage: 13.7V','107','','',20,6,1,11),(2937,2935,3170,NULL,NULL,NULL,213,654,3,'FPN5228A','18288','2022-02-17','Price of new one: $904\r\nUnit after non qualified repair and over heating.\r\n560uF 250V  -2.\r\nC10, C13, C31: 3300uF 25V\r\nR1, R86: 27k 2W.\r\nR62, R60, R21: 220R 2W\r\nOptocouplers CNY17-3  -3.\r\nC41: 1uF 50V, C23: 47uF 25V, C25: 56uF 35V => 100uF 50V,\r\nC19: 220uF 35V.\r\nOutput voltage: 13.7V','107','','',20,6,1,11),(2938,2936,13943,NULL,NULL,NULL,21,664,101,'DSF-2010-MB P/N: 2653-271-06','506401','2022-02-17','New one.\r\nMaking circuit 485.\r\nSimulation U704 (LT1785AI) - ok.\r\nReplacing the U704 (new one).\r\nNo problem founded.\r\n\r\nTested with PC, - ok\r\nPassword 123456','116','','',10,6,1,11),(2939,2937,15000,NULL,NULL,NULL,21,555,19,'10007','364','2022-02-17','364:\r\nLoost phase alarm.\r\nOver voltage alarm.\r\n\r\n220uF 35V -3,\r\n10uF 50V -8.\r\nInrerupted pin tranfformer (cool soldering).\r\n1164: Loost phase alarm. PC817, TCA785  -3.\r\n2199: No problem. Capacitors.\r\n2937: No problem. 220uF -3.','141','','',15,4,1,11),(2940,2938,1950,NULL,NULL,NULL,21,272,6,'TSXCUSB232','PV01RL00DOM1524','2022-02-20','ADM211 PIN 23 (-4V instead -10V)','86, 6','','',10,6,1,11),(2941,2939,2000,NULL,NULL,NULL,21,469,76,'PAXS001U','29103','2022-02-21','751:\r\nCorrosion.\r\nIncorrect parameters\r\n2.5V ref  (0V on AD) => 6000\r\nAD - ok.\r\n1133:\r\nCorrosion.\r\nTested.\r\n1194: Replacing input terminal\r\n2939: jampers, connectors, corrosion.','40','','RedLion',6,2,1,11),(2942,2940,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','4023570','2022-02-21','1808: Corrosion.\r\nCPU + EEPROM.\r\nTest.\r\n2940: cleaning, testing.','103','','',3,1,1,11),(2943,2941,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S A430L250-09','4023897','2022-02-21','2941: cleaning, testing.','103','','',3,1,1,11),(2944,2942,13943,NULL,NULL,NULL,21,664,101,'DSF-2010-MB P/N: 2653-271-06','506442','2022-02-21','2942: \r\nDefaulted \r\n\r\nTested with PC, - ok\r\nPassword 123456','116','','',2,1,1,11),(2945,2943,12000,NULL,NULL,NULL,21,329,19,'RVS-D','7604 4650','2022-02-22','Flash on 10nF 3kV, 100uH.\r\n2200uF 25V, \r\n470uF 16V => 220uF 35V.','','','',5,2,1,11),(2946,2944,12000,NULL,NULL,NULL,21,118,18,'Kent-Taylor MF/E400372101SS5ER33','V-10384/19/1','2022-02-23','2118: Capacitors.\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\n2944: cleaning','','','',10,6,1,11),(2947,2945,1200,NULL,NULL,NULL,21,584,35,'N15-39.4.R.4.0.1.0.00.8-KJ27','22419','2022-02-28','22419 0-650 KG/H\r\nCapacitors.','','','',7,4,1,11),(2948,2946,8300,NULL,NULL,NULL,21,362,35,'A2000','OK4376920007','2022-03-01','PWM PS faulted.\r\nCapacitors.','','','',12,4,5,11),(2949,2947,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5','6080-96','2022-03-01','110VAC!\r\n\r\nCapacitors, \r\nRelay 12V -6.','53','','SOLCON\\MRP2000_D\\X2816.bin',20,6,1,11),(2950,2948,3220,NULL,NULL,NULL,21,551,67,'CE790R','A1F3-156C4-003','2022-03-02','Capacitors.','','','',10,5,2,11),(2951,2949,4500,NULL,NULL,NULL,21,116,98,'M400/2H','1240037113','2022-03-02','No problem','72','','',12,1,1,11),(2952,2950,1000,NULL,NULL,NULL,21,646,90,'SK920','1731','2022-03-03','PS faulted.\r\n6V 1.5A\r\nCapacitors.','','','',4,3,5,11),(2953,2951,6000,NULL,NULL,NULL,213,681,3,'SSM-243-VER4','041-001767','2022-03-03','L296P\r\nCapacitors','','','',20,5,1,11),(2954,2952,6000,NULL,NULL,NULL,213,681,3,'SSM-243-VER4','1830','2022-03-06','L296P\r\nCapacitors','','','',20,5,1,11),(2955,2953,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','5185-95','2022-03-06','Capacitors, \r\nRelay 12V -6.','53','','SOLCON\\MRP2000_D\\X2816.bin',20,6,1,11),(2956,2954,3600,NULL,NULL,NULL,21,500,6,'Gavish 085','','2022-03-07','816: Corrosion!!!\r\nAll terminals, resistors are replaced.\r\nFull test provided.\r\n2954: Corrosion!!!\r\nAll terminals, resistors are replaced.\r\nIncorrect polarity capacitors.','6','','',9,2,5,11),(2957,2955,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','0026460','2022-03-08','Neon lamp\r\n22M\r\nFull test.','8','','',7,4,1,11),(2958,2956,15000,NULL,NULL,NULL,21,555,19,'10007','364','2022-03-20','364:\r\nLoost phase alarm.\r\nOver voltage alarm.\r\n\r\n220uF 35V -3,\r\n10uF 50V -8.\r\nInrerupted pin tranfformer (cool soldering).\r\n1164: Loost phase alarm. PC817, TCA785  -3.\r\n2199: No problem. Capacitors.\r\n2937: No problem. 220uF -3.\r\n2956: Making circuit, corrosion, 1uF -5.','141','','',80,60,1,11),(2959,2957,2600,NULL,NULL,NULL,21,2,1,'SIPART N1SVPB776288001','1137170','2022-03-21','Cleaning.\r\nTest','103','','',12,3,9,11),(2960,2958,2000,NULL,NULL,NULL,21,469,76,'PAXS001U','M2396C','2022-03-21','1416:\r\nCorrosion.\r\nCPU.\r\nInstalled the new AO-module\r\nTested.\r\n1506:\r\nCorrosion. Programming AT24C04.\r\n2958: cleaning. Test','40','','RedLion',10,2,5,11),(2961,2959,5000,NULL,NULL,NULL,21,612,85,'TMS-M','PM08-70-172','2022-03-21','Regeneration of sensor.','77, 50','CMC','',24,24,1,11),(2962,2960,70000,NULL,NULL,NULL,138,497,39,'LKCO Power Sub-Rack And Electro-','6925Z RLW','2022-03-23','Because short tantalum capacitor the PS \r\nLAMBDA Vega 450 V406PCX (2*28V 5A, 2*6V 6A) are flashed.\r\nUsed 4 PS: 2 - 24V; 2 - 5V with regulation potentiometer.','','','',0,25,1,11),(2963,2961,9100,NULL,NULL,NULL,213,533,43,'140ACO03000','21075114974','2022-03-27','LT1013 SOIC to LT1013 DIP','','','',30,20,1,11),(2964,2962,3970,NULL,NULL,NULL,21,2,16,'1XP8001-1','1352','2022-03-28','Repair cable','84','','',10,5,1,11),(2965,2963,8200,NULL,NULL,NULL,21,519,18,'8732ESR2A1NAM4C1','0310682','2022-03-28','2757: PS: shorted Ultra Fust Diode on output voltage.\r\nBecause shorting was faulted PWM controller (OPA425R).\r\nReplacing the new one PS board.\r\nTesting.\r\nok\r\n2963: Device - ok. Absent wire up connector for coil 3 pins).','','','',10,2,2,11),(2966,2964,2200,NULL,NULL,NULL,21,555,52,'TVU 2.0','2964','2022-03-28','Was mistake on the base documentation (connection).\r\nMaking the full circuit.\r\nCapacitors, ISO124P.\r\nCalibration: 0mV/4mA; 60mV/20mA.','73','','',30,30,2,11),(2967,2965,2200,NULL,NULL,NULL,21,555,52,'TVU 2.0','2965','2022-03-29','Capacitors.','73','','',7,3,2,11),(2968,2966,2200,NULL,NULL,NULL,21,555,52,'TVU 2.0','2U3','2022-03-29','Capacitors.','73','','',7,3,2,11),(2969,2967,40000,NULL,NULL,NULL,213,701,143,'POCON 21','12-2256-1','2022-03-31','AC to DC converter 230VAC to 24VDC TMT15124 (repair)','','','',30,10,5,11),(2970,2968,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARBK-0770','2022-04-04','Cleaning.\r\nIncorrect cable sensor inserting. One wire up pin was out of place.','144','','',6,3,1,11),(2971,2969,2000,NULL,NULL,NULL,21,498,43,'TCI-2','608756','2022-04-04','Capacitors\r\nCalibration:\r\n0 grad = 4mA\r\n750 grad = 20 mA','9','','',4,3,5,11),(2972,2970,2000,NULL,NULL,NULL,21,498,43,'TCI-2','608732','2022-04-04','Capacitors\r\nCalibration:\r\n0 grad = 4mA\r\n750 grad = 20 mA','9','','',4,3,5,11),(2973,2971,4600,NULL,NULL,NULL,21,161,7,'AS-B838-032','0043603','2022-04-06','Configuration:\r\n\r\nOOOO-+OOOOOOOO-+OOOO OOOO-+OOOOOOOO-+OOOO\r\n\r\nch_32 IRFD9120\r\ntest ok','','','',9,5,1,11),(2974,2972,4600,NULL,NULL,NULL,21,161,7,'AS-B838-032','0050130','2022-04-06','Configuration:\r\n\r\nOOOO-+OOOOOOOO-+OOOO OOOO-+OOOOOOOO-+OOOO\r\n\r\ntest ok','','','',9,5,1,11),(2975,2973,4600,NULL,NULL,NULL,21,161,7,'AS-B838-032','0052036','2022-04-06','Corrosion on input terminal\r\n\r\nConfiguration:\r\n\r\nOOOO-+OOOOOOOO-+OOOO OOOO-+OOOOOOOO-+OOOO\r\n\r\ntest ok','','','',9,5,1,11),(2976,2974,16500,NULL,NULL,NULL,21,502,11,'3710 ACM','DD-980-4B075-01','2022-04-06','All capacitors.\r\nVFD: fg209c2/m2','','','Magnesium',12,6,1,11),(2977,2975,16500,NULL,NULL,NULL,21,502,11,'3710 ACM','16130','2022-04-07','47:\r\nInternal PS fault.\r\nCapacitors, terminal soldering.\r\nHV PS for VFD.\r\nCapacitors.\r\n\r\n293, 853:\r\nVFD FUTABA FG209M2, 10uF 16V -2, 10uF 63V.\r\n2975: PS-capacitors','','','Magnesium',12,5,1,11),(2978,2976,22500,NULL,NULL,NULL,213,535,3,'LZS1500-3/FOX','0030010401','2022-04-13','LAMBDA PS\r\nOne from two coolers was bad.\r\nOverheat 5.1R => 5,6R, PCB and int. print to cooler\'s supply.\r\nFuse F15A, MOC8101 => CNY17 -4;\r\nLM239 (voltage switching)\r\nLM358 (control PS in module)\r\nCorrosion, screws. \r\nWas unqualified repair.\r\nFor inspection PWM 2*BUZ71 the PS must be loaded.','142','','',70,100,1,11),(2979,2977,2325,NULL,NULL,NULL,213,535,3,'P0922YU','WX06360156','2022-04-14','Bad soldering between base board and connectors board.\r\n1800uF 63V -2 => 3300uF 35V -2.\r\nPFC-voltage was 405Vdc.','','','',15,6,2,11),(2980,2978,4600,NULL,NULL,NULL,21,161,22,'AS-B827-032','0078439','2022-04-24','1097: Full test.\r\nNo problem founded.\r\nCleaning.\r\n1159: Test only.\r\n1673: Test only.\r\n2978: Cleaning and tesing.','41','','',15,8,2,11),(2981,2979,4600,NULL,NULL,NULL,21,161,22,'AS-B827-032','0077878','2022-04-24','1292: Test only.\r\n2979: Cleaning, testing.','41','','',15,5,2,11),(2982,2980,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','0025516','2022-04-25','Neon lamp -11\r\n22M -9\r\nFull test.','8','','',7,4,1,11),(2983,2981,16731,NULL,NULL,NULL,21,702,4,'MSEL controller IXP-4N520-WA-M-P','A60766843','2022-05-01','2981: cooler: MGT4024YB-O10','','','',20,10,1,11),(2984,2982,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','2982','2022-05-02','2982: Capacitors\r\n1M','','','',9,5,1,11),(2985,2983,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049925891','2022-05-02','Capacitors\r\n1M\r\nCNY17-3','','','',9,5,1,11),(2986,2984,8400,NULL,NULL,NULL,21,263,19,'DD10-48C0-0000','002003','2022-05-03','Cleaning.\r\nFull test.','','SCR pdf','',8,5,1,11),(2987,2985,6970,NULL,NULL,NULL,21,434,3,'EWS600-24','609Z33-0010W','2022-05-08','1154: Cooler\r\n2985: Cooler. Regulation','62','','',10,4,2,11),(2988,2986,2400,NULL,NULL,NULL,21,51,52,'OPTA1','253M 2045 0733 AS','2022-05-08','Was presented HV on input terminal\r\n0.1uF was flashed int, print. \r\n24C02 -  bad  (the new one programming)','7 548#','','VACON',10,4,1,11),(2989,2987,12000,NULL,NULL,NULL,21,118,18,'Kent-Taylor F/E','','2022-05-09','516:\r\nCapacitors.\r\n_________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\n\r\n2987: capacitors, 1N5366 (5W;39V), BYT13-1000, CPU.','','','',18,8,2,11),(2990,2988,5200,NULL,NULL,NULL,21,498,24,'AFM-16','630559','2022-05-10','Capacitors.\r\n4051 -ok','9','','bin',15,5,2,11),(2991,2989,5200,NULL,NULL,NULL,21,498,24,'AFM-16','630550','2022-05-10','2018: Corrosion!\r\nConnectors, capacitors, EPROM repair, sockets.\r\n2893: no problem founded.\r\n2989: bad contact J2','9','','bin',10,2,2,11),(2992,2990,870,NULL,NULL,NULL,21,518,11,'305','A159696','2022-05-15','Problem with input terminal.\r\n4mA => 0%\r\n20mA => 600%','','','',4,2,1,11),(2993,2991,870,NULL,NULL,NULL,21,518,11,'305','A170655','2022-05-16','Problem with input terminal.\r\nSwitch -4.\r\nCorrosion.\r\n4mA => 0%\r\n20mA => 200%','','','',4,2,1,11),(2994,2992,2000,NULL,NULL,NULL,21,469,76,'PAXS001U','27AUG20','2022-05-16','Test\r\nNo problem','40','','RedLion',10,2,5,11),(2995,2993,2000,NULL,NULL,NULL,21,469,76,'PAXS001U','29103','2022-05-16','1162: Corrosion.\r\nRepair input terminal.\r\nTested.\r\n2024: Input terminal, jampers.\r\n2210: cleaning. Installing the I/O board.\r\n2993: Input terminal, jampers.','40','','RedLion',8,5,5,11),(2996,2994,2200,NULL,NULL,NULL,21,268,35,'INT69/69T','2994','2022-05-17','100uF 50V  -2,\r\nRelay 24V -3.','','','',5,2,2,11),(2997,2995,12000,NULL,NULL,NULL,21,189,35,'46199001','46193001','2022-05-18','2995: Capacitors.\r\nTerminal, cleaning.','532#','','bin',10,2,5,11),(2998,2996,80450,NULL,NULL,NULL,21,390,98,'1700R12ABAEZZZ','3168056','2022-05-24','Alarm A100\r\nTested PS and 1uF 450V -ok.\r\nNo problem founded.','137','','',10,3,1,11),(2999,2997,260,NULL,NULL,NULL,21,227,8,'788-364','2997','2022-06-01','Cleaning','55','','',2,1,5,11),(3000,2998,650,NULL,NULL,NULL,213,703,35,'UCSA-801','0HV50W75435-38K-04ZU-A01','2022-06-01','Cleaning\r\nThe new thermal conducting paste replacing.','','','',5,4,1,11),(3001,2999,1200,NULL,NULL,NULL,21,505,7,'859-354','2999','2022-06-02','Making the SSD relay DC output.\r\n\r\n542-5959\r\n907-4867\r\n827-4029\r\n896-2436','55','','',7,4,5,11),(3002,3000,1200,NULL,NULL,NULL,21,505,7,'859-354','3000','2022-06-06','Making the SSD relay DC output.\r\n\r\n542-5959\r\n907-4867\r\n827-4029\r\n896-2436','55','','',7,4,5,11),(3003,3001,1200,NULL,NULL,NULL,21,505,7,'859-354','3001','2022-06-06','Making the SSD relay DC output.\r\n\r\n542-5959\r\n907-4867\r\n827-4029\r\n896-2436','55','','',7,4,5,11),(3004,3002,1200,NULL,NULL,NULL,21,505,7,'859-354','3002','2022-06-06','Making the SSD relay DC output.\r\n\r\n542-5959\r\n907-4867\r\n827-4029\r\n896-2436','55','','',7,4,5,11),(3005,3003,1200,NULL,NULL,NULL,21,505,7,'859-354','3003','2022-06-06','Making the SSD relay DC output.\r\n\r\n542-5959\r\n907-4867\r\n827-4029\r\n896-2436','55','','',7,4,5,11),(3006,3004,31000,NULL,NULL,NULL,21,2,13,'MIDIMASTER 6SE3121-7DG40','XAF362MD033D','2022-06-07','305: Internal Power Supply fault.\r\nAll capacitors changed.\r\nStart resistor (330k+330k) 680k.\r\nRepair PCB after heating.\r\nPre driver PCB: 1k (2kII2k) for soft start.\r\nReplacing power board (becouse int. stert resistors).\r\n7&8 pins for start.\r\nTested.\r\n2151: Making the new one.\r\n3004: F003 - overload.\r\nReplaced the Pre-DRV board, PS - ok. Problem was in CPU board.\r\nAfter default drive ok.','20','','',20,10,2,11),(3007,3005,1200,NULL,NULL,NULL,21,505,7,'859-354','3005','2022-06-08','Making the SSD relay DC output.\r\n\r\n542-5959\r\n907-4867\r\n827-4029\r\n896-2436','55','','',7,4,5,11),(3008,3006,1200,NULL,NULL,NULL,21,505,7,'859-354','3006','2022-06-08','Making the SSD relay DC output.\r\n\r\n542-5959\r\n907-4867\r\n827-4029\r\n896-2436','55','','',7,4,5,11),(3009,3007,1200,NULL,NULL,NULL,21,505,7,'859-354','3007','2022-06-08','Making the SSD relay DC output.\r\n\r\n542-5959\r\n907-4867\r\n827-4029\r\n896-2436','55','','',7,4,5,11),(3010,3008,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','0025443','2022-06-12','1493: Neon lamp -10.\r\nFull test.\r\n3008: 22M -15.','8','','',7,4,1,11),(3011,3009,19649,NULL,NULL,NULL,21,396,87,'Nano-L SMJ-A2/2 7,2V','9994434397','2022-06-12','Rotary switches  replacing.\r\n\r\n0/574/143/730','','','',5,2,1,11),(3012,3010,19649,NULL,NULL,NULL,21,396,87,'Nano-L-A2','9997917097','2022-06-12','Repair the wires, making the new connection to rotary switches.\r\n\r\n0/574/143/730','','','',5,2,1,11),(3013,3011,4200,NULL,NULL,NULL,21,707,12,'ika labortechnik rw16','3011','2022-06-13','Input filter flashed.\r\n\r\n2 toroid filters,\r\n2n2 250VAC, T4A => T6.3A','','','',15,8,1,11),(3014,3012,1600,NULL,NULL,NULL,21,486,54,'SD16-408-0000','21001533201-006','2022-06-14','Total corrosion \r\nInstalled the internal resistor 250R\r\n\r\nLim: \r\n4mA => 0\r\n20mA => 600','','pdf','',8,3,2,11),(3015,3013,14100,NULL,NULL,NULL,21,531,98,'AIT-63065 3081/PH0RP','J95-49813','2022-06-16','Replacing CPU\r\n\r\nCalibration','35','','',10,5,1,11),(3016,3014,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','5051-95','2022-06-26','Capacitors, \r\nRelay 12V -6.','53','','SOLCON\\MRP2000_D\\X2816.bin',20,8,1,11),(3017,3015,6970,NULL,NULL,NULL,21,434,3,'EWS600-24','CQT-150K13-0082-P101','2022-06-27','Cooler, cleaning, capacitors.','62','','',10,4,2,11),(3018,3016,4300,NULL,NULL,NULL,21,498,145,'WVT-3','619700','2022-06-29','U2: LT1013\r\nCapacitors','143','','',15,4,2,11),(3019,3017,4300,NULL,NULL,NULL,21,498,145,'WVT-3','821674','2022-06-29','Transformer 230V/2*20 (2*18)','143','','',15,4,2,11),(3020,3018,31000,NULL,NULL,NULL,21,2,13,'MIDIMASTER 6SE3121-7DG40','XAF363MD040D-2','2022-07-03','1348: VFD 11kW\r\nInternal Power Supply fault.\r\nAll capacitors changed.\r\nRepair PCB after heating.\r\nPre driver PCB: 1k for soft start and 1k x4 for HCPL7800.\r\nHCPL7800  -3,\r\nRepair the power board.\r\nCooler 24V. Screws.\r\nRepair the plug X1 of CPU board.\r\n\r\n3018: Pre-DRV board - full refurbished + TL084.\r\nTested: 7&8 pins for start. 3&4 pins are current input 4...20mA\r\nNumeration from right side.','20','','',20,9,2,11),(3021,3019,5570,NULL,NULL,NULL,21,531,98,'2081','L95-52751','2022-07-04','2141: Cleaning.\r\n3019: Tested.\r\nFor testing see manual of 2081 in the folder.\r\n(must be resistor 110R, 2 DIPSW in the down place (marking C).','book','','',10,6,1,11),(3022,3020,12000,NULL,NULL,NULL,21,329,19,'RVS-D 210A','1159-2A','2022-07-04','Problem with current measurement.\r\n210A\r\nCapacitors.\r\nSaving data from EEPROM, EPROM.\r\nReplacing the EEPROM X28C64 to AT28C64B\r\nCleaning.\r\nReturning DIP_SW7 in the OFF state.\r\nReplaced from my box the CPU\r\nPassable current regulation with trimmer.','156','','',25,14,5,11),(3023,3021,80450,NULL,NULL,NULL,21,390,98,'1700R12ABZEZZZ','141234444','2022-07-07','1uF 450V\r\nU14: HCPL-4506\r\nFull 485 sensor simulation.','137','','',40,20,1,11),(3024,3022,6700,NULL,NULL,NULL,21,116,98,'pH 2100/2','54268/1533223/MT','2022-07-07','2880:\r\nCleaning.\r\nCalibration (1100).\r\n3022: \r\nCleaning.\r\nCalibration (1100).','114','','',15,5,1,11),(3025,3023,6700,NULL,NULL,NULL,21,116,98,'pH 2100/2','','2022-07-07','2878: Cleaning.\r\nReplacing the front case panel.\r\nCalibration (1100).\r\n3023: Cleaning.\r\nCalibration (1100).','114','','',15,5,1,11),(3026,3024,80450,NULL,NULL,NULL,21,390,98,'1700R11ABAEZZZ','3014055','2022-07-11','1uF 450V\r\n_________________________________________________\r\nThere was an incomrehensible problem in the PS. \r\nLDO LT1086 did not work.\r\nAt the input of the stabilizer was 46V instead 18V.\r\nThe problem is gone.\r\nLCD stay unclear without repair.\r\n____________________________________________________ \r\nU14: HCPL-4506\r\nFull 485 sensor simulation.','137','','',40,20,1,11),(3027,3025,3570,NULL,NULL,NULL,21,434,3,'EWS300-24','736Z31-0096W','2022-07-12','681: Flash on PCB:\r\nCleaning.\r\nFull test.\r\n\r\n2408: total corrosion.\r\nD201: MUR2100E, Z206: 15V, connectors, fuse holder.\r\nCapacitors.\r\nTest\r\n2556: Fuse socket, test\r\n3025: FUN','62','','',10,3,2,11),(3028,3026,2492,NULL,NULL,NULL,21,396,33,'Ladegerat','31033','2022-07-12','1500: Cable repair.\r\n3026: Transformer PIN bad soldering,\r\nCleaning.','','','',4,2,5,11),(3029,3027,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARAA-0780','2022-07-12','Cleaning.','144','','',6,3,1,11),(3030,3028,2000,NULL,NULL,NULL,21,498,43,'TCI-2','727230','2022-07-14','Capacitors\r\nCalibration:\r\n0 grad = 4mA\r\n750 grad = 20 mA','9','','',4,3,5,11),(3031,3029,6400,NULL,NULL,NULL,21,263,19,'PC11-F30B-0000','00006013','2022-07-14','Corrosion on CPU and POWER PCB.\r\nTesting: phase control/ 4-20mA','','SCR pdf','',15,5,1,11),(3032,3030,6970,NULL,NULL,NULL,21,434,3,'EWS600-24','CQT-811K30-0045-P287','2022-07-17','Good condition, no problem.\r\n3.3uF 250V => 3.9uF 400V\r\n330uF.\r\nVoltage regulation: from 23V to 26V.','62','','',5,3,2,11),(3033,3031,11750,NULL,NULL,NULL,21,540,32,'11123','96','2022-07-21','Input number 243.\r\nProgramming CPU,SRAM.\r\nTest AI PT-100.\r\n_____________________________\r\n646: Lost data M48T08, CPU.\r\nProgramming.\r\n4051 -2 (Bad).\r\n_____________________________\r\n3031: All capacitors changed.\r\n4051 -2, LM324 -1, AD580.\r\nResetting the data.','','','bin',20,12,1,11),(3034,3032,6970,NULL,NULL,NULL,21,434,3,'EWS600-24','CQT-811K30-0025-P287','2022-07-24','836: New.\r\nRegulation +26V\r\n3032: Regulation +26V (from +17V)','62','','',2,1,1,11),(3035,3033,3200,NULL,NULL,NULL,21,498,43,'DCI-2','639379','2022-07-24','1297: Capacitors.\r\n\r\nCalibrated:\r\n0V => 4mA\r\n6V => 20mA\r\n\r\n3033: cleaning,\r\nCalibrated:\r\n0V => 4mA\r\n6V => 20mA','9','','',5,3,1,11),(3037,3035,11750,NULL,NULL,NULL,21,540,32,'11123','112','2022-07-27','319: For menu press and hold MODE key\r\nand apply 230VAC.\r\n\r\nPrg. ver. 5.02 - saved.\r\nAll capacitors replaced.\r\n_____________________________________________\r\nInput number 241.\r\nBecause jamp on power, calculation of PT-100 stay bad.\r\nChanged 4051 -2 (was bad), M48T08, CPU.\r\n_____________________________________________\r\n747: 4051  -2, AD580, LM324, programming CPU.\r\n3035: capacitors, 4051 -1, resetting the CPU \r\n(during pressing the MODE bar applying the power)','','','bin',20,9,1,11),(3036,3034,18000,NULL,NULL,NULL,213,367,146,'INVERTRONIC compact G220E230/6,5','6284141','2022-07-26','Flashed the capacitors 30uF 700V (NP)\r\nUsed the new one: RS-number 174-4501 30uF 800V (NP)\r\nNeed to test the polarity of wires to relay 2 pin\'s because \r\nbroken connector.\r\nWithout test!','','','',50,10,1,11),(3038,3036,11750,NULL,NULL,NULL,21,540,32,'11123','111','2022-07-28','262: For menu press and hold MODE key\r\nand apply 230VAC.\r\n\r\nPrg. ver. 5.02 - saved.\r\nEPROM 5.02 = 5.03\r\nIC9: LM324 - short PIN 3,\r\nM48T08.\r\nAll capacitors replaced.\r\nFuse T1A => F2A.\r\n3036: capacitors, \r\nT315 => T500 (was shorting -15V during the repair)\r\nResetting CPU (pressing the MODE during applying power).','','','bin',20,8,1,11),(3039,3037,6000,NULL,NULL,NULL,21,51,13,'CN45CXL4G5N1','10071731','2022-07-31','1212: Total corrosion.\r\nIGBT, pre-DRV board, connectors, FAN24V.\r\nEEPROM programming.\r\n1427: Corrosion.\r\nFaulted \"External fault 41\" bicause incorrect parameters entered.\r\nFAN 24V.\r\nTested.\r\n1697: sometimes the current drops to zero.\r\nCleaning. Replacing connectors.\r\n1864: Total corrosion.\r\n2421: Flash on PRE-DRV. Connectors.\r\n2641: test only\r\n2037: CPU replaced. \r\nCleaning.','7 548#','','bin',15,9,2,11),(3040,3038,12800,NULL,NULL,NULL,21,367,33,'Gleichrichter D400 G108/30 BWru-','1227406','2022-08-02','Capacitors.\r\nIGBT 1200V 50A (75A). Print.\r\nPre-drv board: 100nF  -7 (OH of plastic).\r\nTested (3*400VAC)\r\nOutput 123VDC.','145','','',40,15,1,11),(3041,3039,12800,NULL,NULL,NULL,21,367,33,'Gleichrichter D400 G116/40 BWru-','1160632','2022-08-03','All capacitors.\r\nPre-drv board: 100nF  -3, 470nF -2 (OH of plastic).\r\nTested (3*400VAC)\r\nOutput voltage regulation from 110.3V to 123VDC.','145','','',30,10,1,11),(3042,3040,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','0026471','2022-08-07','1493: Neon lamp -8.\r\nFull test.\r\n3008: 22M -10.','8','','',7,4,1,11),(3043,3041,4600,NULL,NULL,NULL,21,161,7,'AS-B838-032','31022034833','2022-08-07','Corrosion on input terminal\r\n\r\nConfiguration:\r\n\r\nOOOO-+OOOOOOOO-+OOOO OOOO-+OOOOOOOO-+OOOO\r\n\r\ntest ok','','','',9,5,1,11),(3044,3042,8500,NULL,NULL,NULL,21,51,13,'CN11CXL4G5I1','4216111','2022-08-09','181: Repair of the internal fan (+24V).\r\nCleaning sockets and connectors.\r\nTested.\r\n_________________________________________________\r\n489:\r\nAT28C64+ programming\r\n_________________________________________________\r\n2492: FAN\'s,Pre-DRV: 100R -2, cleaning.\r\nCPU faulted (contactor do not running) Replacing another CPU 11kW.\r\n2707: cleaning. Changing wire-up connector to IDC-flat\r\nChanging plugs on BRAKE IGBT module (making the new).\r\n3042: corrosion! Changing all connectors.','7 548#','','Magnesium',25,10,2,11),(3045,3043,8000,NULL,NULL,NULL,21,51,13,'CN2.2CXL4G5l1','3413733','2022-08-09','641: Corrosion!\r\nCPU: EEPROM, connectors.\r\nOperation panel: case, D-type, flat cable.\r\nPS, pre-driver: cleaning.\r\nCoolers.\r\n2694: Corrosion! Operation panel: case, D-type, flat cable.\r\nPS, pre-driver: cleaning. FAN\'s.\r\n3043: Corrosion! Operation panel: case, D-type, PS, \r\npre-driver: cleaning.','7 548#','book','bin',15,8,1,11),(3046,3044,870,NULL,NULL,NULL,21,518,11,'201','A182197','2022-08-10','Cleaning\r\n\r\n4mA => 0\r\n20mA => 900','pdf','','bin',6,4,2,11),(3047,3045,9650,NULL,NULL,NULL,213,2,48,'PS2494-Y REV C1, D2','2000884','2022-08-14','240 OHm 3W. \r\n\r\nTested: 400VAC to 368VDC with load (3 heater serial).\r\n5V analog input, enable.','104','pdf','',15,5,1,11),(3048,3046,9650,NULL,NULL,NULL,213,2,48,'PS2494-Y REV C1, D2','2000749','2022-08-15','240 OHm 3W. \r\n\r\nTested: 400VAC to 363VDC with load (3 heater serial).\r\n5V analog input, enable.','104','pdf','',15,5,1,11),(3049,3047,9650,NULL,NULL,NULL,213,2,48,'PS2494-Y REV C1, D2','2001512','2022-08-15','240 OHm 3W. \r\n\r\nVS-HFA15TB60-M3  -2  => RURP1560-F085  -2.\r\n\r\nTested: 400VAC to 363VDC with load (3 heater serial).\r\n5V analog input, enable.','104','pdf','',15,5,1,11),(3050,3048,9650,NULL,NULL,NULL,213,2,48,'PS2494-Y REV C1, D2','2020078','2022-08-16','240 OHm 3W. \r\n\r\n1uF 275VAC  -3.\r\n\r\nTested: 400VAC to 363VDC with load (3 heater serial).\r\n5V analog input, enable.','104','pdf','',15,5,1,11),(3051,3049,9650,NULL,NULL,NULL,213,2,48,'PS2494-YE-28-4','2017008','2022-08-16','1931: Broken LAN socket.\r\n240 OHm 3W.\r\n\r\nHV Capacitors instead 7uF*2 => 14uF \r\n\r\nTested: 400VAC to 368VDC with load (3 heater serial).\r\n5V analog input, enable.\r\n\r\n3049: 240 OHm 3W, SCR-transformer, \r\nSCR BTW69-1200,SMD 6R2, SMD Diode 5D=> 4148','104','pdf','',20,15,1,11),(3052,3050,9650,NULL,NULL,NULL,213,2,48,'PS2494-Y REV. C1, D2','2000864','2022-08-21','240 OHm 3W.\r\n\r\nNew set making \r\n\r\nTested: 400VAC','104','pdf','',20,15,1,11),(3053,3051,30000,NULL,NULL,NULL,21,708,9,'XD303V7.1','SD60-MCC','2022-08-22','Making circuit.\r\n\r\nTesting - ok.','146','','bin',30,10,1,11),(3054,3052,1500,NULL,NULL,NULL,21,709,8,'DSR2','3052','2022-08-23','Relay 12V\r\nKSP44 -2 (print OH).','','','',6,4,5,11),(3055,3053,900,NULL,NULL,NULL,21,504,128,'WAS5 CCC 20LP 8581160000','P7140','2022-08-23','10R','','','',6,4,1,11),(3056,3054,3200,NULL,NULL,NULL,21,498,43,'RTI-2','608817','2022-08-24','2103: Capacitors, trim 2k\r\nTest: \r\n0C => 100R => 4mA\r\n400C => 247.09R => 20mA\r\n\r\n3054: Terminal\r\nCalibration','9','','',6,4,2,11),(3057,3055,16000,NULL,NULL,NULL,21,53,6,'FXW Hand Held Communicator','TK7C3377','2022-08-29','99: Making PWM power supply 7.5V 500mA\r\ninstead bad original PS 7VDC 150mA.\r\nMaking back cover for Ni-Cd battery. \r\nBackup 27C512.\r\n1179: Repair the keyboard,\r\nreplacing the new set Ni 1.2V*5 bat.\r\n3055: Repair the keyboard,\r\nreplacing the new set Ni 1.2V*5 bat.\r\nMaking PWM power supply 6V.','','','fuji',12,8,1,11),(3058,3056,870,NULL,NULL,NULL,21,518,11,'201','A150107','2022-08-29','527:\r\nCorrosion.\r\nCleaning.\r\n755: tactile switch -4.\r\n3056: corrosion, cleaning, connectors.\r\n4mA => 0\r\n20mA => 800','','','',7,4,2,11),(3059,3057,870,NULL,NULL,NULL,21,518,11,'201','A140157','2022-08-29','Corrosion, cleaning, connectors.\r\n4mA => 0\r\n20mA => 100','','','',7,4,2,11),(3060,3058,2400,NULL,NULL,NULL,21,689,122,'AM204-MDG','2997','2022-08-30','Flat cable was outside from the socket.','','','',2,1,5,11),(3061,3059,2400,NULL,NULL,NULL,21,689,122,'AM204-MDG','2994','2022-08-30','No problem.','','','',2,1,5,11),(3062,3060,2492,NULL,NULL,NULL,21,396,33,'Ladegerat 2.250.1005','20997','2022-08-30','1501: Cable repair.\r\nCleaning.\r\n1645: capacitors,\r\nCable repair.\r\n2493: int. print on primary of the transformer.\r\n3060: Int. print. \r\nCleaning.','','','',4,2,5,11),(3063,3061,2492,NULL,NULL,NULL,21,396,33,'Ladegerat 2.250.1005','30201','2022-08-30','1646: Capacitors\r\n2488: Interrupted inductor and wire\r\n3061: cleaning','','','',4,2,5,11),(3064,3062,9600,NULL,NULL,NULL,21,710,148,'BP300S','50410098','2022-09-01','Cleaning and soldering \r\nthe print between controller and LCD','','','',15,6,1,11),(3065,3063,2800,NULL,NULL,NULL,21,9,35,'EKE 347','080G5000','2022-09-08','Cleaning the flat of LCD','','','',2,1,1,11),(3066,3064,8500,NULL,NULL,NULL,21,118,1,'SPCD 2055-AB','227535','2022-09-08','Lost data. \r\nProgramming AT28HC64B\r\nRev. 124C','','','magnesium',8,4,1,11),(3067,3065,8500,NULL,NULL,NULL,21,118,1,'SPCD 3D53-AB','275153','2022-09-11','Lost data of EPROM and EEPROM. \r\nProgramming AT28HC64B and AT27C512R (used DIP instead PLCC)\r\nRev. 107C','','','magnesium',15,8,1,11),(3068,3066,8500,NULL,NULL,NULL,21,118,1,'SPCD 3D53-AB','621355','2022-09-11','Braked socket PLCC.\r\nSaving the dump\r\nRev. 107G','','','magnesium',5,2,1,11),(3069,3067,12000,NULL,NULL,NULL,21,505,98,'PMP21','3067','2022-09-13','Making the new project of pressure meter with\r\nrechargeable function and battery indicators. \r\n\r\nUsed the Nokeval 4-20mA and sensor PMP21\r\n\r\nInput: 4=>20\r\nIndication: 0=>10','147','','',50,30,5,11),(3070,3068,2000,NULL,NULL,NULL,21,469,76,'PAXS0010','35041','2022-09-28','Incorrect parameters.\r\nReplacing EEPROM','40','','RedLion',8,2,5,11),(3071,3069,2492,NULL,NULL,NULL,21,396,33,'Ladegerat 2.250.1005','20996','2022-09-28','Capacitors.\r\nInt. soldering on transformer','','','',4,2,5,11),(3072,3070,12000,NULL,NULL,NULL,21,118,18,'Kent-Taylor F/E250372101005','V/34869/1/1','2022-09-29','_________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.','','','',18,8,2,11),(3073,3071,2000,NULL,NULL,NULL,21,569,110,'EVG 2x24W 24VAC','3071','2022-10-03','T1A => T3.15A,\r\n10uF.','43','','',8,4,1,11),(3074,3072,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARJF-0744','2022-10-03','Problem of the 4-20mA.\r\nWas changed isolated board (IO)\r\nTest with sensor (4mA). \r\nCleaning.','144','','',12,8,1,11),(3075,3073,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARZC-0339','2022-10-03','Problem of the 4-20mA.\r\nWas changed isolated board (IO)\r\nTest with sensor (4mA). \r\nCleaning.','144','','',12,4,1,11),(3076,3074,2000,NULL,NULL,NULL,21,569,110,'EVG 2x24W 24VAC','3074','2022-10-18','T1A => T4A, 1000uF 35V => 1000uF 50V -2, IRF640 -2.','43','','',8,4,1,11),(3077,3075,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','5182-95','2022-10-18','Capacitors, \r\nRelay 12V -6.','53','','SOLCON\\MRP2000_D\\X2816.bin',20,8,1,11),(3078,3076,80450,NULL,NULL,NULL,21,390,98,'FIT-99767 1700R12ABZEZZZ','3117482','2022-10-19','1918: 1uF 450V\r\nD2: 8S => SMD SOT23: ZD12V\r\nLT1017 (?)\r\n3076: U14 HCPL-4506 -bad.\r\nU13: also been replaced','137','','',10,5,1,11),(3079,3077,4300,NULL,NULL,NULL,21,552,32,'multi9 RCI 2181','9530','2022-10-19','Label: K12\r\n\r\nCapacitors.\r\nBC547 -2 => BC639 -2.\r\nSYR001, LM324\r\n\r\nTest','30 155','','',10,5,1,11),(3080,3078,1600,NULL,NULL,NULL,21,486,54,'SD16-408-1400','210011372801-003','2022-10-19','696: Electrical Laboratory.\r\n\r\nSW\'s, input terminal - corrosion, 1000uF 10V => 1000uF 16V.\r\nInput scaling:\r\nlimit: (4mA~0; 20mA~4000).\r\n1542: Incorrect data because leakage corrosion.\r\nCleaning.\r\n3078: The external wire/internal resistor installed.','','pdf','',7,4,2,11),(3081,3079,13943,NULL,NULL,NULL,21,664,101,'DSF-2010-MB P/N: 2653-271-06','502480','2022-10-20','Defaulted 4mA','116','','',2,1,1,11),(3082,3080,2600,NULL,NULL,NULL,21,2,1,'SIPART A5E50080933-AA','YSUN7119241  C73451A430L350  RS: 016-031','2022-10-20','Cleaning.\r\nTest','103','','',12,3,9,11),(3083,3081,5655,NULL,NULL,NULL,21,530,22,'140 DDI 353 00','31994103599','2022-10-20','Cleaning.','','','',10,3,5,11),(3084,3082,80450,NULL,NULL,NULL,21,390,98,'FIT-99767 1700R12ABZEZZZ','3117383','2022-10-23','1968: 1uF 450V\r\n3082: C27 22uF 50V, \r\nU13, U14: HCPL-4506 (without testing before repair)\r\nTest sensor I/O (485).','137','','',20,10,1,11),(3085,3083,1800,NULL,NULL,NULL,21,584,35,'N15 0-300MG/H','99667','2022-10-23','568: Internal power supply fault (TOP2** circuit)\r\nCapacitors.\r\n\r\n3083: 10R, corrosion.\r\nCurrent in: 14&15 \r\n4-20mA ~ 0-300','','','',6,3,1,11),(3086,3084,400,NULL,NULL,NULL,21,711,54,'SW-100','9.445','2022-10-30','Making the new cable and circuit.','148','','',60,40,1,11),(3087,3085,2000,NULL,NULL,NULL,21,619,108,'SKM 3S','2003','2022-10-30','2003: Making new circuit for constant current.\r\n3085: 10R -2.','108','','',10,7,1,11),(3088,3086,5600,NULL,NULL,NULL,21,505,52,'Ramzor_1','981','2022-10-30','981: Making the new project.\r\n3086: cleaning and testing.','59','','',4,2,1,11),(3089,3087,8500,NULL,NULL,NULL,21,118,1,'SPCD 3D53-AB','294779','2022-11-02','Lost data of EPROM and EEPROM. \r\nProgramming the new AT28HC64B rev 107G.\r\nThe chip AT27C512R rev D stay from old.\r\nRev. 107C','','','magnesium',15,8,1,11),(3090,3088,2200,NULL,NULL,NULL,21,607,98,'PMC71-5CQ82/101','RA1026010CC','2022-11-03','Corrosion\r\nThe new input module installed.','','','',5,2,1,11),(3091,3089,12000,NULL,NULL,NULL,21,118,18,'F/E000000000005ER3321111','V/32036/2/1','2022-11-06','683: \r\nCapacitors.\r\n_________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\n\r\n2105:\r\nCapacitors.\r\n3089: LCD + panel','','','',5,2,5,11),(3092,3090,1800,NULL,NULL,NULL,21,498,43,'ACT-2-I','1138126','2022-11-06','0-1A => 4-20mA\r\n\r\nCA3130 -bad\r\nUsed in DIP\r\n\r\nCalibrated','136','','',12,10,1,11),(3093,3091,3200,NULL,NULL,NULL,21,498,43,'RTI-2','1319112','2022-11-07','1883: Capacitors,\r\nTest: \r\n0C => 100R\r\n400C => 247.09R\r\n3091: \r\n0C => 100R\r\n100C => 138.5R','9','','',6,4,2,11),(3094,3092,2492,NULL,NULL,NULL,21,396,33,'Ladegerat','24155','2022-11-07','535: Soldering.\r\nCleaning.\r\n1793:Cleaning.1901: Cleaning\r\n2509: capacitors, cleaning\r\n3092: Cleaning','','','',5,3,2,11),(3095,3093,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','563743','2022-11-07','1763: Corrosion,\r\nSW\'s, CPU+EPROM,\r\nProgramming.\r\nTest.\r\n3093: Cleaning, programming.','103','','',12,6,2,11),(3096,3094,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','4297586','2022-11-07','3094: Cleaning.','103','','',12,6,2,11),(3097,3095,35800,NULL,NULL,NULL,213,712,109,'SVERKER 650','4061918','2022-11-07','Input filter 10A short.\r\nChanged to 3A.','','','',8,4,1,11),(3098,3096,1600,NULL,NULL,NULL,21,486,54,'SD16-408-0000','21001616501-012','2022-11-14','Cleaning.\r\n\r\n4-20 mA\r\n1-17','','pdf','',7,4,2,11),(3099,3097,2400,NULL,NULL,NULL,21,51,52,'OPTA1','253M 1825 0656 AS','2022-11-14','24C02 -  bad  (the new one programming)','7 548#','','VACON',10,4,1,11),(3100,3098,3755,NULL,NULL,NULL,21,533,17,'MODICON TM221CE40R','IB203201855','2022-11-16','Q0 -faulted. Changed the relay (was ok).','','','',10,3,1,11),(3101,3099,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','5177-95','2022-11-16','Capacitors, \r\nRelay 12V -6.\r\nSocket CPU.','53','','SOLCON\\MRP2000_D\\X2816.bin',20,8,1,11),(3102,3100,10500,NULL,NULL,NULL,213,533,95,'140DAO84000','WX172001135','2022-11-21','2930: K275 SMD varistor short.\r\nReplaced to S14K275\r\n3100: ch_12 K275 SMD varistor short.\r\nReplaced to S14K275','','','',5,1,1,11),(3103,3101,12000,NULL,NULL,NULL,21,118,18,'F/E201552101001ER1301111','V/87165/1/1','2022-11-22','Capacitors.\r\n_________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\n\r\nPLCC socket','','','',10,3,5,11),(3104,3102,9650,NULL,NULL,NULL,213,2,48,'PS2494-YE REV. C1, D2','2022321','2022-11-22','Total repair (current circuit, power board, \r\nSCR-transformer changed to CD1396C 1:1 Murata).\r\nTest.','104','pdf','',40,25,1,11),(3105,3103,9650,NULL,NULL,NULL,213,2,48,'PS2494-YE REV. C1, D2','2000733','2022-11-23','Corrosion.\r\nIGBT, 240R\r\nTest.','104','pdf','',15,5,1,11),(3106,3104,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240','6242500','2022-11-24','FAN flow corrosion.\r\nInt. print.\r\n470k, 51k -2. \r\nUsed the new silicone paste for PFC and output voltage side PS\r\n\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',50,20,1,11),(3107,3105,3800,NULL,NULL,NULL,21,118,18,'E4','000132119/Y001','2022-11-28','748: Replacing CPU board.\r\nProgramming from original EEPROM.\r\n\r\n3105: \r\nFault: 33mA output instead 3.5mA (without sensor)\r\nPC410L => ACPL-M60L\r\nShort 0.22mF SMD\r\nMAX660 => ICL7660\r\n!!! Without sensor ERROR 2 - low voltage ref is OK','','','',20,8,1,11),(3108,3106,4200,NULL,NULL,NULL,213,367,48,'SLIMLINE RECTIFIYER 1500','5964864','2022-11-28','FAN flow corrosion.\r\nInterrupted wire in pre drive transformer.','132','','',20,5,1,11),(3109,3107,3000,NULL,NULL,NULL,213,661,35,'TRIAD T324','125452LEH','2022-11-29','Unit position: U53.\r\n22uF 400V, 47uF 100V  -2.','113','','',10,5,1,11),(3110,3108,1200,NULL,NULL,NULL,21,713,3,'EDCG120','9500023595','2022-11-29','T5A','','','',2,1,1,11),(3111,3109,1500,NULL,NULL,NULL,213,714,35,'RLE-1C','MON2','2022-12-04','Relay 12V -2,\r\nSMD RED LED -1.','','','',6,3,1,11),(3112,3110,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5','6079-96','2022-12-04','Capacitors, \r\nRelay 12V -6.','53','','SOLCON\\MRP2000_D\\X2816.bin',15,5,1,11),(3113,3111,9650,NULL,NULL,NULL,213,2,48,'PS2494-YE REV. C1, D2','2000878','2022-12-06','IGBT, 240R\r\nTest.','104','pdf','',15,5,1,11),(3114,3112,400,NULL,NULL,NULL,21,715,33,'DC18RC','0206018','2022-12-07','Control transformer faulted.\r\nReplacing to PS 12V and DC to DC\r\n12V to 24V.','149','','',40,20,5,11),(3115,3113,10000,NULL,NULL,NULL,21,489,65,'ICS-668','93','2022-12-08','93: Repair compensation sensor.\r\nTest.\r\n3113: 20mA - incorrect current output.\r\nuA741 -2 => OP07 -2.\r\n47uF -3.\r\nTrimer 100R.\r\nCalibration 4-20mA.','','','',15,4,2,11),(3116,3114,12000,NULL,NULL,NULL,21,507,76,'AD-4322A','C7916067','2022-12-08','60: Interrupted power plug.\r\nCapacitors.\r\nTest.\r\nBridge color wire load cell connection:\r\n1, 2 - green exc+ (+12V),\r\n3, 4 - black exc- (0V),\r\n5 - white sig+,\r\n6 - red sig -.\r\n\r\n2789: Complaint about accuracy of weight. \r\nWas replaced LC-input module from another unit, \r\nfew capacitors 33uF, and cleaning.\r\n\r\nStay problem with VFD-controller because artefacts on display (DP)\r\nChip: HD6433712B90H\r\n\r\n3114: +5V problem because bad contact \r\nbetween PCB PS and PCB CPU (connector).\r\n2200uF 35V -2 => 2200 50V -2.\r\nCleaning.','','PDF/AD-4322','BIN/Magnesium/ad4322/27512',10,3,1,11),(3117,3115,6700,NULL,NULL,NULL,21,116,98,'pH 2100/2H','54270/8333019/0832','2022-12-11','Cleaning.\r\nConfiguration (1200).\r\nCalibration (1100).','114','','',15,5,1,11),(3118,3116,6700,NULL,NULL,NULL,21,116,98,'pH 2100/2H','061562','2022-12-11','570: Corrosion!\r\n74HC4066 (ok)\r\nFull testing.\r\n2074: Corrosion!\r\nKeyboard cleaning, \r\ncalibration.\r\nPassword: 1100\r\n2194: Corrosion.\r\nProgramming temperature = +25C\r\nCONF/1200/ manual TC=25C\r\n2569: calibration\r\n3116: Corrosion!\r\nCalibration','114','','',15,6,1,11),(3119,3117,18000,NULL,NULL,NULL,21,51,13,'45CXL4G5N1','9435879','2022-12-13','439:\r\nFAN24V, Base contactor,\r\nStart resistor => 3 II 100R,\r\nBridge SKD 160/16,\r\nProgramming AT28C64B.\r\nCleaning.\r\nFull test.\r\n_______________________\r\n1214: Total. IGBT, BRIDGE, Pre-DRV\r\n____________________________________________\r\n514: Unqualified repair (not by laboratory)\r\nChanged 100R soft start,\r\ncleaing.\r\nTest.\r\n____________________________________________\r\n596: Alarm: Inverter fault.\r\nCleaing.\r\nTest.\r\n____________________________________________\r\n759: IGBT 150A -1 => 200A; \r\nModules IGBT istead 200A used 150A.\r\nPre driver: 100R -2 smd, connectors, 5R6.\r\nCPU: connectors.\r\nPower terminal changed (L2).\r\n____________________________________________\r\n845: New flat connection applyed X3.\r\n1511: Short SMD transistor of remote relay K4.\r\nRelay K3 was changed because bad contact between NC&COM\r\n_________________________________________________________\r\n3117: IGBT, Pre-DRV (5R6 -2), all plugs, Large FAN, Small FAN.','7 548#','','folder: PC00061A',30,16,1,11),(3120,3118,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D34-AA','311331','2022-12-14','Bad EEPROM\r\nThe new chip EEPROM programming 043E','','','magnesium',8,2,1,11),(3121,3119,2230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','277572','2022-12-14','Capacitors.\r\nRelays.','','','',7,3,2,11),(3122,3120,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240','5778861','2022-12-27','3120: FAN flow corrosion.\r\nInt. print.\r\nPFC: IC\r\n470k (was ok), SMD n-p-n for relay, T20A.\r\nUsed the new silicone paste for PFC and output voltage side PS\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',50,24,1,11),(3123,3121,5200,NULL,NULL,NULL,21,716,32,'CEF 1-12','020289','2022-12-27','Relay 24V, capacitors, soldering!','','','',5,2,5,11),(3124,3122,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240','5987636','2022-12-29','FAN flow corrosion.\r\nInt. print.\r\nPower side: ZD15V (was ok) -2.\r\n470k (was ok), SMD n-p-n for relay, T20A, 2k2, 4148, ZD15V\r\nUsed the new silicone paste for PFC and output voltage side PS\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',30,16,1,11),(3125,3123,1400,NULL,NULL,NULL,21,611,149,'6400-2716-01','X13650728060','2023-01-02','Part number BRD04873\r\n\r\nCleaning.\r\nTesting with magnetic stick.\r\n\r\nhttps://www.chillerparts.ae/product/brd04873-dual-binary-input-board/','','','',3,2,2,11),(3126,3124,1800,NULL,NULL,NULL,21,611,3,'X13110756-02','2183806527','2023-01-02','Part number BRD02102\r\n\r\nCleaning.\r\n\r\nhttps://www.chillerparts.ae/product/brd02102-power-supply-board/https://www.chillerparts.ae/product/brd02102-power-supply-board/','','','',3,2,2,11),(3127,3125,700,NULL,NULL,NULL,21,512,35,'ATR121-AD','M420-120B','2023-01-02','470uF 16V\r\nCleaning.\r\nTest: 4-20 => 0-10.0','36','pdf','',5,4,1,11),(3128,3126,8000,NULL,NULL,NULL,21,611,35,'UC800 X13651144-01 Rev G','1365114401041173000050','2023-01-02','Corrosion on IC (Ethernet controller):\r\nTERIDIAN 78Q2133\r\n\r\nThe chip was extracted, cleaning and soldering.','','','',10,4,1,11),(3129,3127,600,NULL,NULL,NULL,21,717,62,'ES21 34.15','00122952','2023-01-03','Diode SMD 1000V 1.5A -2\r\nMosfet SMD 7A 600V\r\nTest.','','','',8,4,1,11),(3130,3128,600,NULL,NULL,NULL,21,717,62,'ES21 34.15','001223128','2023-01-03','Diode SMD 1000V 1.5A -2\r\nMosfet SMD 7A 600V\r\nTest.','','','',8,4,1,11),(3131,3129,600,NULL,NULL,NULL,21,717,62,'ES21 34.15','001223129','2023-01-03','Diode SMD 1000V 1.5A -2\r\nMosfet SMD 7A 600V\r\nTest.','','','',8,4,1,11),(3132,3130,600,NULL,NULL,NULL,21,717,62,'ES21 34.15','001223130','2023-01-03','Diode SMD 1000V 1.5A -2\r\nMosfet SMD 7A 600V\r\nTest.','','','',8,4,1,11),(3133,3131,2000,NULL,NULL,NULL,21,469,76,'PAXS0010','M2396E','2023-01-03','Cleaning','40','','RedLion',8,2,5,11),(3134,3132,4200,NULL,NULL,NULL,213,367,48,'SLIMLINE RECTIFIYER 1500','6066482','2023-01-05','FAN flow corrosion.\r\nFlash under capacitor and print of DC BUS after PFC.\r\nMosfet 600V 20A, Diode 16V 15A fast, \r\nUC3854DW, SMD 0.01R to 0.02R II 0.02R.\r\nSMD 0.23uF to 0.15uF II 0.15uF.','132','','',30,15,1,11),(3135,3133,2100,NULL,NULL,NULL,21,530,3,'140CPS11420','330004800932901007','2023-01-08','Power side flashed.\r\nB9NK60Z -2, T4A, 0.2R II 0.2R -2 => 0.3R II 0.3R -3,\r\nLE Z240 (?) => 2*12V ZD.','','','',20,10,2,11),(3136,3134,1500,NULL,NULL,NULL,21,599,8,'MSR 020-I','911','2023-01-09','911: 100uF 50V -2,\r\nRelay 24V -2.\r\n1459: no problem founded.\r\nSoldering PCB, relay transistor.\r\n1509: no problem founded.\r\n3134: no problem founded.','','','',5,4,1,11),(3137,3135,2200,NULL,NULL,NULL,21,268,35,'INT69/69T','52A1442135','2023-01-09','100uF 50V  -2,\r\nRelay 24V -3.\r\nTest with PTC=100R','','','',5,2,2,11),(3138,3136,1600,NULL,NULL,NULL,21,486,54,'SD16-408-0000','21001588901-010','2023-01-09','Cleaning.\r\n\r\n4-20 mA\r\n0-150','','pdf','',7,2,2,11),(3139,3137,16200,NULL,NULL,NULL,21,530,35,'140CPU43412A','21081305100','2023-01-11','Cleaning and soldering the IC\'s\r\nTest - ok.','','','',10,4,2,11),(3140,3138,1200,NULL,NULL,NULL,213,569,108,'OPUS Mini LED 4303023-51-AC','3138','2023-01-11','SMD LED -4 (used MXA8-PW35-H001).\r\nT1.6A to T2A SMD','150','','',9,4,5,11),(3141,3139,1000,NULL,NULL,NULL,21,499,12,'4.728.0051','1728001002662','2023-01-15','3139: New PS installed.','22','','',4,2,1,11),(3142,3140,2492,NULL,NULL,NULL,21,396,33,'Ladegerat 2.250.1005','30201','2023-01-15','1646: Capacitors\r\n2488: Interrupted inductor and wire\r\n3061: cleaning\r\n3140: interrupted PIN inductor 150mH','','','',4,2,5,11),(3143,3141,1200,NULL,NULL,NULL,213,569,108,'OPUS Mini LED 4303023-51-AC','3141','2023-01-15','1uF 250V\r\nT1.6A to T2A SMD','150','','',9,4,5,11),(3144,3142,1200,NULL,NULL,NULL,213,569,108,'OPUS Mini LED 4303023-51-AC','3142','2023-01-15','SMD LED -1 (used MXA8-PW35-H001).\r\nT1.6A to T4A SMD, IRF640, LM358.','150','','',9,4,5,11),(3145,3143,1200,NULL,NULL,NULL,213,569,108,'OPUS Mini LED 4303023-51-AC','3143','2023-01-15','T1.6A to T4A.','150','','',9,4,5,11),(3146,3144,2600,NULL,NULL,NULL,21,2,1,'SIPART A430-L1','4091546','2023-01-16','Cleaning.\r\nTest.','103','','',12,3,9,11),(3147,3145,1200,NULL,NULL,NULL,213,569,108,'OPUS Mini LED 4303023-51-AC','3145','2023-01-16','T1.6A to T3.15A.\r\nMaking the transformer (5:1 75:17 500uH PRI).','150','','',20,10,5,11),(3148,3146,1200,NULL,NULL,NULL,213,718,5,'XW183K-5P0C2','305.0072.01.00','2023-01-17','Flash on PCB\r\nBTA08-600B => BTA16-600B','','','',5,3,2,11),(3149,3147,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARAJ-0598','2023-01-22','The problem is not in the unit, but in the external sensor box.\r\nTesting with with sensor.','144','','',12,4,1,11),(3150,3148,600,NULL,NULL,NULL,21,719,62,'Bridge','3148','2023-01-23','Bridge SMD 1000V 2A => 800V 2A.','','','',3,1,1,11),(3151,3149,4000,NULL,NULL,NULL,21,116,68,'USB-P25','SNR B911359027','2023-01-23','2571: Factory making print fault.\r\nShort between shield USB and RC-circuit (GND & earth)\r\n3149: cleaning.\r\nTest - ok.','','','',4,3,1,11),(3152,3150,6700,NULL,NULL,NULL,21,524,104,'P3U 8317275','ARJB-0013','2023-01-23','2091: Wireup plug repair.\r\nCleaning\r\n3150: Repair the terminal (4 pin).','151','','',20,10,1,11),(3153,3151,500,NULL,NULL,NULL,21,236,3,'ADP-230EB T','M0LW67P03H3','2023-01-29','Was applied 400VAC.\r\nVaristor 471, T6.3A.','','','',4,3,1,11),(3154,3152,4600,NULL,NULL,NULL,21,161,7,'AS-B838-032','0043642','2023-01-29','Configuration:\r\n\r\nOOOO-+00000000-+0000 OOOO-+00000000-+0000\r\n\r\n1767: ch_18 IRFD9120\r\n\r\n1860: ch_6  IRFD9120\r\n3152: ch_22 IRFD9120','','','',9,5,1,11),(3155,3153,3000,NULL,NULL,NULL,21,539,54,'Fluke 52','38620556WS','2023-01-30','2849: Making TC-terminal.\r\n3153: changed the battery. \r\nThe new TC plug installed.','','','',5,4,5,11),(3156,3154,2000,NULL,NULL,NULL,21,498,43,'DCI-2','714469','2023-01-30','1293: Capacitors.\r\n\r\nCalibrated:\r\n0V => 4mA\r\n6V => 20mA\r\n1879: test\r\n2064: changing configuration\r\n0...70V \r\nR30=240R\r\non-on-on-off-off-off\r\nCalibrated:\r\n0V => 4mA\r\n70V => 20mA\r\n3154: Test only.','9','','',5,3,1,11),(3157,3155,2492,NULL,NULL,NULL,21,396,33,'Ladegerat 2.250.1005','24201','2023-02-01','3155: Capacitors\r\nSoldering the transformer pin,\r\ncleaning.','','','',4,2,5,11),(3158,3156,15000,NULL,NULL,NULL,21,708,12,'XD303V7.1','3156','2023-02-05','Instead flashed motor used universal (12/24VDC)\r\nUsed DC BUS only for supplying the solenoid without PWM circuit.\r\n \r\nProvided works for install and regulation the solenoid. \r\n\r\nTesting - ok.','146','','bin',30,15,1,11),(3159,3157,15000,NULL,NULL,NULL,21,708,12,'XD303V7.1','3157','2023-02-05','Instead flashed motor used universal (12/24VDC)\r\nUsed DC BUS only for supplying the solenoid without PWM circuit.\r\n \r\nProvided works for install and regulation the solenoid. \r\n\r\nTesting - ok.','146','','bin',30,15,1,11),(3160,3158,2000,NULL,NULL,NULL,21,469,76,'PAXS0010','35041','2023-02-06','3158: Incorrect parameters.\r\nReplacing EEPROM\r\n3158: no problem founded.','40','','RedLion',8,2,5,11),(3161,3159,1000,NULL,NULL,NULL,21,646,90,'SK920','1731','2023-02-06','2950: PS faulted.\r\n6V 1.5A\r\nCapacitors.\r\n3159: 32768kHz, LED, 100uF -2, soldering.','','','',4,2,5,11),(3162,3160,9316,NULL,NULL,NULL,21,535,52,'P0400YE FBM4 REV-0P','1896506231','2023-02-08','2446: Capacitors.\r\nProblem with AT28C64B.\r\nEEPROM was changed, DUMP by 0P \r\n\r\nch_2: 49R9\r\n3160: no problem found. Transformers soldering.','32','','',5,2,1,11),(3163,3161,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192105','2023-02-08','2450: Capacitors.\r\n28C64 -bad. \r\nProgramming the new AT28CC64 with DUMP 0L\r\n3161: ch_1 - problem not found.','32','','',10,3,1,11),(3164,3162,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1894518981','2023-02-08','2338: Capacitors.\r\nNQ28C64 -bad. \r\nProgramming the new AT28CC64 with DUMP 0L\r\n3162: no problem found','32','','',10,3,1,11),(3165,3163,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896464415','2023-02-08','2838: Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P\r\n3163: no problem found.','32','','',10,3,1,11),(3166,3164,1200,NULL,NULL,NULL,21,648,35,'JDR10','29024','2023-02-08','Capacitors, Relay 12V.','','','',2,1,1,11),(3167,3165,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA FBM 01','1896505260','2023-02-08','1571: 330uF 35V -10,\r\n1000uF 10V => 1000uF 16V - all.\r\nSoldering.\r\n\r\nFull test - ok.\r\n2172: NO PROBLEM.\r\n2476: new AT28C64B installed with updated DUMP DA-0P\r\n3165: short (non constant) between EEPROM pin\'s.','32','','',8,1,1,11),(3168,3166,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0R FBM1','SF98025630','2023-02-08','2428: Capacitors.\r\n28C64 -bad. \r\nProgramming the new AT28CC64 with DUMP 0R\r\n3166: no problem found. Corrosion.','32','','',10,2,1,11),(3169,3167,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0S FBM1','SF00046356','2023-02-09','2369: Capacitors.\r\n28C64 -bad. \r\nProgramming the new AT28CC64 with DUMP 0S\r\n3167: no problem found','32','','bin',10,3,1,11),(3170,3168,9316,NULL,NULL,NULL,21,535,52,'P0400YE FBM4 REV-0P','1896506232','2023-02-09','Capacitors.\r\nEEPROM was changed, DUMP by 0P','32','','',10,3,1,11),(3171,3169,7467,NULL,NULL,NULL,21,535,134,'I/A Series P0400YG-0L FBM 06','1895191675','2023-02-09','Capacitors.\r\nAT28C64 - bad. Programming the new chip\r\nwith original revision 0L','32','','',10,3,1,11),(3172,3170,7467,NULL,NULL,NULL,21,535,134,'I/A Series P0400YG-0R FBM 06','1896462492','2023-02-12','Capacitors.\r\nAT28C64 - bad. Programming the new chip\r\nwith original revision 0R','32','','',10,3,1,11),(3173,3171,7467,NULL,NULL,NULL,21,535,134,'I/A Series P0400YG-0R FBM 06','1896092137','2023-02-13','Capacitors.\r\nAT28C64 - bad. Programming the new chip\r\nwith original revision 0R','32','','',10,3,1,11),(3174,3172,7467,NULL,NULL,NULL,21,535,134,'I/A Series P0400YG-0R FBM 06','1896472503','2023-02-13','Capacitors.\r\nAT28C64 - bad. Programming the new chip\r\nwith original revision 0R','32','','',10,3,1,11),(3175,3173,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0R FBM 1','SF98156927','2023-02-15','Capacitors\r\nEEPROM: NQ28C64A => AT28C64B\r\nProgramming rev 0R','32','','bin',10,3,1,11),(3176,3174,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192122','2023-02-15','Capacitors.\r\nProgramming the new AT28CC64 with DUMP 0L','32','','',10,3,1,11),(3177,3175,3755,NULL,NULL,NULL,21,533,17,'MODICON TM221CE40R','IB184600137','2023-02-15','Regeneration.','','','',10,6,1,11),(3178,3176,50000,NULL,NULL,NULL,21,609,86,'MultiSentry DMSTK40AA0DI','MR06UT708540001','2023-02-20','Incorrect negative battery PIN applied.\r\nFlash on battery charger board.\r\nReplacing the board with addition parts \r\nfor save original configuration.','79','','',40,20,1,11),(3179,3177,4700,NULL,NULL,NULL,21,530,3,'140CPS22400','31980573882','2023-02-21','Capacitors + 100nF 63V -2.','134','','',10,3,2,11),(3180,3178,4700,NULL,NULL,NULL,21,530,3,'140CPS22400','31980573002','2023-02-21','Capacitors + 100nF 63V -2.','134','','',10,3,2,11),(3181,3179,14000,NULL,NULL,NULL,21,530,1,'140CPU42402','31981753635','2023-02-21','Cleaning.','','','',4,2,5,11),(3182,3180,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301662','2023-02-21','Capacitors.\r\nR1, R2: 100k 2W\r\nR85: 39R 2W\r\nTesting, - ok.','123','','',15,6,1,11),(3183,3181,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301600','2023-02-21','Capacitors.\r\nR1, R2: 100k 2W\r\nR85: 39R 2W\r\nTesting, - ok.','123','','',15,6,1,11),(3184,3182,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301601','2023-02-22','Capacitors.\r\nR1, R2: 100k 2W\r\nR85: 39R 2W\r\nTesting, - ok.','123','','',15,6,1,11),(3185,3183,1800,NULL,NULL,NULL,21,584,35,'N15 0-1300KG/H','3183','2023-02-22','Capacitors.\r\nCurrent in: 14&15 \r\n4-20mA ~ 0-1300','','','',6,3,1,11),(3186,3184,4349,NULL,NULL,NULL,21,535,24,'FBM201','1808398124','2023-02-26','Changed transformer (dark isolator tape color)\r\nTest ok.','32','','',10,4,1,11),(3187,3185,45750,NULL,NULL,NULL,21,611,11,'AdaptView X13760326050','X1376032605001160720127','2023-02-26','Replacing the Touch Screen UC800\r\n(PIN to PIN compatible)','152','','',50,30,1,11),(3188,3186,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1897399557','2023-02-27','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3189,3187,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896505303','2023-02-27','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3190,3188,2492,NULL,NULL,NULL,21,396,33,'Ladegerat 2.250.1005','20674','2023-02-28','2510: Transformer 12V 320mA => 15V (RS 504-391)\r\n3188: Bridge','','','',5,3,2,11),(3191,3189,17980,NULL,NULL,NULL,21,612,85,'TMS-M','PM08-70-172','2023-03-01','Regeneration of sensor.','77, 50','CMC','',22,2,1,11),(3192,3190,18000,NULL,NULL,NULL,21,548,6,'HART 275','1364','2023-03-01','295: Repair input DC terminal.\r\n1364: Installed the new battery pack:\r\ninstead 1.2V 700mA Ni-Cd SANYO N-700AAC => RKB 49AA 1500 NiMH 1.2V\r\nbattery from remote control RF\r\n2823: Charging after very low battery level.\r\n3190: refreshing the battery pack. \r\nReplacing the fuse 100mA to T500.','','','',25,15,1,11),(3193,3191,3000,NULL,NULL,NULL,213,651,35,'D2494-YE REV. B0-C9','2000221','2023-03-05','Applied HV instead 12Vdc PS.\r\nFlashed LM358; 10R.','','','',20,8,1,11),(3194,3192,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R1','091095','2023-03-06','788: Electrican Laboratory.\r\nRelay.\r\nCapacitors.\r\n3192: Capacitors.','','','',7,3,2,11),(3195,3193,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','100495','2023-03-06','Capacitors.','','','',7,3,2,11),(3196,3194,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D40-AA','083095','2023-03-06','3194: Bad EEPROM\r\nThe new chip EEPROM programming 081C','','','magnesium',8,2,1,11),(3197,3195,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896505188','2023-03-06','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3198,3196,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515462','2023-03-07','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3199,3197,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515542','2023-03-07','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3200,3198,30000,NULL,NULL,NULL,21,560,6,'HART 375 Field Communicator','12072847','2023-03-09','390:  Touch screen AMT98234 replaced.\r\n1016: Braked DC plug (RS: 526-0825).\r\n1117: Making new battery block 1.2/1500mA*5 NiMH\r\n1277: Test only.\r\n1540: system card error.Bad placing becase shock during after fall.\r\n1576: recovery system provided. Flashing OS from internal dump.\r\nMaking the new cable.\r\n1628: Installed the new power socket. Soldering the MB, connectors\r\nSMD components. Re-Flashing EEPROM.\r\n1967: total fault. Changed battery pack without battery, \r\n- only board with closed module.\r\n2829: Very low battery level. PS-plug changed. \r\nCleaning and charging. \r\nTesting.\r\n3198: Falling unit. The CPU module was outside from the place.','60','','',20,10,1,11),(3201,3199,6700,NULL,NULL,NULL,21,116,98,'pH 2100/2H','54270/8294583/0747','2023-03-12','3199: Key -ENTER- int. print.\r\nCorrosion!','114','','',15,6,1,11),(3202,3200,7200,NULL,NULL,NULL,21,434,3,'EWS1500-24','CTH-405Z30-0010','2023-03-12','2417: Was working.\r\nCleaning.\r\n1N5408  -2.\r\nTest.\r\n3200: FAN, cleaning.','62','','',10,5,2,11),(3203,3201,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192066','2023-03-13','Capacitors.\r\nProgramming the new AT28CC64 with DUMP 0L','32','','',10,3,1,11),(3204,3202,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192116','2023-03-14','Capacitors.\r\nProgramming the new AT28CC64 with DUMP 0L','32','','',10,3,1,11),(3205,3203,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192072','2023-03-14','Capacitors.\r\nProgramming the new AT28CC64 with DUMP 0L','32','','',10,3,1,11),(3206,3204,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1897035898','2023-03-14','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3207,3205,47497,NULL,NULL,NULL,21,329,19,'TPS 820 400-230-230-5-8-D-S','WO18149429004','2023-03-19','SCRC471000 -2 => ESTEL T353-1000-20-72 -2.\r\n1N4007.\r\nFAN -3 (capacitor 1uF 400V)\r\nFull test 4-20mA 0-400VAC\r\nUsed 2 phases.\r\nPrice:\r\nhttps://www.radwell.co.uk/en-GB/Buy/SOLCON/SOLCON/TPS-820-600-115-115/?redirect=true','','','',40,20,1,11),(3208,3206,9316,NULL,NULL,NULL,21,535,52,'P0400YE FBM4 REV-0P','1896506219','2023-03-19','Capacitors.\r\nEEPROM was changed, DUMP by 0P','32','','',10,3,1,11),(3209,3207,9316,NULL,NULL,NULL,21,535,52,'P0400YE FBM4 REV-0P','1896465304','2023-03-19','Capacitors.\r\nEEPROM was changed, DUMP by 0P','32','','',10,3,1,11),(3210,3208,2492,NULL,NULL,NULL,21,396,33,'Ladegerat 2.250.1005','20287','2023-03-20','3208: 100uF 50V\r\nCleaning the terminal.','','','',5,3,2,11),(3211,3209,2492,NULL,NULL,NULL,21,396,33,'Ladegerat','30540','2023-03-20','760: 220uF 35V  -3.\r\nCable repair.\r\nCleaning.\r\n1499: cable repair\r\n2511: Int. print. Soldering\r\n3209: cleaning the terminal','','','',4,2,5,11),(3212,3210,2492,NULL,NULL,NULL,21,396,33,'Ladegerat','31033','2023-03-20','1500: Cable repair.\r\n3026: Transformer PIN bad soldering,\r\nCleaning.\r\nThe green LED was outside from the place.','','','',4,2,5,11),(3213,3211,47497,NULL,NULL,NULL,21,329,19,'TPS 820 400-230-230-5-8-D-S','WO18149429002','2023-03-20','SCRC471000 -2 => T9G0-20-11-0-3 JB56 11A This is: 2kV 1.1kA\r\n(570/900/221) -2.\r\n\r\nFAN -3 (capacitor 1uF 400V)\r\n\r\nFull test 4-20mA 0-400VAC\r\n\r\nWorking with 2 phases (L1 and L3).\r\n\r\nPrice:\r\nhttps://www.radwell.co.uk/en-GB/Buy/SOLCON/SOLCON/TPS-820-600-115-115/?redirect=true','','','',40,10,1,11),(3214,3212,9650,NULL,NULL,NULL,213,2,48,'PS2494-YE REV. C1, D2','2001464 0520','2023-03-21','PS board are flashed\r\nAll parts was replaced.\r\n240R\r\nTest.','104','pdf','',15,5,1,11),(3215,3213,9650,NULL,NULL,NULL,213,2,48,'PS2494-YE REV. C1, D2','2000939 1219','2023-03-21','PS board are flashed\r\nAll parts was replaced.\r\n240R\r\nTest.','104','pdf','',15,5,1,11),(3216,3214,9650,NULL,NULL,NULL,213,2,48,'PS2494-YE REV. C1, D2','2000730  5218','2023-03-22','PS board are flashed\r\nAll parts was replaced.\r\n240R\r\nCPU board: flashed D5 -2 BSS138 -2 and all IC\'s.\r\nTest.','104','pdf','',30,10,1,11),(3217,3215,870,NULL,NULL,NULL,21,518,11,'201','A197330','2023-03-22','2536: Incorrect indication.\r\nFactory reset and calibration\r\n4mA => 0\r\n20mA => 100\r\n3215: corrosion (U6) \r\n4mA => 0\r\n20mA => 200','pdf','','bin',2,1,2,11),(3218,3216,870,NULL,NULL,NULL,21,518,11,'201','A210379','2023-03-22','Corrosion \r\n4mA => 0\r\n20mA => 1500','pdf','','bin',2,1,2,11),(3219,3217,870,NULL,NULL,NULL,21,518,11,'201','A210384','2023-03-22','Corrosion \r\n4mA => 0\r\n20mA => 300','pdf','','bin',2,1,2,11),(3220,3218,870,NULL,NULL,NULL,21,518,11,'201','A208195','2023-03-22','Corrosion \r\nTest','pdf','','bin',2,1,2,11),(3221,3219,2100,NULL,NULL,NULL,213,530,3,'140CPS11420','330004800932901019','2023-03-23','Power side flashed.\r\nB9NK60Z -2, T4A, 0.2R II 0.2R -2 => 0.1R + 0.1R (My Mistake!!!!!!),\r\nLE Z240 (?) => 2*12V ZD.','','','',20,10,2,11),(3222,3220,2400,NULL,NULL,NULL,213,689,35,'AM530D VER-1.09-1 1.02','14480215','2023-03-26','C33: 470nF 275VAC -2\r\n220uF 35V - incorrect using polarity.\r\nCold soldering the terminals.','','','',10,5,1,11),(3223,3221,9650,NULL,NULL,NULL,213,2,48,'PS2494-YE REV. C0, D2','2020091  3513','2023-03-26','240R\r\nTest.','104','pdf','',15,3,1,11),(3224,3222,15000,NULL,NULL,NULL,21,143,35,'Smart Pro','IC96-030-0346-C','2023-03-26','Internal PS faulted.\r\nAll capacitors changed.\r\n\r\nhttps://www.ebay.com/itm/394170794520?hash=item5bc6690218:g:LaIAAOSw9Y1i1x3k&amdata=enc%3AAQAHAAAA4M%2Bauu1EVaVqH0AUnxTfcpgGxv1hWIoHCBKpteu8BjvCxLKqVT1Uo%2FpgZpCZPweLB6LtFroXoRUCE%2Bej%2BbVKhNMinlssNsY8K4%2FD5NCvk7FbLoc%2FuLjlJDbFv5Nb81AlH%2BdgwfoDVwb2aS9e%2FiDo7CWr2rgpFPeMgfG89y5QAlazV6749I5yqOWGGnE8%2Fq%2FogVkocW350p%2FW1yp8sOkeB71cmY0s3jyQatWCnFDnSzhBinFMUqn3xDDToGRbM1zaFPuJxtDc2UZW3f45Nea1j%2FOrlhtwH5tjGYeCR09iOa97%7Ctkp%3ABk9SR5y66tXjYQ','','','',10,3,1,11),(3225,3223,2000,NULL,NULL,NULL,21,469,76,'PAXS0010','3223','2023-03-27','Cleaning.','40','','RedLion',8,2,5,11),(3226,3224,2000,NULL,NULL,NULL,21,469,76,'PAXS001U','12073','2023-03-27','588: A/D faulted.\r\nChip: CIRRUS 5509-ASZ\r\n3224: 32768Hz - changed.\r\nCleaned and tested.','40','','RedLion',6,5,1,11),(3227,3225,3600,NULL,NULL,NULL,21,514,32,'12XR 40/6I 0-1A 4-20mA','3225','2023-03-27','Internal DC/DC 30V faulted.\r\nZD5V1. Corrosion.\r\nCalibration 4mA (offset).','','','',10,5,1,11),(3228,3226,2400,NULL,NULL,NULL,21,421,52,'Sineax V604 604-122','97308328','2023-03-27','Capacitors:\r\n47uF 35V => 25V\r\n220uF 16V => 6.3V\r\n4.7uF 50V\r\nT80mA => T125mA','','','',6,3,2,11),(3229,3227,15000,NULL,NULL,NULL,21,143,35,'Smart Pro','IS96-030-0346-B','2023-03-29','1207: Internal PS faulted.\r\nAll capacitors changed.\r\n3227: Testing TB4/1&2\r\nTB4/1&2 = input U5 6N136 after resistor R46 120R (5V ore pulses) \r\n-was ok.\r\nChanging the capacitors in PS. (was working before repair).','','','',10,6,1,11),(3230,3228,9500,NULL,NULL,NULL,215,721,45,'VON XF28-1960 Reflection System','3228','2023-04-02','1. Sabotage of working circuit for the purpose of destroying.\r\n(replacing black and white wires on panel connections). T10A\r\n2. Leakage in P-ch MOSFET RFG60P05E - charge/discharge system.\r\nWas replaced to 2 P-ch (17A+17A).','153','','',50,35,1,11),(3231,3229,3570,NULL,NULL,NULL,21,434,3,'EWS300-24','713Z14-0001W','2023-04-02','583: Flash secondary PS:\r\n2SK513, ZD15V, F2.5A => F3.15A, 4R3 -2.\r\nFAN repair.\r\nCleaning.\r\nFull test.\r\n2148: Cleaning. Capacitors, BYV26E.\r\nFull test.\r\n3229: Flash primary PS. The terminal repair.\r\n0.47uF.\r\n3300uF 35V -6.','62','','',15,8,2,11),(3232,3230,9650,NULL,NULL,NULL,213,2,48,'PS2494-YE REV. C0, D2','2020081  3513','2023-04-03','Corrosion\r\n240R\r\nTest.','104','pdf','',15,3,1,11),(3233,3231,9650,NULL,NULL,NULL,213,2,48,'PS2494-YE REV. C0, D2','2000738  5218','2023-04-03','Corrosion\r\n240R\r\nCPU from another one.\r\nTest.','104','pdf','',15,3,1,11),(3234,3232,2000,NULL,NULL,NULL,21,723,3,'AVPQ181M245155','K18440474','2023-04-03','Input circuit (filter) flashed.\r\nRepair, test.','','','',5,3,5,11),(3235,3233,2000,NULL,NULL,NULL,21,469,76,'PAXS0010','35041','2023-04-04','3158: Incorrect parameters.\r\nReplacing EEPROM\r\n3233: Incorrect parameters.\r\nReplacing EEPROM','40','','RedLion',8,2,5,11),(3236,3234,2600,NULL,NULL,NULL,21,2,1,'SIPART N1SVPB776288001','1137170','2023-04-04','2957: Cleaning.\r\nTest\r\n3234:  Cleaning.\r\nTest','103','','',12,3,9,11),(3237,3235,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00236622','2023-04-16','2543: Capacitors.\r\nCorrosion!\r\nTesting, - ok.\r\n3235: 4710uF 250V -2 => 680uF 250V -2.','123','','',15,6,1,11),(3238,3236,19120,NULL,NULL,NULL,21,724,17,'6E10 6C01','3226','2023-04-17','Soldering.\r\n100uF 10V => 100uF 25V','','','',10,5,5,11),(3239,3237,22660,NULL,NULL,NULL,21,2,13,'MIDIMASTER 6SE3121-7DG40','XAF362MD023D','2023-04-17','870: Internal Power Supply fault.\r\nAll capacitors changed.\r\nRepair PCB after heating.\r\nPre driver PCB: 1k for soft start and 1k x4 for HCPL7800.\r\nHCPL7800  -3,\r\nRepair the power board: \r\nchanged SS-resistors 22R (10R+10R)\r\n7&8 pins for start. Numeration from right side.\r\nTested.\r\n1081: Replacement Pre driver board after repair.\r\nFAN 24V.\r\nTested.\r\n1481: installed the DC BUS outputs and terminal.\r\n1484: because fault voltage of PS unit out of work PRE-DRV\r\nTotal.\r\n1624: no problem. Test, -ok. SW I-V was in V instead I (4-20mA)\r\n3237: IGBT 50A 1200V -1. FUSE FAST 60A. PS - full repair.\r\n7&8 pins for start. Numeration from right side.\r\nTested.','20','','',25,15,2,11),(3240,3238,2400,NULL,NULL,NULL,21,51,52,'OPTA1','253M18250656AS','2023-04-18','24C02 - (the new one programming)\r\nFlash on the DI. Shorted protection diode, int. print.','7 548#','','VACON',10,4,1,11),(3241,3236,19120,NULL,NULL,NULL,21,724,17,'6E10 6C01','3239','2023-04-18','Soldering.\r\nCleaning.','','','',10,5,5,11),(3242,3240,4300,NULL,NULL,NULL,21,552,32,'multi9 RCI 2181','0720','2023-04-18','Label: K12\r\n\r\nSMD: 4148, n-p-n 6Bn => FMMT619\r\n\r\nTest','30 155','','',10,3,1,11),(3243,3241,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','5174-95','2023-04-19','Capacitors, \r\nRelay 12V -6.','53','','SOLCON\\MRP2000_D\\X2816.bin',15,5,1,11),(3244,3242,4900,NULL,NULL,NULL,213,607,98,'t-mass 150 6BABL1-26Q1/0','T7071102000','2023-04-24','Logic MOSFET fault: PHT4NQ10T => IRLL014','','','',5,1,1,11),(3245,3243,4900,NULL,NULL,NULL,213,607,98,'t-mass 150 6BABL1-26Q1/0','T7071602000','2023-04-24','Logic MOSFET fault: PHT4NQ10T => IRLL014','','','',5,1,1,11),(3246,3244,4900,NULL,NULL,NULL,213,607,98,'t-mass 150 6BABL1-26Q1/0','T7071402000','2023-04-24','Logic MOSFET fault: PHT4NQ10T => IRLL014','','','',5,1,1,11),(3247,3245,4900,NULL,NULL,NULL,213,607,98,'t-mass 150 6BABL1-26Q1/0','T7070F02000','2023-04-24','Logic MOSFET fault: PHT4NQ10T => IRLL014','','','',5,1,1,11),(3248,3246,4900,NULL,NULL,NULL,213,607,98,'t-mass 150 6BABL1-26Q1/0','T7071002000','2023-04-24','Logic MOSFET fault: PHT4NQ10T => IRLL014','','','',5,1,1,11),(3249,3247,4900,NULL,NULL,NULL,213,607,98,'t-mass 150 6BABL1-26Q1/0','T7071502000','2023-04-24','Logic MOSFET fault: PHT4NQ10T => IRLL014','','','',5,1,1,11),(3250,3248,4900,NULL,NULL,NULL,213,607,98,'t-mass 150 6BABL1-26Q1/0','T7071502000','2023-04-24','Logic MOSFET fault: PHT4NQ10T => IRLL014','','','',5,1,1,11),(3251,3249,4900,NULL,NULL,NULL,213,607,98,'t-mass 150 6BABL1-26Q1/0','T7071302000','2023-04-24','Logic MOSFET fault: PHT4NQ10T => IRLL014','','','',5,1,1,11),(3252,3250,2700,NULL,NULL,NULL,21,498,98,'ACT-4','801116','2023-04-24','LM336-2.5, CA3130E\r\n\r\nCalibration:\r\n0...5A\r\n4...20mA\r\n\r\n(For 1A => 16/5+4=7.2mA)','136','','',10,5,1,11),(3253,3251,6500,NULL,NULL,NULL,21,498,6,'FOM-10','630572','2023-05-07','3251: Broken positive contact of the input PIN terminal.\r\nCapacitors.\r\n\r\nTested.','9','','',7,3,2,11),(3254,3252,6500,NULL,NULL,NULL,21,498,6,'FOM-10','630582','2023-05-07','Broken positive contact of the input PIN terminal.\r\nCapacitors.\r\n\r\nTested.','9','','',7,3,2,11),(3255,3253,6500,NULL,NULL,NULL,21,498,6,'FOM10','733042','2023-05-07','138 from 17.08.2014\r\nFounded problem in transmitter.\r\nFull tested.\r\n\r\n265 - cleaning all fiber optic sensors.\r\n3253: no problem founded. (changed BS170, 1N4148).','9','','',7,3,2,11),(3256,3254,14152,NULL,NULL,NULL,21,35,4,'SGDV-260D11A','D009YI218610002','2023-05-08','1706:\r\n0/996/682/930\r\nAXIS-2 3V2\r\nModule: PM100RLA120 -ok\r\nCorrosion!\r\nFlash on power PCB.\r\nLED DC BUS.\r\n2752: COOLER 24V: NMB-MAT MODEL 2410ML-05W-B69 (with digital TACHO) \r\n3 units are replaced to:\r\nRS: 335-7012 TUDC24B4 - without wire TACHO \r\nbut with internal TACHO circuit and 100% compatible.\r\nRS: 191-9277 - pin to pin 100% compatible. \r\nwith 2410ML-05W-B69 (only no pin to pin). \r\nCorrosion.\r\nCleaning.\r\n3254:\r\nCorrosion.\r\nCleaning.','101','','',20,8,1,11),(3257,3255,2400,NULL,NULL,NULL,21,2,99,'SIPART PS2 6DR4004-8J','A430-L3','2023-05-08','Broken input terminal.','','','',3,2,2,11),(3258,3256,2600,NULL,NULL,NULL,21,498,29,'PS-3','715710','2023-05-08','3256: C26: 22uF 50V.\r\n4u7\r\nScrews\r\nTest.','','','',5,2,5,11),(3259,3257,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','135295','2023-05-09','Capacitors.\r\nTesting, - ok.','123','','',15,6,1,11),(3260,3258,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00258636','2023-05-09','Capacitors.\r\nTesting, - ok.','123','','',15,6,1,11),(3261,3259,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00265085','2023-05-10','Capacitors.\r\nTesting, - ok.','123','','',15,6,1,11),(3262,3260,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00258465','2023-05-10','Capacitors.\r\nTesting, - ok.','123','','',15,6,1,11),(3263,3261,14600,NULL,NULL,NULL,21,530,3,'AS-P840-000','3659','2023-05-14','1987: MBR3045PT - short.\r\nCapacitors\r\nFuse 6A => 10A\r\n3261: MRB3045 => STPS3045 (another position on board)  \r\nTested.','123','','',8,4,1,11),(3264,3262,14152,NULL,NULL,NULL,21,35,4,'SGDV-260D11A','D0095B754910003','2023-05-15','1709: \r\n0/996/682/930\r\nModule: PM100RLA120 -ok\r\nAXIS-1 2V2\r\nCorrosion!\r\nFlash on power PCB.\r\nFast fuse 63A => 3*25A fast.\r\n2753: COOLER 24V: NMB-MAT MODEL 2410ML-05W-B69 \r\n(with digital TACHO) \r\n3 units are replaced to:\r\nRS: 335-7012 TUDC24B4 - without wire TACHO \r\nbut with internal TACHO circuit and 100% compatible.\r\nRS: 191-9277 - pin to pin 100% compatible. \r\nwith 2410ML-05W-B69 (only no pin to pin). \r\nCorrosion.\r\nCleaning.\r\n3262: Corrosion.\r\nCleaning.','101','','',5,2,1,11),(3265,3263,12000,NULL,NULL,NULL,21,118,18,'MS/E4003621011001ER3321111','V/12031/9/1','2023-05-15','1209: Capacitors.\r\n_________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\n\r\nBroken socket-PLCC, Inductor.\r\nReplaced the new screws.\r\n3263: Very dusty. Corrosion. Capacitors.','','','',20,10,1,11),(3266,3264,1920,NULL,NULL,NULL,213,514,43,'1MV3','9411640','2023-05-16','100uF 50V (closed PS)\r\n\r\n0-600VAC\r\n0-20mA\r\n\r\nCalibration.','','','',5,3,1,11),(3267,3265,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0K FBM1','1894053537','2023-05-17','Corrosion! Capacitors.\r\n28C64 -bad. \r\nProgramming the new AT28HC64 with DUMP 0K\r\nShort 0.22uF (ch8)\r\nShort transformer ch8 & ch7\r\nTest.','32','','bin',10,3,1,11),(3268,3266,1100,NULL,NULL,NULL,21,725,3,'GT-93020-0324','3266','2023-05-21','24VAC input\r\n\r\n22Uf 35v','','','',2,1,1,11),(3269,3267,4900,NULL,NULL,NULL,213,607,98,'t-mass 150 6BABL1-26Q1/0','T7071002000','2023-05-21','3246: Logic MOSFET fault: PHT4NQ10T => IRLL014\r\n3267: IRLL014','','','',5,1,1,11),(3270,3268,4900,NULL,NULL,NULL,213,607,98,'t-mass 150 6BABL1-26Q1/0','T7071102000','2023-05-21','3242: Logic MOSFET fault: PHT4NQ10T => IRLL014\r\n3268: IRLL014','','','',5,1,1,11),(3271,3269,4900,NULL,NULL,NULL,213,607,98,'t-mass 150 6BABL1-26Q1/0','T7071202000','2023-05-21','IRLL014','','','',5,1,1,11),(3272,3270,4900,NULL,NULL,NULL,213,607,98,'t-mass 150 6BABL1-26Q1/0','S8052B02000','2023-05-21','IRLL014','','','',5,1,1,11),(3273,3271,9316,NULL,NULL,NULL,21,535,52,'P0400YE FBM4 REV-0P','1896506231','2023-05-22','2446: Capacitors.\r\nProblem with AT28C64B.\r\nEEPROM was changed, DUMP by 0P \r\n\r\nch_2: 49R9\r\n3160: no problem found. Transformers soldering.\r\n3271: leakage in ceramic capacitors 350nF near AD-chip\r\n350nF => 1uF + 47nF.','32','','',10,3,1,11),(3274,3272,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA FBM 01 REV-0L','1895192086','2023-05-22','2176: Faulted after self reset\r\n\r\nVery bad condition (nozle):\r\n330uF 35V -10,\r\n1000uF 10V => 1500uF 16V - all.\r\n\r\n3272: Programming the new EEPROM of 0L dump.','32','','',8,3,1,11),(3275,3273,2400,NULL,NULL,NULL,213,689,35,'AM528D 1.25 1.18-1','3273','2023-05-24','470nF 275VAC -1\r\n220uF 35V.\r\nCold soldering the terminals.','','','',10,2,1,11),(3276,3274,2400,NULL,NULL,NULL,213,689,35,'AM528D 1.25 1.18-1','3274','2023-05-24','470nF 275VAC -2\r\n220uF 35V -2.\r\nCold soldering the terminals.','','','',10,2,1,11),(3277,3275,1100,NULL,NULL,NULL,21,725,3,'GT-93020-0324','3275','2023-05-28','24VAC input\r\n\r\n22Uf 35v','','','',2,1,1,11),(3278,3276,6600,NULL,NULL,NULL,213,326,3,'LLS-7120','90K030001','2023-05-29','Repair the case (box).\r\nInt. print.\r\nMUR460 => 2*UF5408\r\n\r\n\r\n\r\nhttps://www.ebay.com/itm/144415765150?hash=item219fd9b29e:g:8KQAAOSwoNhj0k4f&amdata=enc%3AAQAIAAAAwBlHM5DnTi6CLUqbh3uyufi30uOZDuAblE0LjvhDmLOlZC6r3aLRKYxEMVTFISoTyAiciMQ13YZK632Rr6lNMJ1UHse%2FvdlhmiQfis3W4N%2BZHdpgnIhQNCkQT3jGiQ5IFsvKYNy4ZV3sqpyx6i8lqz8xHlUC01l60eK3rpN2KlWNqjH6Lu3OFRJu7hF0jP0N2xxa9HCdnV1pJYoHUHOlK%2FSOCIrBpvGuvcwGDyF0RxQbH68whIp6uJWWMZkZl8U4QA%3D%3D%7Ctkp%3ABk9SR5LZ8e-MYg','','','',30,20,2,11),(3279,3277,2200,NULL,NULL,NULL,21,533,3,'ABL8 RPS24100','0231232','2023-05-31','Cleaning\r\nTest.','','','',15,10,1,11),(3280,3278,14100,NULL,NULL,NULL,21,531,98,'3081','PH H00-51454','2023-05-31','1544: Cleaning. \r\nAdd screws, cables, inputs.\r\nTest without calibration.\r\nTest after calibration.\r\n3278: Cleaning.','35','','',20,8,1,11),(3281,3279,6700,NULL,NULL,NULL,21,116,98,'pH 2100/2H','54270/8294583/0747','2023-05-31','3199: Key -ENTER- int. print.\r\nCorrosion!\r\n3279: Corrosion. Calibration.','114','','',15,6,1,11),(3282,3280,5200,NULL,NULL,NULL,21,498,24,'AFM-16','630561','2023-06-04','1684: Capacitors\r\n56R -2.\r\n3280: To ch_1 was applied power voltage.\r\nLM358, 1N4148, TIP31C, BS250 - shorted, F3/4A -2 => T500mA -2.\r\n21R 1% => 22R II 470R 1%.','9','','',20,10,1,11),(3283,3281,2100,NULL,NULL,NULL,213,530,3,'140CPS11420','330004800932901766','2023-06-06','Power side flashed.\r\nB9NK60Z -2, T4A, 0.2R II 0.2R -2 => 0.1R 1.5W 1%,\r\nLE Z240 (?) => 2*12V ZD.\r\n3300uF 6.3V -3.','','','',20,5,2,11),(3284,3282,2100,NULL,NULL,NULL,213,530,3,'140CPS11420','330004800932901723','2023-06-06','Power side flashed.\r\nB9NK60Z -2, T4A, 0.2R II 0.2R -2 => 0.1R 1.5W 1%,\r\nLE Z240 (?) => 2*12V ZD.\r\n3300uF 6.3V -3.','','','',20,5,2,11),(3285,3283,55000,NULL,NULL,NULL,213,726,45,'220120-47  88-T-011','7748','2023-06-08','Voltmeter plug was outside from the place.\r\nAdd extended resistor 8k to divider voltage.\r\nOutput voltage calibration.','154','','',40,20,1,11),(3286,3284,1500,NULL,NULL,NULL,21,599,8,'MSR 020','M2-U3','2023-06-11','2921: 100uF 50V, 330uF 63V => 330uF 50V 125C\r\nRelay 24V -2.\r\nTr-r: 230V/24V/3W => 2.8W 24V (732-0395).\r\n3284: 2.8W 24V (732-0395).','','','',5,4,1,11),(3287,3285,9316,NULL,NULL,NULL,21,535,52,'P0400YE FBM4 REV-0L','1894516788','2023-06-11','2415: Capacitors.\r\nProgramming the new AT28C64B rev 0L\r\n\r\nch_3: 49R9, J11-SMD => 4148\r\n3285: testing.','32','','',15,5,5,11),(3288,3286,2500,NULL,NULL,NULL,213,727,6,'i-7022','I7022CR000BBEBI00249','2023-06-12','ZD5V 0.5W => ZD5V 3W\r\n470uF 16V, 47uF 50V.\r\nMC34063','','','',5,3,1,11),(3289,3287,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','CM902WL-AS95180013','2023-06-13','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3290,3288,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','CM902WL-AS95180096','2023-06-13','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3291,3289,1600,NULL,NULL,NULL,21,591,54,'DMA14-31A20R-S1','937559-024','2023-06-15','1069: Capacitors.\r\n\r\nCalibration: \r\n\r\nTerminal pin 8&1 = 0.06V\r\nScale: 0V => 0kV; 60mV => 20000kV\r\n3289: Capacitors.','','','',6,3,1,11),(3292,3290,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARRE-1279','2023-06-15','No problem found.\r\nFull test with sensor.','144','','',12,4,1,11),(3293,3291,26000,NULL,NULL,NULL,21,272,67,'174CEV20040','31084470201','2023-06-18','FAN 12V with tacho CPU, FAN chassis 12V.\r\n9/991/118/313','','','',30,20,1,11),(3294,3292,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D34-AA','311327','2023-07-03','Bad EEPROM\r\nThe new chip programming 043E','','','magnesium',8,2,1,11),(3295,3293,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','0027478','2023-07-03','Neon lamp -8.\r\n22M -10.','8','','',7,4,1,11),(3296,3294,12100,NULL,NULL,NULL,21,161,3,'P810','94489','2023-07-04','1989: Capacitors.\r\n3294: applied 230V instead 120V\r\n470uF 200V -2 => 560uF 200V -2.\r\n330k -2, 39R 2W -2.\r\nThe input terminal soldering.\r\nTest.','123','','',15,4,1,11),(3297,3295,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301829','2023-07-04','267: 1N5404, \r\nT4A => T7A.\r\nAll capacitors changed.\r\nTested.\r\n1990: capacitors.\r\n3295: 230VAC was applied instead 120VAC\r\n1.5KE220 single to 1.5KE220 dual.\r\n100k 2W -2.\r\n10000uF 6.3V -2.','123','','',15,4,1,11),(3298,3296,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301582','2023-07-05','Capacitors.\r\n100k 2W -2.\r\nTested.','123','','',15,4,1,11),(3299,3297,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301578','2023-07-05','Capacitors.\r\n100k 2W -2.\r\nTested.','123','','',15,4,1,11),(3300,3298,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301589','2023-07-05','Capacitors.\r\n100k 2W -2.\r\nTested.','123','','',15,4,1,11),(3301,3299,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301810','2023-07-06','Capacitors.\r\n100k 2W -2.\r\nTested.','123','','',15,4,1,11),(3302,3300,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301830','2023-07-06','Capacitors.\r\n100k 2W -2.\r\nTested.','123','','',15,4,1,11),(3303,3301,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301584','2023-07-09','Capacitors.\r\n100k 2W -2.\r\nTested.','123','','',15,4,1,11),(3304,3302,14600,NULL,NULL,NULL,21,161,3,'AS-P840-000','4559','2023-07-09','2001: Capacitors\r\n3302: 4700uF\r\nTest','123','','',10,4,1,11),(3305,3303,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301596','2023-07-09','Capacitors.\r\n100k 2W -2.\r\nTested.','123','','',15,5,1,11),(3306,3304,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00202674','2023-07-10','Capacitors.\r\nTest.','123','','',15,4,1,11),(3307,3305,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301566','2023-07-10','Capacitors.\r\n100k 2W -2.\r\nTested.','123','','',15,5,1,11),(3308,3306,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','20050750','2023-07-10','Relay ch_7 (LZ-5H-C)','','','',4,3,1,11),(3309,3307,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301554','2023-07-11','Capacitors.\r\n100k 2W -2.\r\nTested.','123','','',15,5,1,11),(3310,3308,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301580','2023-07-11','Capacitors.\r\n100k 2W -2.\r\nTested.','123','','',15,5,1,11),(3311,3309,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','311202','2023-07-11','1988: No problem, test only.\r\n3309: 10000uF 6.3V  -2.','','','',10,3,1,11),(3312,3310,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','0040087','2023-07-11','1031: Fuse T7A (Ch_1).\r\n3310: Relay_ch5','','','',4,3,1,11),(3313,3311,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','309278','2023-07-12','Capacitors.\r\nTested.','123','','',15,5,1,11),(3314,3312,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301574','2023-07-12','Capacitors.\r\n100k 2W -2.\r\nTested.','123','','',15,5,1,11),(3315,3313,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARKE-0602','2023-07-16','No problem found.\r\nFull test with sensor.','144','','',12,4,1,11),(3316,3314,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARYB-0791','2023-07-16','1468: Corrosion,\r\ncleaning.\r\n3314: no problem\r\nTest','144','','',6,3,1,11),(3317,3315,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA FBM 01','1896505260','2023-07-16','1571: 330uF 35V -10,\r\n1000uF 10V => 1000uF 16V - all.\r\nSoldering.\r\n\r\nFull test - ok.\r\n2172: NO PROBLEM.\r\n2476: new AT28C64B installed with updated DUMP DA-0P\r\n3165: short (non constant) between EEPROM pin\'s.\r\n3315: soldering the all IC\'s. 6264 - changed.','32','','',8,3,1,11),(3318,3316,14600,NULL,NULL,NULL,21,161,3,'AS-P840-000','002754','2023-07-18','Capacitors\r\nMBR3045 => STP3045\r\nTest','123','','',10,4,1,11),(3319,3317,9200,NULL,NULL,NULL,21,605,116,'AW2601','030-2082-001','2023-07-20','2558: Corrosion\r\nInt. print\r\n3317: corrosion\r\nTest','121','','',10,6,1,11),(3320,3318,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','311190','2023-07-20','Capacitors.\r\nTested.','123','','',15,5,1,11),(3321,3319,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00238504','2023-07-20','Corrosion!\r\nCapacitors. 27000uF => 33000uF\r\nLeakage in MC7912 isolator + MC7912\r\nTested.','123','','',15,5,1,11),(3322,3320,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','153360','2023-07-23','Capacitors. 27000uF => 33000uF\r\nTested.','123','','',15,5,1,11),(3323,3321,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301254','2023-07-24','Capacitors.\r\n100k 2W -2.\r\n39R 2W\r\nTested.','123','','',15,4,1,11),(3324,3322,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301831','2023-07-24','Capacitors.\r\n100k 2W -2.\r\n39R 2W\r\nTested.','123','','',15,4,1,11),(3325,3323,17980,NULL,NULL,NULL,21,612,85,'TMS-M','PM08-70-172','2023-07-24','Regeneration of sensor.','77, 50','CMC','',22,2,1,11),(3326,3324,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301577','2023-07-24','Capacitors.\r\n100k 2W -2.\r\n39R 2W\r\nTested.','123','','',15,4,1,11),(3327,3325,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301936','2023-07-25','Capacitors.\r\n100k 2W -2.\r\n39R 2W\r\nTested.','123','','',15,4,1,11),(3328,3326,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301579','2023-07-25','Capacitors.\r\n100k 2W -2.\r\n39R 2W\r\nTested.','123','','',15,4,1,11),(3329,3327,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301816','2023-07-26','Capacitors.\r\n100k 2W -2.\r\n39R 2W\r\nTested.','123','','',15,4,1,11),(3330,3328,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00246296','2023-07-27','Corrosion!\r\nCapacitors.\r\nTested.','123','','',15,4,1,11),(3331,3329,5200,NULL,NULL,NULL,21,498,24,'AFM-16','2725907','2023-07-30','3329: Capacitors\r\n56R -2.','9','','',10,5,1,11),(3332,3330,3200,NULL,NULL,NULL,21,498,43,'DCI-2','623069','2023-07-30','848:\r\nCapacitors.\r\nCalibrated:\r\n0V => 4mA\r\n6V => 20mA\r\n\r\n1026:\r\nCalibrated:\r\n0V => 4mA\r\n6V => 20mA\r\n3330: 3.3uF 400V\r\ncapacitors','9','','',10,4,1,11),(3333,3331,6970,NULL,NULL,NULL,21,434,3,'EWS600-24','CQT-821Z12-0014','2023-07-30','1025, 166\r\n\r\n3331: 3.9uF 400V, FAN\r\ncapacitors','62','','',8,4,1,11),(3334,3332,6970,NULL,NULL,NULL,21,434,3,'EWS600-24','720Z12-0072W','2023-07-30','FAN','62','','',8,4,1,11),(3335,3333,2492,NULL,NULL,NULL,21,396,33,'Ladegerat','24155','2023-08-03','535: Soldering.\r\nCleaning.\r\n1793:Cleaning.1901: Cleaning\r\n2509: capacitors, cleaning\r\n3092: Cleaning\r\n3333: Cleaning','','','',5,3,2,11),(3336,3334,1200,NULL,NULL,NULL,21,552,32,'multi9 RCI','21181 0720','2023-08-06','2534: Total flash\r\n1k, 10nF 400V, 100uF 50V, PCB\r\n3334: LT1413 => LT1013 \r\nCapacitors.','30 155','','',20,10,5,11),(3337,3335,9200,NULL,NULL,NULL,21,605,116,'081-5212-322/041-5062-010','668880-01-002','2023-08-06','R20: 56R 1W  -ok\r\nIC\'s - ok\r\nChanged 470uF 50V','121','','',10,6,1,11),(3338,3336,12100,NULL,NULL,NULL,21,161,3,'AS-P800-003','2787H','2023-08-07','220uF 50V.\r\nFuse Holder + F10A\r\nTested.','123','','',15,6,1,11),(3339,3337,1800,NULL,NULL,NULL,21,498,43,'ACT-2','519832','2023-08-08','0-5A => 4-20mA (1A => 7.2mA)\r\n \r\nCalibrated','136','','',6,2,1,11),(3340,3338,12500,NULL,NULL,NULL,21,572,76,'FIP 0401 FBW INTECONT PLUS','G0003991','2023-08-08','482: Corrozion.\r\n1uF - broken (circuit LM337T)\r\nSaving data.\r\nCD4046 -2 (impossible test).\r\nMaking circuit.\r\nCleaning.\r\nNo problem found.\r\n2032: No problem found.\r\nX2 - new socket installed.\r\n!!! LC give pulses from controller instead voltage.\r\n2302: BTZ03C36 (V8, V10) broken PIN\'s corrosion.\r\nIonistor 0.47F 5.5V\r\n2766: no problem\r\n3338: no problem. Replaced 4052 -2, 4046 -2, AD708 - tested all.','44','meir','bin',10,6,1,11),(3341,3339,4300,NULL,NULL,NULL,21,552,32,'multi9 RCI 21181','9547','2023-08-09','2819: Label: K12\r\n\r\nCapacitors.\r\nBC547 => BC639.\r\nSYR001.\r\n22R\r\nTest 1ADC\r\n3339: \r\nSYR001, BC639 => 2N2222\r\nTest 1ADC','30 155','','',10,5,1,11),(3342,3340,80450,NULL,NULL,NULL,21,390,98,'FIT-63293 2700R11BBAEZZZ','3032065','2023-08-14','2116: 1uF 450V\r\n3340: tested the HCNW4506 -ok, but replaced the new (U13, U14).','137','','',10,5,1,11),(3343,3341,80450,NULL,NULL,NULL,21,390,98,'FIT-99507 1700R12ABZEZZZ','3117313','2023-08-14','3341: 1uF 450V\r\nU14: HCNW-4506 are bad. Replaced every one (U13, U14).\r\nTesting.','137','','',10,5,1,11),(3344,3342,2492,NULL,NULL,NULL,21,396,33,'Ladegerat','30540','2023-08-16','760: 220uF 35V  -3.\r\nCable repair.\r\nCleaning.\r\n1499: cable repair\r\n2511: Int. print. Soldering\r\n3209: cleaning the terminal\r\n3342: soldering the PIN of transformer.','','','',4,2,5,11),(3345,3343,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00249125','2023-08-16','Capacitors. \r\n27000uF => 33000uF\r\nTested.','123','','',15,5,1,11),(3346,3344,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','301607','2023-08-16','Capacitors.\r\n100k 2W -2.\r\n39R 2W\r\nTested.','123','','',15,4,1,11),(3347,3345,17980,NULL,NULL,NULL,21,612,85,'TMS-M','PM08-70-172','2023-08-17','Regeneration of sensor.','77, 50','CMC','',22,2,1,11),(3348,3346,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','304234','2023-08-20','Capacitors.\r\n39R 2W\r\nTested.','123','','',15,4,1,11),(3349,3347,1000,NULL,NULL,NULL,21,652,3,'PA1015-1HE','1866','2023-08-20','1866: Capacitors\r\n3347: HV was applied.\r\n10uF 400V, 22uF 400V, varistor, T2A => T3.15A','','','',4,3,1,11),(3350,3348,1000,NULL,NULL,NULL,21,728,3,'BA018-050250AXE','PT00-161004843','2023-08-20','Capacitors, T1A','','','',3,2,1,11),(3351,3349,12000,NULL,NULL,NULL,21,329,19,'RVS-DN','885-95','2023-08-21','31A, 400V, control: 230V.\r\n2200uF 25V, \r\n470uF 16V => 220uF 35V.\r\nIncorrect flat connector inserting.','','','',7,3,1,11),(3352,3350,12000,NULL,NULL,NULL,21,329,19,'RVS-D','3/1/96','2023-08-21','400V, control: 230V.\r\nIncorrect flat connector inserting.','','','',7,3,1,11),(3353,3351,47497,NULL,NULL,NULL,21,329,19,'6878-13/04 HMCP-050K2-E 11kW','3TF45','2023-08-21','SKKT92/12E -2 => SKKT92/16E -2\r\nPrice:\r\nhttps://www.radwell.co.uk/en-GB/Buy/SOLCON/SOLCON/TPS-820-600-115-115/?redirect=true','','','',16,8,1,11),(3354,3352,12000,NULL,NULL,NULL,21,396,103,'00071','0713','2023-08-21','The new IDC connector installed (673-7742).','65','','',3,1,1,11),(3355,3353,1000,NULL,NULL,NULL,138,729,33,'24C 15A','3353','2023-08-22','No problem \r\nWiring and full test\r\nMeir Avitan\r\n052-2963079\r\n\r\n891540080 1000 shekel','','','',5,2,1,11),(3356,3354,6500,NULL,NULL,NULL,21,498,6,'FOM-10','630572','2023-08-24','3251: Broken positive contact of the input PIN terminal.\r\nCapacitors.\r\nTested.\r\n3354: Broken bayonet of the photo sensor, Repairing.\r\nTested 74LS251 -2 and all channels.','9','','',12,6,2,11),(3357,3355,6500,NULL,NULL,NULL,21,498,6,'FOM10','733042','2023-08-24','138 from 17.08.2014\r\nFounded problem in transmitter.\r\nFull tested.\r\n265 - cleaning all fiber optic sensors.\r\n3253: no problem founded. (changed BS170, 1N4148).\r\n3355: ch8; ch10 - was bad, cleaning.\r\nFull test with fiber optic.','9','','',8,4,2,11),(3358,3356,18000,NULL,NULL,NULL,21,51,13,'30CX4G2N0','1245352','2023-08-27','1678: VFD without BRAKE function.\r\nWas flashed SS resistors.\r\nTerminal replaced.\r\nMaking the CHOPER function: SKM195GAL123D\r\nHV wires.\r\n3356: Corrosion. FAN 24V, EEPROM, Relay 12V -2.\r\nTest.','7 548','','bin',20,8,1,11),(3359,3357,18000,NULL,NULL,NULL,21,51,13,'45CX4G2N0','9468156','2023-08-28','3300uF 430V -1 (changed 2 units). \r\n100R smd on relay.\r\nNew EEPROM installed.\r\nFor test PIN 10 (external fault function must be open).\r\nTest.','7 548','','bin',25,15,1,11),(3360,3358,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192068','2023-09-03','2460: Capacitors.\r\n28C64 -bad. \r\nProgramming the new AT28CC64 with DUMP 0L\r\n3358: ch_1 faulted\r\nSoldering and cleaning.','32','','',10,3,1,11),(3361,3359,3500,NULL,NULL,NULL,21,605,116,'811-1206-EOO/X41-5005-024','','2023-09-04','1098: Relay 12V 2CHO,\r\n10uF 50V.\r\nTesting of all IC\'s.\r\n3359: Soldering the connectors, Relay 12V 2CHO.','pdf','','',10,3,1,11),(3362,3360,3500,NULL,NULL,NULL,21,605,116,'811-1206-EOO/X41-5005-024','30-4502-004','2023-09-04','CA3260E => TLC2272\r\n10uF 50V.\r\nSoldering the wire connector. LED.','pdf','','',10,5,1,11),(3363,3361,3500,NULL,NULL,NULL,21,605,116,'AW2601','030-2082-001','2023-09-04','Corrosion\r\nInt. print','121','','',10,6,1,11),(3364,3362,5655,NULL,NULL,NULL,21,530,22,'140 DDI 353 00','31993606712','2023-09-05','328: Channel 5 and 28 was bad.\r\nLackage in SMD thantalium capacitors 10nF.\r\n3362: corrosion (ch_22)','','','',10,5,5,11),(3365,3363,2335,NULL,NULL,NULL,213,359,10,'RMA-19-02','115 D0','2023-09-06','Capacitors\r\n100uF 10V -4\r\nNo problem.\r\n\r\nhttps://www.radwell.co.uk/Buy/SIEMENS/MILLTRONICS/RMA-19-02/?redirect=true','pdf','','bin',12,6,1,11),(3366,3364,3500,NULL,NULL,NULL,21,605,116,'AW1839 09-5129-001','3364','2023-09-06','Calibration.','121','','',6,3,1,11),(3367,3365,3500,NULL,NULL,NULL,21,605,116,'AW1839 09-5129-001','3365','2023-09-06','Calibration.','121','','',6,3,1,11),(3368,3366,3500,NULL,NULL,NULL,21,605,116,'AW1839 09-5129-001','3366','2023-09-06','Calibration.','121','','',6,3,1,11),(3369,3367,80450,NULL,NULL,NULL,21,390,98,'FIT-63293 2700R12BBAEZZZ','3269520','2023-09-07','Communication with sensor is ok.\r\n1uF 450V was not good condition, - replaced.\r\nC27: 22uF 25V SMD - bad.\r\nTest.','137','','',10,5,1,11),(3370,3368,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D40-AA','','2023-09-10','3194: Bad EEPROM\r\nThe new chip EEPROM programming 081C\r\n3368: internal ERROR 56\r\nProblem in EEPROM revision (not correct). \r\nInstalled the another revision 081D (according with datasheet ABB 4D40)','','','magnesium',8,3,1,11),(3371,3369,2400,NULL,NULL,NULL,21,2,99,'N1WN04AD1800101','4063074','2023-09-13','3369: Corrosion.','103','','',3,2,2,11),(3372,3370,2400,NULL,NULL,NULL,21,2,99,'N1WN04AD1800101','087398','2023-09-13','3370: Corrosion.','103','','',3,2,2,11),(3373,3371,2400,NULL,NULL,NULL,21,2,99,'C73451','036579','2023-09-13','Corrosion.','103','','',3,2,2,11),(3374,3372,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','1053223','2023-09-13','1872: Cleaning.\r\nTest.\r\n3372: corrosion','103','','',12,3,9,11),(3375,3373,1400,NULL,NULL,NULL,21,611,149,'6400-2716-01','TE1709 24495','2023-09-13','Part number BRD04873\r\n\r\nCorrosion. Instead SMD 3k -3 =. 1k\r\nTesting with magnetic stick.\r\n\r\nhttps://www.chillerparts.ae/product/brd04873-dual-binary-input-board/','','','',8,4,2,11),(3376,3374,1400,NULL,NULL,NULL,21,611,149,'6400-2716-01','0220591122','2023-09-13','Part number BRD04873\r\n\r\nCorrosion. Instead SMD 3k -3 =. 1k\r\nTesting with magnetic stick.\r\nShort on 485 chip ISL81487, - changed from another good board.\r\n\r\nhttps://www.chillerparts.ae/product/brd04873-dual-binary-input-board/','','','',8,4,2,11),(3377,3375,2400,NULL,NULL,NULL,21,2,99,'N1A910AF4473203','4146998','2023-09-14','Corrosion.\r\nLCD changed\r\nTest','103','','',8,4,2,11),(3378,3376,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049784392','2023-09-18','Varistor \r\nT2A; fuse holder','','','',9,5,1,11),(3379,3377,4700,NULL,NULL,NULL,21,530,3,'140CPS12400','20001002','2023-09-18','330uF 200V - was empty, changed 2\r\n1000uF 25V, 1000uF 10V -2.','134','','',10,5,2,11),(3380,3378,4700,NULL,NULL,NULL,21,530,3,'140CPS12400','20000998','2023-09-19','330uF 200V - was good, changed 2\r\n1000uF 25V, 1000uF 10V -2.','134','','',10,5,2,11),(3381,3379,4700,NULL,NULL,NULL,21,530,3,'140CPS12400','20002556','2023-09-19','330uF 200V - 1 from 2 was good, changed 2\r\n1000uF 25V, 1000uF 10V -2.','134','','',10,5,2,11),(3382,3380,12000,NULL,NULL,NULL,21,396,1,'E-AN08A0 V1/1','1287','2023-09-20','1287: Total flash (230V was on analog output)\r\nRepair PCB, inductors + 0.1uF.\r\nBoard E-AN02A4 V1/2 -2 unit - total:\r\nEX=TVS 39V, A82=1N4148\r\nTLC272C 4.\r\n1289: Unknown autside  problem (HV)\r\nFlash on ground cable. Relay 12V.\r\nRepair.\r\nOA -4 (TLE2022C and other CMOS compatible).','','','nbb',20,12,1,11),(3383,3381,3570,NULL,NULL,NULL,21,175,3,'HWS600-24','5CT-821CJT-0201W5007','2023-09-20','10R smd - flashed\r\n470R smd over heating.\r\nAborting the CN1, making the internal connection','','','',10,4,2,11),(3384,3382,12000,NULL,NULL,NULL,21,329,19,'TPS 820 400-230-230-5-8-D-S','3382','2023-09-21','2666: SCRC471000 -2 => ESTEL T353-1000-20-72 -2.\r\n\r\n\r\n3382: Reset -key, 220uF, cleaning.','','','',10,4,1,11),(3385,3383,9316,NULL,NULL,NULL,21,535,52,'P0400YE FBM4 REV-0S','SF02307529','2023-09-26','3383: Capacitors.\r\nProgramming the new AT28C64B rev 0S','32','','',15,5,5,11),(3386,3384,9316,NULL,NULL,NULL,21,535,52,'P0400YE FBM4 REV-0K','1892290053','2023-09-26','Capacitors. Corrosion!\r\nProgramming the new AT28C64B rev 0K\r\n49R9 - flashed.','32','','',15,5,5,11),(3387,3385,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240','5908984','2023-09-27','FAN flow corrosion.\r\nInt. print.\r\nPower side: ZD15V (was ok) -2.\r\n470k, T20A, 2k2, 4148, ZD15V, UC3855.\r\nAttiny45 + programming.\r\nUsed the new silicone paste for PFC and output voltage side PS\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','bin',30,16,1,11),(3388,3386,47497,NULL,NULL,NULL,21,329,19,'TPS820 400-230-230-5-8-D-S','WO19184390002','2023-09-28','SKKT92/12E -2 => SKKT92/16E -2\r\nFAN repair (1uF)\r\nPrice:\r\nhttps://www.radwell.co.uk/en-GB/Buy/SOLCON/SOLCON/TPS-820-600-115-115/?redirect=true','','','',10,4,1,11),(3389,3387,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240','6460610','2023-10-09','FAN flow corrosion.\r\nInt. print on power board between 51k and 51k SMD.\r\nChanged many SMD resistors on the power board.\r\n470k CPU side.\r\nUsed the new silicone paste for PFC and output voltage side PS\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','bin',30,16,1,11),(3390,3388,80450,NULL,NULL,NULL,21,390,98,'FIT-99607 1700R12ABZEZZZ','14123454','2023-10-10','1uF 450V \r\nU14: HCNW-4506 are ok. Replaced every one (U13, U14).\r\nTesting.','137','','',10,8,1,11),(3391,3389,9650,NULL,NULL,NULL,213,2,48,'PS2494-YE REV. C0, D2','202090','2023-10-11','Corrosion!\r\n240R, IGBT\r\nReplacing SMD resistors because corrosion. Cleaning.\r\nTest.','104','pdf','',40,20,1,11),(3392,3390,5100,NULL,NULL,NULL,21,607,98,'Solicap M FTI55-AAD2RGJ43A3A','S8006E10FE','2023-10-12','Sensor cable repair','','','',7,3,1,11),(3393,3391,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','278689','2023-10-15','464:Capacitors,\r\nrelays 24V -2.\r\n\r\n3391: Capacitors.','','','',5,2,2,11),(3394,3392,14100,NULL,NULL,NULL,21,531,98,'3081','A00-36344','2023-10-16','1760: Cleaning. \r\nAdd screws, cables, inputs.\r\nResistor 2 GOhm - broken.\r\nTest without calibration.\r\nTest after calibration.\r\n2788: Add screws.\r\nCalibration.\r\n3392: Cleaning and full test.','35','book 3081','',20,8,1,11),(3395,3393,20000,NULL,NULL,NULL,21,396,1,'E-AN08A0 V1/1','0559856690','2023-10-17','Total flash (230V was on analog output)\r\nRepair PCB, Varistor: VC12063D650DP + 0.1uF.\r\nCPU SAB80C537 infenion - bad.\r\nBoard E-AN02A4 V1/2 -2 unit - total flash:\r\nEX=TVS 39V, A82=1N4148\r\n\r\nFlash on ground cable. \r\n\r\nOA -4','157','','nbb',20,12,1,11),(3396,3394,5216,NULL,NULL,NULL,21,272,35,'TSX MOMENTUM 170ADI74050','21014190808','2023-10-18','PS: R1, R2 - 2R2.\r\n\r\nprice:\r\nhttps://www.newark.com/schneider-electric/170adi74050/programmable-logic-controller/dp/18C3748?CMP=AFC-DESIGNFAST','','','',5,3,1,11),(3397,3395,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D40-AA','259593','2023-10-18','Bad EEPROM\r\nThe new chip EEPROM programming 043E','','','magnesium',8,3,1,11),(3398,3396,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','272231','2023-10-18','Capacitors, relay 24V -2.','','','',7,2,5,11),(3399,3397,1100,NULL,NULL,NULL,21,725,3,'GT-93020-0324','3397','2023-10-19','24VAC input\r\n\r\nAll capacitors','','','',2,1,1,11),(3400,3398,6970,NULL,NULL,NULL,21,434,3,'EWS600-24','CQT-811K30-0025-P287','2023-10-23','836: New.\r\nRegulation +26V\r\n3032: Regulation +26V (from +17V)\r\nRegulation +26V (incorrect regulation)','62','','',2,1,1,11),(3401,3399,20000,NULL,NULL,NULL,21,396,1,'E-AN08A0 V1/1','0553829099','2023-10-24','Total flash (230V was on analog output)\r\nRepair PCB, Varistor: VC12063D650DP + 0.1uF.\r\nCPU SAB80C537 infenion - bad.\r\nChip: 28C64A - bad\r\nLM358 - flashed.\r\nUsed the left side of the CPU board for use: LM358 -2 and changes \r\nfor compatibility with analog modules. \r\nBoard E-AN02A4 V1/2 -2 unit - total flash:\r\nEX=TVS 39V, A82=1N4148\r\nOA: TLE2022 (TLC272 compatible).  \r\nFull test.','157','','nbb',50,36,1,11),(3402,3400,4600,NULL,NULL,NULL,21,161,7,'AS-B838032','0043529','2023-10-24','Configuration:\r\nOOOO-+00000000-+0000 OOOO-+00000000-+0000\r\n\r\n1126: CH9, CH10 - faulted.\r\nIRFD9024 -2.\r\n3400: CH13, CH10: IRFD9024 -2, P6KE20C => P6KE20A, 1N4007.','','','',9,5,1,11),(3403,3401,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00267289','2023-10-25','Capacitors + 27000u.\r\nTested.','123','','',15,4,1,11),(3404,3402,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','306832','2023-10-25','Capacitors.\r\nTested.','123','','',15,4,1,11),(3405,3403,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','153371','2023-10-25','Capacitors( 27000uF - ok).\r\nTested.','123','','',15,4,1,11),(3406,3404,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','275039','2023-10-25','No problem','','','',7,3,2,11),(3407,3405,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','310394','2023-10-25','228: Electrican Laboratory.\r\n\r\nCapacitors,\r\nrelays 24V.\r\n3405: \r\n47uF 400V => 56uF 450V, 22uF, 470uF 16V.','','','',5,2,2,11),(3408,3406,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','275040','2023-10-25','Electrican Laboratory.\r\n\r\n470uF 16V.','','','',5,2,2,11),(3409,3407,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D34-AA','1443858','2023-10-25','917: Lost data.\r\nProgramming AT28C64 (043G).\r\n3407: Programming AT28C64 (043G).','','','magnesium',8,4,1,11),(3410,3408,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D34-AA','274786','2023-10-25','918: Lost data.\r\nProgramming AT28C64 (043G).\r\n3408: Programming AT28C64 (043G).','','','magnesium',8,4,1,11),(3411,3409,20000,NULL,NULL,NULL,21,607,98,'Cerabar S PMC71-5CQ82/101','RA10300109C','2023-10-30','Making the mobile version','147','','',60,40,1,11),(3412,3410,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D34-AA','398618','2023-10-30','Lost data.\r\nProgramming the old AT28C64 (043F).','','','magnesium',8,4,1,11),(3413,3411,3800,NULL,NULL,NULL,21,322,10,'ACP-300H','2021-189','2023-11-01','PCB flashed because manufacturing quality','','','',10,3,1,11),(3414,3412,1000,NULL,NULL,NULL,21,499,12,'4.728.0051','1728001002662','2023-11-01','3139: New PS installed.\r\n3412: LM358','22','','',8,4,1,11),(3415,3413,80450,NULL,NULL,NULL,21,390,98,'FIT-99507 1700R12ABZEZZZ','3117313','2023-11-02','3341: 1uF 450V\r\nU14: HCNW-4506 are bad. Replaced every one (U13, U14).\r\nTesting.\r\n3413: C27 22uF 50V (total).','137','','',10,5,1,11),(3416,3414,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00161475','2023-11-06','Capacitors( 27000uF - ok).\r\nTested.','123','','',15,4,1,11),(3417,3415,2492,NULL,NULL,NULL,21,396,33,'Ladegerat 2.250.1005','20287','2023-11-07','3208: 100uF 50V\r\nCleaning the terminal.\r\n3415: no problem','','','',5,3,2,11),(3418,3416,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS951B0172','2023-11-12','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3419,3417,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS95125038','2023-11-12','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3420,3418,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS95180180','2023-11-12','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3421,3419,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS95180171','2023-11-13','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3422,3420,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS96091092','2023-11-13','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3423,3421,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS95180173','2023-11-15','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3424,3422,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS96091100','2023-11-15','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3425,3423,4349,NULL,NULL,NULL,21,535,24,'FBM201','P0914S0-0N-RY1025-2558','2023-11-21','Ch_8: SMD varistor (RS:464-8959)','32','','',10,4,1,11),(3426,3424,3570,NULL,NULL,NULL,21,730,3,'JWS600-24','41K-622CCW-0196W4705','2023-11-23','PFC - flashed.\r\nT4A, FA5502, S3V52 => UF5408, \r\n2SA1244, 2SC3074 => FZN652, FZN752\r\nSMD 270R -2, 2R4, 100K.\r\nSecondary PS side: T5A => T6.3A, SMD 10R 0.5W -2.\r\nTested from VFD power test station.','','','',50,25,1,11),(3427,3425,345,NULL,NULL,NULL,21,731,90,'CP4-A','3425','2023-11-26','400VAC applied. 100R - flashed.\r\nClearing.','64','','',2,1,1,11),(3428,3426,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3426','2023-11-26','400VAC applied. 100R - flashed.\r\nClearing.','64','','',2,1,1,11),(3429,3427,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3427','2023-11-26','3427: 400VAC applied. 100R - flashed.\r\nClearing.','64','','',2,1,1,11),(3430,3428,12200,NULL,NULL,NULL,213,607,54,'FTR20-1265/0','T80029KIR40','2023-11-27','Flash on the board because HV was applied to NC contact of the relay.\r\nSMD: 100k, new print installed.','','','',15,10,1,11),(3431,3429,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3429','2023-11-27','Replaced the board','64','','',2,1,1,11),(3432,3430,8750,NULL,NULL,NULL,21,272,135,'140CHS11000','','2023-11-30','Making circuit\r\nhttps://www.nexinstrument.com/140chs11000','158','','',40,20,1,11),(3433,3431,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3431','2023-11-30','100R, 22nF 630V, ZD10V 1W.','64','','',5,4,1,11),(3434,3432,7000,NULL,NULL,NULL,21,732,10,'MX-3','35745','2023-12-04','Keyboard problem, - repairing.','159','','',20,10,1,11),(3435,3433,1000,NULL,NULL,NULL,21,499,12,'4.728.0051','3433','2023-12-04','No problem','22','','',8,1,1,11),(3436,3434,1000,NULL,NULL,NULL,21,499,12,'4.728.0051','3434','2023-12-04','No problem','22','','',8,1,1,11),(3437,3435,1000,NULL,NULL,NULL,21,499,12,'4.728.0051','3435','2023-12-04','No problem','22','','',8,1,1,11),(3438,3436,7200,NULL,NULL,NULL,21,733,150,'ESC-012A','ESC-012A 11851','2023-12-05','DC to DC converter\'s:\r\n24V to 3.3V 0.5A => 5A (aliexpress)\r\n24V to 5V 0.5A => 5V 1A (Wurth Elektronik 173010578 144-7813)','','','',15,8,1,11),(3439,3437,25000,NULL,NULL,NULL,21,96,9,'591C/3600/9/1/0/0/0/00/000','16229021002129','2023-12-07','526: Corrosion on PCB \r\n(int. print on power line for supply\r\ntransformers for field SCR\'s).\r\n\r\n3437: Varistor int. from corrosion   \r\n(RS: 240-157  PRO Metal Oxide Varistor Clamping 1200V, Varistor 710V)\r\nFAN 110VAC => FAN 24VDC + PS (RS: 739-6547)\r\nLC-capacitor 100nF+100R (RS: 206-7881)\r\n\r\nPrice from ebay','','book','',10,6,1,11),(3440,3438,1200,NULL,NULL,NULL,21,51,8,'OPTA2 CM090499 PC00253G','3689761187','2023-12-10','1782: K1: 24V.\r\n3438: 24C02 + programming.','7 548#','','VACON',5,3,1,11),(3441,3439,6500,NULL,NULL,NULL,21,498,6,'FOM10','733042','2023-12-10','138 from 17.08.2014\r\nFounded problem in transmitter.\r\nFull tested.\r\n265 - cleaning all fiber optic sensors.\r\n3253: no problem founded. (changed BS170, 1N4148).\r\n3355: ch8; ch10 - was bad, cleaning.\r\n3439: Laser transmitter was bad. Changed.\r\nFull test with fiber optic.','9','','',8,4,2,11),(3442,3440,13943,NULL,NULL,NULL,21,664,101,'DSF-2010-MB P/N: 2653-271-06','506877','2023-12-12','Defaulted 4mA','116','','',2,1,1,11),(3443,3441,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','5082-95','2023-12-14','Capacitors + capacitors with very bad condition in PWM PS. \r\nCleaning.','53','','SOLCON\\MRP2000_D\\X2816.bin',15,5,1,11),(3444,3442,1200,NULL,NULL,NULL,21,326,3,'JWS50-24/A','VKL-144C83-0474W480','2023-12-14','Total because zener 220V (OVP) PFC\r\nInstead zener used 220k 0.4W.\r\n56uF 450V','88','','',10,4,9,11),(3445,3443,4800,NULL,NULL,NULL,21,653,98,'IFC300','A23059538','2023-12-19','Broken the plastic bracket. Repairing.','','','',8,4,1,11),(3446,3444,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R1','091495','2023-12-21','3444: Electrican Laboratory.\r\n\r\nRelay\r\nCapacitors','','','',5,2,2,11),(3447,3445,4100,NULL,NULL,NULL,21,734,45,'1196','3445','2023-12-24','612-6L7-IGT120.220.6000V Ignition transformer\r\n\r\nBroken central HV output pin.\r\nRepairing.','','','',10,5,1,11),(3448,3446,3570,NULL,NULL,NULL,21,434,3,'EWS600-24','CQT-822Z12-0023','2023-12-25','583: Flash secondary PS:\r\n2SK513, ZD15V, F2.5A => F3.15A, 4R3 -2.\r\nFAN repair.\r\nCleaning.\r\nFull test.\r\n837: Cooler, SENS cables.\r\n1538: Test. Ok\r\n3446: Total corrosion. Replaced: ZD15V -3, BYV26, capacitors.\r\nRegulated to 26V.','62','','',10,6,1,11),(3449,3447,1200,NULL,NULL,NULL,21,513,98,'Isotec 4(H) Terminal Assignment','98021142','2023-12-28','3447\r\nTest:\r\n5: -4...20mA\r\n6: +4...20mA\r\n\r\n1: +4...20mA\r\n2: -4...20mA\r\nNo problem','','','',4,1,5,11),(3450,3448,1200,NULL,NULL,NULL,21,513,98,'Isotec 4(H) Terminal Assignment','9701715','2023-12-28','3448\r\nTest:\r\n5: -4...20mA\r\n6: +4...20mA\r\n\r\n1: +4...20mA\r\n2: -4...20mA\r\nNo problem','','','',4,1,5,11),(3451,3449,2600,NULL,NULL,NULL,21,2,1,'SIPART A430-L1','4297553','2024-01-02','Cleaning.\r\nTest.','103','','',12,3,9,11),(3452,3450,2600,NULL,NULL,NULL,21,2,1,'SIPART A430-L1','4005748','2024-01-02','Cleaning.\r\nTest.','103','','',12,3,9,11),(3453,3451,2600,NULL,NULL,NULL,21,2,1,'SIPART A430-L1','4041196','2024-01-02','Cleaning.\r\nTest.','103','','',12,3,9,11),(3454,3452,5970,NULL,NULL,NULL,21,434,3,'EWS600-24','731Z02-0010W','2024-01-02','826:\r\nCooler.\r\n2051: cooler, capacitors.\r\n3452: voltage output regulation','62','','',10,2,2,11),(3455,3453,17980,NULL,NULL,NULL,21,612,85,'TMS-M','NM 05-70-155','2024-01-03','Regeneration of sensor.','77, 50','CMC','',22,2,1,11),(3456,3454,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D34-AA','271272','2024-01-07','940: Lost data.\r\nNew chip programming AT28C64 (043E).\r\n3454: no problem','','','magnesium',8,4,1,11),(3457,3455,3600,NULL,NULL,NULL,21,500,6,'Gavish 085','292','2024-01-08','292: Total corrosion.\r\nFull test provided.\r\n3455: Making correct configuration polarity for 1uF.\r\nCapacitors, cleaning.','6','','',9,3,1,11),(3458,3456,3600,NULL,NULL,NULL,21,500,6,'Gavish 085','3456','2024-01-08','Making correct configuration polarity for 1uF.\r\nCapacitors, cleaning.','6','','',9,3,1,11),(3459,3457,3600,NULL,NULL,NULL,21,500,6,'Gavish 085','3457','2024-01-08','Making correct configuration polarity for 1uF.\r\nCapacitors, cleaning.\r\nFuse holder','6','','',9,3,1,11),(3460,3458,3600,NULL,NULL,NULL,21,500,6,'Gavish 085','3458','2024-01-08','Making correct configuration polarity for 1uF.\r\nCapacitors, cleaning.\r\nFuse holder','6','','',9,3,1,11),(3461,3459,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515469','2024-01-09','2845: Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P\r\n3459: No problem founded.','32','','',10,3,1,11),(3462,3460,9316,NULL,NULL,NULL,21,535,52,'P0400YE FBM4 REV-0L','1894516777','2024-01-10','2406: Capacitors.\r\nProgramming the new AT28C64B rev 0L\r\n3460: Was curved output PIN\'s','32','','',15,5,5,11),(3463,3461,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA FBM 01','1896505260','2024-01-10','1571: 330uF 35V -10,\r\n1000uF 10V => 1000uF 16V - all.\r\nSoldering.\r\n\r\nFull test - ok.\r\n2172: NO PROBLEM.\r\n2476: new AT28C64B installed with updated DUMP DA-0P\r\n3165: short (non constant) between EEPROM pin\'s.\r\n3315: soldering the all IC\'s. 6264 - changed.\r\n3461: no problem.','32','','',8,3,1,11),(3464,3462,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0S FBM1','SF08347509','2024-01-10','Capacitors.\r\nProgramming the new AT28CC64 with DUMP 0S','32','','bin',10,3,1,11),(3465,3463,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA FBM 01','1895192069','2024-01-11','1662: All PS\'s capacitors changed (corrosion on PAD\'s)\r\nTest.\r\n3463: The new EEPROM programming and installing.\r\nTest.','32','','bin',20,3,1,11),(3466,3464,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192095','2024-01-14','Capacitors.\r\nProgramming the new AT28CC64 with DUMP 0L','32','','',10,3,1,11),(3467,3465,3000,NULL,NULL,NULL,21,543,125,'DFW W-indicator','DFW06XPCC/0100175370','2024-01-14','Repair the keyboard','82','','',10,5,1,11),(3468,3466,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896494888','2024-01-22','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3469,3467,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515539','2024-01-23','2842: Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P\r\n3467: LDO, 6264','32','','',10,3,1,11),(3470,3468,12000,NULL,NULL,NULL,21,189,35,'46199001','46193001','2024-01-23','2995: Capacitors.\r\nTerminal, cleaning.\r\n3468: Capacitors','532#','','bin',10,2,5,11),(3471,3469,8400,NULL,NULL,NULL,21,263,19,'DD10-48C0-0000','002007','2024-01-23','2268: 65A FUSE -2 => 63A FUSE 182-1464.\r\n3469: test only','','SCR pdf','',3,2,1,11),(3472,3470,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192153','2024-01-24','3470: Capacitors.\r\nProgramming the new AT28CC64 with DUMP 0L','32','','',10,3,1,11),(3473,3471,18498,NULL,NULL,NULL,21,15,35,'invensys MODBUS Mini8','IS15845-001-001-11-05','2024-01-24','Int. soldering of the terminal FC, CC\r\n\r\nhttps://www.ebay.com/itm/123213047482?hash=item1cb011e6ba:g:dAIAAOSwWspbMZrZ&amdata=enc%3AAQAIAAAA8KCzgZ9IVshquwBFKLAnVWSuZa8MUKVUSD3jr0VhdhsydMSrGrYUbhBuLapDNpu6bTl%2B3qhoWDbrwdJGmxad9iMfu88fgVHLKSZklmsQHyaob9mh3DM%2FN8RkkSvs7wmzQRFQ8x9KN4eCJeRcyoZFVxwzv9gBgXsUwQ1IiS258oR93%2BM%2BiBXOF46qfs60dNp9tVmL63WDZEkYY6GJ8AwnuzRbLzezin3D6O86JeYP%2B67I0Od%2FRytruXREauB57nYEgTUNdpPoi8YWR6kCTZCYGn5AIyNUCdv8C8YyUinxvvkkmtH%2BlbGhqir6rdG09aoafg%3D%3D%7Ctkp%3ABk9SR7q8n7GnYw','160','','',4,2,1,11),(3474,3472,5000,NULL,NULL,NULL,213,520,9,'BSCEN-602T','050058','2024-01-28','Int. pot. 10k. Repair.','','','',4,1,1,11),(3475,3473,1200,NULL,NULL,NULL,21,513,98,'ISO-4H FTB-33136','3473','2024-01-31','Test:\r\n5: -4...20mA\r\n6: +4...20mA\r\n\r\n1: +4...20mA\r\n2: -4...20mA\r\nNo problem','','','',4,1,5,11),(3476,3474,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','259105','2024-02-01','Electrican Laboratory.\r\n\r\nRelay\r\nCapacitors','','','',5,2,2,11),(3477,3475,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D34-AA','258624','2024-02-01','Lost data.\r\nNew chip programming AT28C64 (043E).','','','magnesium',8,2,1,11),(3478,3476,1500,NULL,NULL,NULL,21,2,3,'SITOP Power 5 6EP1333-1AL11','Q6MD370142','2024-02-04','All capacitors are bad.\r\nCorrosion.\r\nTest.','','','',10,5,1,11),(3479,3477,4600,NULL,NULL,NULL,21,161,7,'AS-B824-016','31033739914','2024-02-04','Configuration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /+24V 0V/  8 7 6 5 4 3 2 1\r\n\r\nTest','','','',9,1,1,11),(3480,3478,4600,NULL,NULL,NULL,21,161,7,'AS-B824-016','31980462543','2024-02-04','Corrosion.\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /+24V 0V/  8 7 6 5 4 3 2 1\r\n\r\nTest','','','',9,2,1,11),(3481,3479,13943,NULL,NULL,NULL,21,664,101,'DSF-2010-MB P/N: 2653-271-06','506442','2024-02-04','2942: \r\nDefaulted \r\n\r\nTested with PC, - ok\r\nPassword 123456\r\n3479: No problem (4-20mA) founded.\r\nDefaulted','116','','',2,1,1,11),(3482,3480,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896464428','2024-02-04','Capacitors.\r\nCh_5: 49R9, 15uF 35V.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3483,3481,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1895232585','2024-02-05','3481: Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3484,3482,2000,NULL,NULL,NULL,21,469,76,'PAXS0010','46035','2024-02-05','Cleaning.\r\nWire pin','40','','RedLion',8,3,5,11),(3485,3483,6700,NULL,NULL,NULL,21,524,104,'P3U 8317275','ARSC-0022','2024-02-05','Cleaning\r\nNo problem.','151','','',5,2,1,11),(3486,3484,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896464435','2024-02-06','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3487,3485,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896464481','2024-02-07','Programming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3488,3486,1500,NULL,NULL,NULL,21,735,3,'AT1621-20','12189','2024-02-08','Test internal supply only (30.5V).','','','',3,2,5,11),(3489,3487,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192156','2024-02-08','Capacitors. Corrosion!\r\nProgramming the new AT28CC64 with DUMP 0L','32','','',10,3,1,11),(3490,3488,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896494995','2024-02-11','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3491,3489,80450,NULL,NULL,NULL,21,390,98,'FIT-99507 1700R12ABAEZZZ','14144708','2024-02-11','1uF 450V (was good)\r\nU14: HCNW-4506 are bad. Replaced every one (U13, U14).\r\nTesting.\r\nC27 22uF 50V (was good).','137','','',10,5,1,11),(3492,3490,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896464490','2024-02-11','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3493,3491,2000,NULL,NULL,NULL,213,736,86,'SPS 800 SOHO+','232016C01367','2024-02-14','Capacitors','','','',3,2,1,11),(3494,3492,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D29-AB','273059','2024-02-15','Test only\r\nEPROM\'s data is correct (056F).','','','magnesium',8,2,1,11),(3495,3493,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D29-AB','278890','2024-02-15','Test only\r\nEPROM\'s data is correct (056F).','','','magnesium',8,2,1,11),(3496,3494,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D29-AB','304627','2024-02-15','Test only\r\nEPROM\'s data is correct (056F).','','','magnesium',8,2,1,11),(3497,3495,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D29-AB','278900','2024-02-15','Test only\r\nEPROM\'s data is correct (056F).','','','magnesium',8,2,1,11),(3498,3496,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192091','2024-02-25','Capacitors.\r\nProgramming the new AT28CC64 with DUMP 0L','32','','',10,3,1,11),(3499,3497,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896464409','2024-02-25','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3500,3498,2600,NULL,NULL,NULL,21,2,1,'SIPART C73451-A430-P300-08','015-025','2024-02-29','Cleaning.\r\nTest.','103','','',12,3,9,11),(3501,3499,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192168','2024-03-03','Capacitors. Corrosion!\r\nProgramming the new AT28CC64 with DUMP 0L','32','','',10,3,1,11),(3502,3500,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192143','2024-03-03','Capacitors. \r\nProgramming the new AT28CC64 with DUMP 0L','32','','',10,3,1,11),(3503,3501,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192125','2024-03-03','Capacitors. Corrosion!\r\nProgramming the new AT28CC64 with DUMP 0L','32','','',10,3,1,11),(3504,3502,1500,NULL,NULL,NULL,21,599,8,'MSR 020','M2-U3','2024-03-03','2921: 100uF 50V, 330uF 63V => 330uF 50V 125C\r\nRelay 24V -2.\r\nTr-r: 230V/24V/3W => 2.8W 24V (732-0395).\r\n3284: 2.8W 24V (732-0395).\r\n3502: 2.8W 24V (732-0395).','','','',5,4,1,11),(3505,3503,9316,NULL,NULL,NULL,21,535,52,'P0400YE-0P FBM 04','1896506226','2024-03-03','2618: Capacitors\r\nBad EEPROM Programming rev 0P\r\n15uF 25V, Transformer flashed.','32','','bin',20,3,1,11),(3506,3504,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896464506','2024-03-04','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3507,3505,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896464450','2024-03-04','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P','32','','',10,3,1,11),(3508,3506,9316,NULL,NULL,NULL,21,535,52,'I/A Series P0400YE-0N FBM4','1893032870','2024-03-04','Capacitors. Corrosion!\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400YE-0N','32','','',10,3,1,11),(3509,3507,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM 01','1895049585','2024-03-04','2452: Capacitors\r\nProgramming the new EEPROM by 0P revision\r\nStay problem with accuracy of ch_5...ch_8\r\nReplacing 74HC240, SRAM\r\n3507: tested','32','','bin',10,3,1,11),(3510,3508,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1895232585','2024-03-04','3481: Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P\r\n3508: Correction: \r\nch_5: 65301 => 2k4\r\nch_6: 65147 => 2k73','32','','',10,3,1,11),(3511,3509,40000,NULL,NULL,NULL,21,737,35,'FIZEPR-SW100','22996','2024-03-05','Test - ok.','','','',10,5,1,11),(3512,3510,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS95180159','2024-03-05','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3513,3511,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS95180124','2024-03-05','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3514,3512,2000,NULL,NULL,NULL,213,533,7,'140SD095300S','21091701439','2024-03-06','Incorrect polarity applied \r\nTVS GFK76A SMD to TVS36V\r\n\r\n×ž×§×˜: 973800672  ×¢×œ×•×ª 548 $','','','',10,3,1,11),(3515,3513,2000,NULL,NULL,NULL,213,533,7,'140SD095300S','21091701445','2024-03-06','Incorrect polarity applied \r\nTVS GFK76A SMD to TVS36V\r\n\r\n×ž×§×˜: 973800672  ×¢×œ×•×ª 548 $','','','',10,3,1,11),(3516,3514,12500,NULL,NULL,NULL,21,580,3,'036A P/N1009701000','0000000333','2024-03-06','539: All capacitors changed \r\n(Original large capacitors 1500uF 50V)\r\nC13: 1nF 1000V (Was leakege 15k)\r\n\r\n2191: F5A -2 (Original F6.3A), PFC: IRFP460,\r\nR243, R244: 18R 2W\r\nR47, R48: 0.22R 2W\r\nAll IC\'s and optocouplers on PCB module instead UC3844\r\nCooler 12V Current must be 100 - 140mA (original 12V 140mA) \r\nOptional to use 24V Cooler\r\nCleaned and tested with loading 24V - lamps.\r\n3514: IRFP460 -1. Cooler 12V - repair.  18R 2W => 39R II 33R -2 sets.\r\n0.22R -2 => 0.2R -2.','119','','',40,30,1,11),(3517,3515,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS95180146','2024-03-07','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3518,3516,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS95180174','2024-03-07','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3519,3517,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS95180150','2024-03-10','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3520,3518,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS96091098','2024-03-10','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3521,3519,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS95180177','2024-03-10','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3522,3520,2500,NULL,NULL,NULL,21,505,109,'AVO_Tester','3520','2024-03-11','Make the new one. \r\nProject.','161','','',50,40,5,11),(3523,3521,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS95180148','2024-03-12','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3524,3522,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS96091099','2024-03-12','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3525,3523,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS96091097','2024-03-12','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3526,3524,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS96091106','2024-03-12','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3527,3525,4700,NULL,NULL,NULL,21,303,3,'AA16560 I/A Series REV B','AS96091096','2024-03-12','Foxboro PS\r\nCapacitors.','120','','',12,6,1,11),(3528,3526,4700,NULL,NULL,NULL,21,535,3,'CM902WX','SP95160149','2024-03-12','Foxboro PS\r\nCapacitor 330uF 50V.\r\nSet battery bad condition and not used from manufacture.','120','','',12,6,1,11),(3529,3527,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D34-AA','111326','2024-03-13','531: 043G\r\nLost data 2864.\r\nProgramming.\r\n3527: Lost data 2864.\r\nProgramming.','','','magnesium',8,2,1,11),(3530,3528,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','274003','2024-03-13','Electrican Laboratory.\r\n\r\nRelay\r\nCapacitors','','','',5,2,2,11),(3531,3529,4700,NULL,NULL,NULL,21,535,3,'CM902WX','SP95160161','2024-03-13','Foxboro PS\r\nCapacitor 330uF 50V.\r\nSet battery bad condition and not used from manufacture.','120','','',12,6,1,11),(3532,3530,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D29-AB','278895','2024-03-17','2014: EEPROM X28HC64 PLCC (056F).\r\n3530: Test ok.','','','magnesium',8,4,1,11),(3533,3531,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D29-AB','','2024-03-17','573: Lost data.\r\nReplacing and programming AT28HC64B (PLCC 44).\r\n3531: Test ok.','','','magnesium',10,3,1,11),(3534,3532,9316,NULL,NULL,NULL,21,535,52,'I/A Series P0400YE-0L FBM4','1895251540','2024-03-17','Capacitors. Corrosion!\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400YE-0L','32','','',10,3,1,11),(3535,3533,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D34-AA','281474','2024-03-18','2494: EEPROM Programming AT28HC64  (043G).\r\n3533: test ok','','','magnesium',8,4,1,11),(3536,3534,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D34-AA','271909','2024-03-18','(043E).\r\ntest ok','','','magnesium',8,4,1,11),(3537,3535,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D34-AA','1004661','2024-03-18','test ok','','','magnesium',8,4,1,11),(3538,3536,4500,NULL,NULL,NULL,21,118,1,'SPCJ 4D34-AA','281476','2024-03-18','2442: EEPROM AT28HC64 PLCC (043E).\r\n3536: test','','','magnesium',8,4,1,11),(3539,3537,18000,NULL,NULL,NULL,21,51,13,'NXS00725A2H0SSVA1A2000000','10104316','2024-03-18','2483: Present from Yam Hamelah\r\nOperation Panel changed only.\r\nTested.\r\n3537: (Saturation trip). 2u2 -4, relay -3.\r\nTest','7 548#','','',20,10,2,11),(3540,3538,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','383969','2024-03-20','Electrican Laboratory.\r\n\r\nRelay\r\nCapacitors','','','',5,2,2,11),(3541,3539,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R2','259778','2024-03-20','Electrican Laboratory.\r\n\r\nRelay\r\nCapacitors','','','',5,2,2,11),(3542,3540,3570,NULL,NULL,NULL,21,434,3,'EWS300-24','708Z23-0015W','2024-03-25','2212: Flash on secondary PS: replacing the another one.\r\nFAN repair.\r\nCleaning.\r\nFull test.\r\n3540: Flash on PCB. \r\nF2.5A => F3.15A, connector? cleaning.','62','','',20,10,2,11),(3543,3541,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00267264','2024-03-26','Capacitors.\r\nTested.','123','','',15,4,1,11),(3544,3542,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00241634','2024-03-26','Capacitors.\r\nTested.','123','','',15,4,1,11),(3545,3543,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','00261083','2024-03-26','Capacitors.\r\nTested.','123','','',15,4,1,11),(3546,3544,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0K FBM1','1892176047','2024-04-01','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0K','32','','',10,3,1,11),(3547,3545,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895242686','2024-04-01','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0K','32','','',10,3,1,11),(3548,3546,5000,NULL,NULL,NULL,21,505,109,'HV_Tester','3546','2024-04-03','Make the new one. \r\nProject.','161','','',100,50,5,11),(3549,3547,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R1','091495','2024-04-04','3444: Electrican Laboratory.\r\n\r\nRelay\r\nCapacitors\r\n3547: test','','','',5,2,2,11),(3550,3548,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895242725','2024-04-07','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0K','32','','',10,3,1,11),(3551,3549,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0R FBM 1','1898131029','2024-04-07','Capacitors\r\nEEPROM: NQ28C64A => AT28C64B\r\nProgramming rev 0R','32','','bin',10,3,1,11),(3552,3550,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0R FBM 1','1898131009','2024-04-07','Capacitors\r\nEEPROM: NQ28C64A => AT28C64B\r\nProgramming rev 0R','32','','bin',10,3,1,11),(3553,3551,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','4297586','2024-04-09','3094: Cleaning.\r\n3551: Cleaning','103','','',5,2,2,11),(3554,3552,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','4016344','2024-04-09','2069: Test. ok\r\n3552: cleaning.','103','','',12,3,1,11),(3555,3553,9316,NULL,NULL,NULL,21,535,24,'P0400DA-0R FBM 1','1898131040','2024-04-10','Capacitors\r\nEEPROM: NQ28C64A => AT28C64Bing rev 0R','32','','bin',10,3,1,11),(3556,3554,9316,NULL,NULL,NULL,21,535,52,'I/A Series P0400YE-0L FBM4','1895251527','2024-04-10','Capacitors. Corrosion!\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400YE-0L','32','','',10,3,1,11),(3557,3555,9316,NULL,NULL,NULL,21,535,52,'I/A Series P0400YE-0L FBM4','1895251557','2024-04-11','Capacitors. Corrosion!\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400YE-0L','32','','',10,3,1,11),(3558,3556,400,NULL,NULL,NULL,21,738,3,'TRG10R150','3556','2024-04-11','Plug replacing.','','','',3,2,1,11),(3559,3557,400,NULL,NULL,NULL,21,738,3,'TRG10R150-01E03 VI','3557','2024-04-14','Plug replacing.','','','',3,2,1,11),(3560,3558,400,NULL,NULL,NULL,21,738,3,'TRG10R150-01E03 level V','3558','2024-04-14','Plug replacing.','','','',3,2,1,11),(3561,3559,700,NULL,NULL,NULL,21,739,3,'PSU30A-3','446117','2024-04-15','Flashed.\r\nCapacitors, T2A\r\nTOP225 => TOP204','','','',5,3,5,11),(3562,3560,1200,NULL,NULL,NULL,21,117,3,'C6409-60014','T5002054010','2024-04-15','Plug installed.','','','',2,1,5,11),(3563,3561,9316,NULL,NULL,NULL,21,535,52,'I/A Series P0400YE-0L FBM4','1895251550','2024-04-16','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400YE-0L','32','','',10,3,1,11),(3564,3562,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','6081-96','2024-04-17','2671: Capacitors,\r\nrelays 12V -6.\r\n3562: LCD replaced: PVC160206QGL01 => 1620BCBC DISPLAYTECH 162D\r\nRS: 532-6486\r\nRL1 68R => 13R (LED backlight) pluse => minus','53','','SOLCON\\MRP2000_D\\X2816.bin',10,3,1,11),(3565,3563,9316,NULL,NULL,NULL,21,535,52,'I/A Series P0400YE-0L FBM4','1895251556','2024-04-18','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400YE-0L','32','','',10,3,1,11),(3566,3564,80450,NULL,NULL,NULL,21,390,98,'1700R12ABAEZZZ','3031406','2024-05-01','3564: 1uF 450V - was bad.\r\nU14: HCNW-4506 replaced not tested\r\nC27 22uF 50V (was good).\r\nNot tested','137','','',10,5,1,11),(3567,3565,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','5177-95','2024-05-02','3099: Capacitors, \r\nRelay 12V -6.\r\nSocket CPU.\r\n3565: incorrect current meter (L1)\r\nU43: LM324 (replaced all)','53','','SOLCON\\MRP2000_D\\X2816.bin',20,16,1,11),(3568,3566,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','1179969','2024-05-05','1842: Cleaning.\r\nTest.\r\n1871: test only\r\n2252: cleaning.\r\n3566: cleaning.','103','','',12,3,9,11),(3569,3567,7600,NULL,NULL,NULL,21,161,17,'AS-H827-100','0045217','2024-05-05','D-SUB connector 37 pin changed.\r\nCleaning.','','','',4,3,9,11),(3570,3568,10497,NULL,NULL,NULL,21,329,19,'TPS 820 400-230-230-5-8-D-S','WO18149429003','2024-05-12','Over temperature (orange wire & orange wire must be shorting \r\nduring the temperature is normal for system)\r\nResetting the alarm:\r\npressing on select/reset knobe.','','','',4,4,1,11),(3571,3569,10497,NULL,NULL,NULL,21,329,19,'TPS 820 400-230-230-5-8-D-S','WO1814929004','2024-05-12','3205: SCRC471000 -2 => ESTEL T353-1000-20-72 -2.\r\n1N4007.\r\nFAN -3 (capacitor 1uF 400V)\r\nFull test 4-20mA 0-400VAC\r\nUsed 2 phases.\r\nPrice:\r\nhttps://www.radwell.co.uk/en-GB/Buy/SOLCON/SOLCON/TPS-820-600-115-115/?redirect=true\r\n3569: Over temperature (orange wire & orange wire must be shorting \r\nduring the temperature is normal for system)\r\nResetting the alarm:\r\npressing on select/reset knobe.','','','',40,20,1,11),(3572,3570,10497,NULL,NULL,NULL,21,329,19,'TPS 820 400-230-230-5-8-D-S','WO18149429002','2024-05-12','Over temperature (orange wire & orange wire must be shorting \r\nduring the temperature is normal for system)\r\nResetting the alarm:\r\npressing on select/reset knobe.','','','',4,4,1,11),(3573,3571,8300,NULL,NULL,NULL,21,2,11,'6AV6647-0AB11-3AX0','ZVC2YUD022114','2024-05-12','Corrosion on touch cable corrosion.','','','',8,3,1,11),(3574,3572,5200,NULL,NULL,NULL,21,498,24,'AFM-16','2725906','2024-05-15','Capacitors\r\nOutput_2 (OPF 694-2) -bad\r\nTested.','9','','',10,5,1,11),(3575,3573,900,NULL,NULL,NULL,21,504,128,'WAS5 CCC LP 844950000','O6N40','2024-05-20','No problem\r\nTested.','','','',2,1,1,11),(3576,3574,870,NULL,NULL,NULL,21,518,11,'305','A203722','2024-05-20','Corrosion.\r\n4mA => 0%\r\n20mA => 200%','pdf','','',5,3,5,11),(3577,3575,870,NULL,NULL,NULL,21,518,11,'201','A182197','2024-05-20','3044: Cleaning\r\n4mA => 0\r\n20mA => 900\r\n3575: Cleaning\r\n4mA => 0\r\n20mA => 800','pdf','','bin',6,4,2,11),(3578,3576,870,NULL,NULL,NULL,21,518,11,'201','A230617','2024-05-20','Test only\r\n4mA => 0\r\n20mA => 900','pdf','','bin',3,1,2,11),(3579,3577,870,NULL,NULL,NULL,21,518,11,'201','A202099','2024-05-20','Cleaning\r\nCalibration: \r\n4mA => 0\r\n20mA => 100','pdf','','bin',8,5,2,11),(3580,3578,870,NULL,NULL,NULL,21,518,11,'201','A230622','2024-05-20','Cleaning\r\nThe new wireup and tactile switches are installed. \r\nCalibration: \r\n4mA => 0\r\n20mA => 100','pdf','','bin',8,5,2,11),(3581,3579,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','4016344','2024-05-21','2069: Test. ok\r\n3552: cleaning.\r\n3579: indication problem. Cleaning.','103','','',12,3,1,11),(3582,3580,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','4223239','2024-05-21','Cleaning.\r\nProgramming (parameter 7)','103','','',12,3,1,11),(3583,3581,450,NULL,NULL,NULL,21,740,6,'PC-27','3581','2024-05-21','The new terminal installed.','','','',1,1,5,11),(3584,3582,2700,NULL,NULL,NULL,21,498,98,'ACT-2','625163','2024-05-23','The devise marked by 5A but internal configuration\r\nresistor Rd1 is 402R - for 1A.\r\n\r\nLM336-2.5, CA3130E, Z1 36V, D3 1N4148, LM334Z, R6 100R\r\nCorrosion, repairing the terminal. \r\nMaking full circuit.\r\n\r\nCalibration:\r\n0...1A\r\n4...20mA','136','','',20,10,1,11),(3585,3583,2700,NULL,NULL,NULL,21,498,98,'ACT-2','630673','2024-05-23','The devise marked by 5A but internal configuration\r\nresistor Rd1 is 402R - for 1A.\r\n\r\nCorrosion, repairing the terminal. \r\nMaking full circuit.\r\n\r\nCalibration:\r\n0...1A\r\n4...20mA','136','','',20,10,1,11),(3586,3584,1200,NULL,NULL,NULL,21,697,3,'HDR-60-24','EB93C52357','2024-05-23','MOSFET shorted because mosquito on the pins. \r\n13A 600V => 19A 950V','','','',5,1,1,11),(3587,3585,345,NULL,NULL,NULL,21,731,90,'70222HI2','f11','2024-05-27','Making circuit.\r\n\r\nFor setting the longer (up to 6H) time replace\r\nthe J2 to upper position.','162','','',20,10,1,11),(3588,3586,5200,NULL,NULL,NULL,21,498,24,'AFM-16','630535','2024-05-27','Capacitors\r\nFlashed the board IO. LM324, LM358, BSS250.\r\nFuses, ZD30V','9','','',40,15,1,11),(3589,3587,2000,NULL,NULL,NULL,21,498,43,'TCI-2','608746','2024-05-28','2587: Capacitors\r\nCalibration:\r\n0 grad = 4mA\r\n750 grad = 20 mA\r\n3587: test ok.','9','','',4,3,5,11),(3590,3588,3200,NULL,NULL,NULL,21,498,43,'DCI-2','608869','2024-06-03','Capacitors.\r\n\r\n0V => 4mA\r\n6V => 20mA','9','','',10,4,1,11),(3591,3589,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','502159','2024-06-03','Cleaning.\r\nTest.','103','','',12,3,9,11),(3592,3590,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049744498','2024-06-04','Varistor: RMS275V => RMS250V\r\nT2A, C10: 220uF 25V => 220uF 35V, 1000uF 25V => 35V.\r\nTest.','','','',3,1,1,11),(3593,3591,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049704291','2024-06-04','3591: C10: 220uF 25V => 220uF 35V, 1000uF 25V => 35V.\r\nTest.','','','',3,1,1,11),(3594,3592,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049315899','2024-06-04','3592: Varistor: RMS275V => RMS250V\r\nT2A, C10: 220uF 25V => 220uF 35V, CNY17-3.\r\nSoldering, cleaning.\r\nTest.','','','',6,3,1,11),(3595,3593,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1896241540','2024-06-05','CH_8: flashed 49R9\r\n\r\nCapacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0K','32','','',10,3,1,11),(3596,3594,7600,NULL,NULL,NULL,21,161,17,'AS-H819-209','0028104','2024-06-05','Soldering the D-Type connector.  \r\nCleaning.','','','',4,3,9,11),(3597,3595,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','0024722','2024-06-06','1632: Neon lamp -5.\r\n3595: 22M -16.','8','','',7,4,2,11),(3598,3596,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','0025244','2024-06-06','1756: Test.\r\n3596: 22M -16.','8','','',7,4,2,11),(3599,3597,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','0024743','2024-06-06','Neon lamp -14.\r\n22M -16.','8','','',7,4,2,11),(3600,3598,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','00225257','2024-06-06','Neon lamp -6.\r\n22M -16.','8','','',7,4,2,11),(3601,3599,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','20513223','2024-06-06','Tested','','','',4,3,1,11),(3602,3600,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','0039064','2024-06-06','Ch3, Ch5: relays.','','','',4,3,1,11),(3603,3601,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240','6496792','2024-06-09','FAN flow corrosion.\r\nInt. print on power board between 51k and 51k SMD.\r\n470k CPU side.\r\nUsed the new silicone paste for PFC and output voltage side PS\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','bin',30,16,1,11),(3604,3602,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049654499','2024-06-09','3602: Varistor: RMS275V => RMS250V\r\nT2A, C10: 220uF 25V => 220uF 35V, CNY17-3.\r\nTest.','','','',6,3,1,11),(3605,3603,7600,NULL,NULL,NULL,21,161,17,'AS-9534-002','0007499','2024-06-09','Cleaning.','','','',4,3,9,11),(3606,3604,14600,NULL,NULL,NULL,21,161,67,'ASP890300','31062530085','2024-06-10','4700uF 50V => 4700uF 40V\r\nTested.','','','',10,4,5,11),(3607,3605,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3605','2024-06-10','100R, ZD10V 1W.','64','','',5,4,1,11),(3608,3606,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3606','2024-06-10','100R, ZD10V 1W, set of the switches.','64','','',5,4,1,11),(3609,3607,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3607','2024-06-10','3607: 100R, ZD10V 1W.','64','','',5,4,1,11),(3610,3608,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3608','2024-06-10','100R, ZD10V 1W, 22nF 630V','64','','',5,4,1,11),(3611,3609,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3609','2024-06-10','3609: 100R, ZD10V 1W, 22nF 630V','64','','',5,4,1,11),(3612,3610,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3610','2024-06-10','3610: 100R, ZD10V 1W.','64','','',5,4,1,11),(3613,3611,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3611','2024-06-10','Board flashed, - replaced.','64','','',5,4,1,11),(3614,3612,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3612','2024-06-10','100R, ZD10V 1W, 22nF 630V','64','','',5,4,1,11),(3615,3613,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3613','2024-06-10','3613: 100R, ZD10V 1W, 22nF 630V','64','','',5,4,1,11),(3616,3614,2750,NULL,NULL,NULL,21,741,35,'ST140','S6114LH747 0711','2024-06-10','Chemical laboratory.\r\nPS: capacitors','','','',5,2,1,11),(3617,3615,1600,NULL,NULL,NULL,21,543,76,'DINI ARGEO Srl','DGT1IO/100965336','2024-06-17','Switches -5.','41','','',4,2,1,11),(3618,3616,4500,NULL,NULL,NULL,21,366,13,'ATV16U54N4','8S9610005820 03847820311A11','2024-06-18','Driver: Lost phase: 180k 1W (changed all)\r\nCapacitors. Relay 12V 16A.\r\nOperation panel: 22uF -2.','','','',20,10,9,11),(3619,3617,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','0035138','2024-06-18','ok.','','','',4,3,1,11),(3620,3618,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','0046862','2024-06-18','ok.','','','',4,3,1,11),(3621,3619,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','0027169','2024-06-18','Relay 5 units.','','','',4,3,1,11),(3622,3620,5682,NULL,NULL,NULL,21,535,52,'FBM204','P0914SY 0N RY1024 1240','2024-06-20','Total flash.\r\nInput transformer, IO-transformer are flashed.\r\nVaristor 30V -4.\r\nT2A -2.','45, 32','','',10,6,1,11),(3623,3621,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3605','2024-06-27','3605: 100R, ZD10V 1W.\r\n3621: 100R','64','','',5,4,1,11),(3624,3622,2400,NULL,NULL,NULL,21,2,99,'N1WN04AD1800101','087398','2024-06-27','3370: Corrosion.\r\n3622: Corrosion.\r\nProgramming: parameter 7; 38 => instead rise => fall','103','','',3,2,2,11),(3625,3623,2400,NULL,NULL,NULL,21,2,99,'N1WN04AD1800101','4063074','2024-06-27','3369: Corrosion.\r\n3623: Corrosion.','103','','',3,2,2,11),(3626,3624,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','4297586','2024-06-27','3094: Cleaning.\r\n3551: Cleaning\r\n3624: Leakage between few PIN\'s CPU.\r\nCleaning.','103','','',5,2,2,11),(3627,3625,20000,NULL,NULL,NULL,21,505,35,'V300EX','3625','2024-06-27','Making the new controller board for spark registration.\r\nUsed integrated method. Passible time regulation from 0.04s - 19s.','163','','',100,70,5,11),(3628,3626,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','087408','2024-06-27','Corrosion on touch PIN\'s\r\nCleaning.','103','','',5,2,2,11),(3629,3627,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','31983000368','2024-06-30','10M -1 => ch_16\r\n\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1\r\n        \r\nTested.','','','',9,5,1,11),(3630,3628,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','F008200','2024-07-01','Neon lamp -12.\r\n22M -8.\r\n470k -1.','8','','',7,4,2,11),(3631,3629,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','F004783','2024-07-01','Neon lamp -8.\r\n22M -16.','8','','',7,4,2,11),(3632,3630,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','F004832','2024-07-01','Neon lamp -10.\r\n22M -16.','8','','',7,4,2,11),(3633,3631,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','F004792','2024-07-01','Neon lamp -7.\r\n22M -16.','8','','',7,4,2,11),(3634,3632,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','F004141','2024-07-01','Neon lamp -5.\r\n22M -16.','8','','',7,4,2,11),(3635,3633,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','31984305355','2024-07-01','ch15 => flashed 1k\r\n\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1\r\n        \r\nTested.','','','',9,5,1,11),(3636,3634,7500,NULL,NULL,NULL,21,742,22,'AS-B819-232','11467','2024-07-01','1k2 ch_30','','','',4,2,5,11),(3637,3635,7500,NULL,NULL,NULL,21,742,22,'AS-B819-232','11465','2024-07-01','no problem','','','',4,2,5,11),(3638,3636,7500,NULL,NULL,NULL,21,161,7,'AS-B808-016','F003280','2024-07-02','Ch_11: 0.1uF 500V, 1k\r\n7 neon lamp','','','',8,4,5,11),(3639,3637,7500,NULL,NULL,NULL,21,161,7,'AS-B808-016','F006811','2024-07-02','5 neon lamp\r\nCorrosion','','','',8,2,5,11),(3640,3638,7500,NULL,NULL,NULL,21,161,7,'AS-B808-016','0011365','2024-07-03','13 neon lamp\r\nF8A -2.','','','',8,4,5,11),(3641,3639,7600,NULL,NULL,NULL,21,161,17,'AS-H827-100','0048285','2024-07-03','Connector two side changed.\r\nCleaning.','','','',4,3,9,11),(3642,3640,45000,NULL,NULL,NULL,21,2,11,'SIMATIC IPC277E','VPK5950704 2018','2024-07-04','3640: Over heating CPU\r\nInserting the new thermal pad.\r\nCleaning','','','',10,4,1,11),(3643,3641,12000,NULL,NULL,NULL,21,329,19,'RVS-DN','66295','2024-07-07','Flashed PS.\r\n10nF 3kV, 220uF 16V.\r\nCleaning.','156','','',7,3,1,11),(3644,3642,9316,NULL,NULL,NULL,21,535,52,'P0400YE-0P FBM 04','1896506238','2024-07-10','2629: Capacitors\r\nBad EEPROM Programming rev 0P\r\n3642: lost DATA from EEPROM. Programming YE-OP.\r\nChanged: LDO, 6264.','32','','bin',20,5,1,11),(3645,3643,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA FBM 01','1896505260','2024-07-10','1571: 330uF 35V -10,\r\n1000uF 10V => 1000uF 16V - all.\r\nSoldering.\r\n\r\nFull test - ok.\r\n2172: NO PROBLEM.\r\n2476: new AT28C64B installed with updated DUMP DA-0P\r\n3165: short (non constant) between EEPROM pin\'s.\r\n3315: soldering the all IC\'s. 6264 - changed.\r\n3461: no problem.\r\n3643: no problem, changed LDO.','32','','',8,3,1,11),(3646,3644,80450,NULL,NULL,NULL,21,390,98,'2700R12BBAEZZZ','3269520','2024-07-14','Problem in internal PS.\r\nOver heating TVS because bad optocoupler HCPL0501 (764-4457)\r\n1uF 450V, 330uH (PS), was not good condition, - replaced.\r\nHCNW4506 -2 - replaced without testing.\r\nRepair the LCD panel\r\nTest.','137','','',15,8,1,11),(3647,3645,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049704291','2024-07-14','3591: C10: 220uF 25V => 220uF 35V, 1000uF 25V => 35V.\r\nTest.\r\n3645: Varistor: RMS275V, T2A\r\nTest.','','','',3,1,1,11),(3648,3646,2600,NULL,NULL,NULL,21,498,29,'PS-4','649907','2024-07-16','MIC4427 => TC4427\r\n22uF 50V, 4.7uF 25V.','','','',5,6,1,11),(3649,3647,2600,NULL,NULL,NULL,21,498,29,'PS-4','716752','2024-07-16','Capacitors ok, - changed.\r\nCleaning the pin\'s of the red led.','','','',5,4,1,11),(3650,3648,2600,NULL,NULL,NULL,21,498,29,'PS-4','716751','2024-07-16','Capacitors ok.\r\nTesting.','','','',5,4,1,11),(3651,3649,2600,NULL,NULL,NULL,21,498,29,'PS-3','715711','2024-07-16','95: C26: 22uF 50V.\r\nTested.\r\n3649: test.','','','',12,4,1,11),(3652,3650,2600,NULL,NULL,NULL,21,498,29,'PS-3','716749','2024-07-16','475: Test.\r\n3650: testing.','','','',12,2,1,11),(3653,3651,2600,NULL,NULL,NULL,21,498,29,'PS-3','648899','2024-07-16','129: C26: 22uF 50V.\r\nTested.\r\n3651: testing.','','','',12,1,1,11),(3654,3652,2600,NULL,NULL,NULL,21,498,29,'PS-3','715710','2024-07-16','3256: C26: 22uF 50V.\r\n4u7\r\nScrews\r\nTest.\r\n3652: testing.','','','',5,2,5,11),(3655,3653,2600,NULL,NULL,NULL,21,498,29,'PS-2','715706','2024-07-16','161:\r\nTested.\r\nNo problem founded.\r\n552: 22uF 50V.\r\n3653: testing','','','',5,3,1,11),(3656,3654,2600,NULL,NULL,NULL,21,498,29,'PS-2','638162','2024-07-16','533: 22uF 50V - all.\r\n3654: testing','','','',5,3,1,11),(3657,3655,2600,NULL,NULL,NULL,21,498,29,'PS-2','638162','2024-07-16','553: 22uF 50V - all.\r\n3655: testing.','','','',5,3,1,11),(3658,3656,2600,NULL,NULL,NULL,21,498,29,'PS-3','644698','2024-07-16','94: C26: 22uF 50V.\r\nTested.\r\n3656: testing.','','','',12,2,1,11),(3659,3657,2600,NULL,NULL,NULL,21,498,29,'PS-3','643651','2024-07-16','2008: C26: 22uF 50V.\r\nFuse holder + fuse F10A\r\nScrews\r\nTest\r\n3657: testing.','','','',5,2,5,11),(3660,3658,30000,NULL,NULL,NULL,21,560,6,'HART 375 Field Communicator','12072847','2024-07-18','390:  Touch screen AMT98234 replaced.\r\n1016: Braked DC plug (RS: 526-0825).\r\n1117: Making new battery block 1.2/1500mA*5 NiMH\r\n1277: Test only.\r\n1540: system card error.Bad placing becase shock during after fall.\r\n1576: recovery system provided. Flashing OS from internal dump.\r\nMaking the new cable.\r\n1628: Installed the new power socket. Soldering the MB, connectors\r\nSMD components. Re-Flashing EEPROM.\r\n1967: total fault. Changed battery pack without battery, \r\n- only board with closed module.\r\n2829: Very low battery level. PS-plug changed. \r\nCleaning and charging. \r\nTesting.\r\n3198: Falling unit. The CPU module was outside from the place.\r\n3658: Not working. Cleaning all board.','60','','',20,10,1,11),(3661,3659,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARRN-1295','2024-07-18','cleaning.\r\nno problem founded.\r\nTest.','144','','',6,3,1,11),(3662,3660,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARMH-0006','2024-07-18','2187: Broken PCB LCD plug.\r\nReplacing the board.\r\n3660: no problem founded.\r\nReplaced the board: LP SIOS P3 8317112-05\r\nTested','144','','',6,3,1,11),(3663,3661,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','2982','2024-07-30','2982: Capacitors\r\n1M\r\nVaristor 275V, T2A (original) changed to F2A.','','','',9,5,1,11),(3664,3662,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049315899','2024-07-30','3592: Varistor: RMS275V => RMS250V\r\nT2A, C10: 220uF 25V => 220uF 35V, CNY17-3.\r\nSoldering, cleaning.\r\nTest.\r\n3662: Varistor 275V, T2A (original) changed to F2A','','','',6,3,1,11),(3665,3663,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049975295','2024-07-30','3663: Varistor 275V between earth and fuse line,\r\nT2A (original was ok) changed to F2A.','','','',6,3,1,11),(3666,3664,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','5183-95','2024-07-30','Capacitors, relays 12V -6.\r\nLCD replaced: PVC160206QGL01 => 1620BCBC DISPLAYTECH 162D\r\nRS: 532-6486\r\nRL1 68R => 13R (LED backlight) pluse => minus','53','','SOLCON\\MRP2000_D\\X2816.bin',10,7,1,11),(3667,3665,12000,NULL,NULL,NULL,21,118,18,'MF/E000000000001ER3321111','V/36786/1/2','2024-07-31','791: Capacitors.\r\n_________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\nLCD: 10uF -2.\r\n3665: Capacitors. PLCC socket stay broken. Test.','','','',10,5,1,11),(3668,3666,80450,NULL,NULL,NULL,21,390,98,'1700R12ABAEZZZ','3031406','2024-08-01','3564: 1uF 450V - was bad.\r\nU14: HCNW-4506 replaced not tested\r\nC27 22uF 50V (was good).\r\nNot tested.\r\n3666: Operation panel (LCD) was bad, - replaced.\r\nUsed another (from repairing box and full repair) PS board.\r\nTested.','137','','',30,15,1,11),(3669,3667,9500,NULL,NULL,NULL,21,329,11,'MPR2000/5/D','5177-95','2024-08-04','3099: Capacitors, \r\nRelay 12V -6.\r\nSocket CPU.\r\n3565: incorrect current meter (L1)\r\nU43: LM324 (replaced all).\r\n3667: Damaged because falling. Rpaire.','53','','SOLCON\\MRP2000_D\\X2816.bin',5,3,1,11),(3670,3668,10497,NULL,NULL,NULL,21,329,19,'TPS820 480-230-24-3M-5-8','WO24296812008','2024-08-05','Incorrect optocoupler input current. Must be 27k *3 per channel.\r\nK814P -2.\r\nIncorrect AUX parameter. Must be /Remote reset/.  \r\nPrice:\r\nhttps://www.radwell.co.uk/en-GB/Buy/SOLCON/SOLCON/TPS-820-600-115-115/?redirect=true','','','',15,6,5,11),(3671,3669,10497,NULL,NULL,NULL,21,329,19,'TPS820 480-230-24-3M-5-8','WO19184390002','2024-08-05','Reset after temperature strip.  \r\nPrice:\r\nhttps://www.radwell.co.uk/en-GB/Buy/SOLCON/SOLCON/TPS-820-600-115-115/?redirect=true','','','',5,2,5,11),(3672,3670,2000,NULL,NULL,NULL,21,498,43,'TCI-2','608741','2024-08-07','Capacitors. Soldering.\r\nCalibration:\r\n0 grad = 4mA\r\n750 grad = 20 mA','9','','',4,3,5,11),(3673,3671,2000,NULL,NULL,NULL,21,498,43,'TCI-2','714466','2024-08-07','Capacitors.\r\nCalibration:\r\n0 grad = 4mA\r\n750 grad = 20 mA\r\n________________________\r\nWork 456:\r\ncalibration.\r\n3671: calibration. Replacing the screws.\r\n________________________\r\nWork 990:\r\ncalibration.','9','','',4,2,1,11),(3674,3672,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049925891','2024-08-07','2983: Capacitors\r\n1M\r\nCNY17-3\r\n3683: Varistor 240V, Fuse T2A (original) to F2A.','','','',9,5,1,11),(3675,3673,3600,NULL,NULL,NULL,21,2,11,'1P6SE7090-0XX84-2FF0','*S SIE001G9539*','2024-08-12','Leakage in keyboard between line 3 from right side \r\nand 3 from left side.\r\n\r\nLCD stay with broken glass. Changed backlight lamps (5V -2).','','','',20,10,1,11),(3676,3674,9316,NULL,NULL,NULL,21,535,52,'P0400YE-0S FBM 04','SF04333820','2024-08-12','Capacitors\r\nBad EEPROM Programming rev 0S','32','','bin',20,5,1,11),(3677,3675,3600,NULL,NULL,NULL,21,66,113,'1673534','3424231673534','2024-08-13','Making the cable 24 pin.','','','',12,8,1,11),(3678,3676,9316,NULL,NULL,NULL,21,535,52,'I/A Series P0400YE-0L FBM4','1895251566','2024-08-14','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400YE-0L','32','','',10,3,1,11),(3679,3677,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049654499','2024-08-15','3602: Varistor: RMS275V => RMS250V\r\nT2A, C10: 220uF 25V => 220uF 35V, CNY17-3.\r\nTest.\r\n3677: RMS275V, T2A => F2A','','','',6,3,1,11),(3680,3678,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049975295','2024-08-15','3663: Varistor 275V between earth and fuse line,\r\nT2A (original was ok) changed to F2A.\r\n3678: Varistor 275V, T2A (original was ok) changed to F2A.\r\n220uF 25V (very bad condition).','','','',6,3,1,11),(3681,3679,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0S FBM1','SF00379733','2024-08-15','Capacitors.\r\nProgramming the new AT28CC64 with DUMP 0S\r\nch_2 - flashed. 49R9, A11 (cmd code) => BAV70','32','','bin',10,5,1,11),(3682,3680,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895242721','2024-08-18','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0L','32','','',10,3,1,11),(3683,3681,1500,NULL,NULL,NULL,21,743,8,'245TCR-4','95.03 A','2024-08-19','Cleaning.','164','','',10,5,1,11),(3684,3682,1500,NULL,NULL,NULL,21,743,8,'245TCR-4','95.26 A','2024-08-19','Cleaning.','164','','',10,5,1,11),(3685,3683,1500,NULL,NULL,NULL,21,743,8,'245TCR-4','95.42 A','2024-08-19','Cleaning.','164','','',10,5,1,11),(3686,3684,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895242729','2024-08-20','Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0L','32','','',10,3,1,11),(3687,3685,66000,NULL,NULL,NULL,21,51,13,'NXS01685-A2H0SSV-A1A2000000','V00002104335','2024-08-26','Nozzle on CPU board.\r\nCorrosion, cleaning.\r\n\r\nhttps://www.ebay.com/itm/305582608732?_nkw=NXS01685&itmmeta=01J66TFS7BGCPAFF7AXD5DPJM7&hash=item472624915c:g:n5MAAOSw6bBl7~ih&itmprp=enc%3AAQAJAAAA8HoV3kP08IDx%2BKZ9MfhVJKkRQOo85%2BsKKct00ZD5Uncy7j%2BGSMZPj04D3P0RtaiGM8X%2FYxxzBQ2KhbBywcfpTkn0JFLwCtjnOS79gm5dfuD0yocTNJnJNDozLy%2F3YXr2qukgnXeLsdBYMyzFhUdRwUIt5Ge6J8wCqmZmHlZUIcmHylv%2ByOOTqdQANaflpJkV2%2FFcoNwhxtu3TRZwe1ijUzshv5bH9l2FYBLfvPhepRXlGL5dqZI5zmqDxqQb3DCtW8cznkvWERQy5It%2F9D3sOS0JMnVqh52zh7bPkzPW0dxhCcH6s9%2FadFpzXTYZojUBUw%3D%3D%7Ctkp%3ABk9SR-KTv9qxZA','7','','',25,10,1,11),(3688,3686,2000,NULL,NULL,NULL,21,514,43,'20/6VD','9801085','2024-08-26','No problem.\r\n0-60VAC => 4-20mA','12','','',4,2,1,11),(3689,3687,6300,NULL,NULL,NULL,21,631,35,'LAE 1','000007','2024-08-28','3687: 24V relay 1CH/O -3\r\n470uF 40V => 680uF 35V\r\n470uF 16V\r\n100k.\r\nCleaning.','','','',4,2,1,11),(3690,3688,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240G110/2','5987639','2024-09-02','FAN flow corrosion!!!\r\nUnit after non qualification service of BENNING.\r\n470k SMD open.\r\nIPW60R045CP -2 (PFC) 752-8397; \r\nT20A, ZD18V, 51k MELF -2, 10k MELF.\r\nRepairing the socket ISA -2.\r\n\r\nUsed the new silicone paste for PFC and output voltage side PS.\r\n\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.','132','','',60,20,1,11),(3691,3689,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240G110/2','5909013','2024-09-04','2745: FAN flow corrosion.\r\nInt. print between PCB and ISA-connector (Flashed).\r\nTested. Wile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.\r\n3689: FAN flow corrosion. Flash on PCB CPU and ISA-connector.\r\nCPU-board: Attiny45 Microchip are short. Programming the new one \r\n(Flash; EEPROM; Device configuration Lock&Fuses; Override EESAVE setting).\r\n8MHz Int.osc.LC). Possible to check outside from board. Pin_1 must be opened.\r\n78M05, BC817, 4148, 470k MELF.\r\nPFC-module: UC3855, LM2902 => LM324.','132','','',40,20,1,11),(3692,3690,10000,NULL,NULL,NULL,213,744,135,'MIO-AX8','AX8-0161129','2024-09-05','Instead 24V was appliyed 230Vac.\r\nFlashed D12: uni-direction TVS 33V => bi-direction 39V','','','',5,2,1,11),(3693,3691,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240G110/2','5999856','2024-09-09','FAN flow corrosion.\r\nISA connector repairing.\r\nCleaning.\r\nTesting.','132','','',40,20,1,11),(3694,3692,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3692','2024-09-10','100R','64','','',5,1,1,11),(3695,3693,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3693','2024-09-10','100R','64','','',5,1,1,11),(3696,3694,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3694','2024-09-10','100R, 22nF 630V','64','','',5,1,1,11),(3697,3695,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3695','2024-09-10','100R.','64','','',5,1,1,11),(3698,3696,345,NULL,NULL,NULL,21,731,90,'CP4-EST','','2024-09-10','100R, 22nF 630V','64','','',5,1,1,11),(3699,3697,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3697','2024-09-10','100R, 22nF 630V','64','','',5,1,1,11),(3700,3698,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3698','2024-09-10','100R, 22nF 630V','64','','',5,1,1,11),(3701,3699,13000,NULL,NULL,NULL,213,367,48,'TEBECHOP 3000 HDI E110-240','5778861','2024-09-11','3120: FAN flow corrosion.\r\nInt. print.\r\nPFC: IC\r\n470k (was ok), SMD n-p-n for relay, T20A.\r\nUsed the new silicone paste for PFC and output voltage side PS\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. Tested on real 230VAC after safe testing \r\nwith isolated 230VAC input.\r\n3699: Power side: 390k MELF (one from two serial).','132','','',50,24,1,11),(3702,3700,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA FBM 01','1896505260','2024-09-15','1571: 330uF 35V -10,\r\n1000uF 10V => 1000uF 16V - all.\r\nSoldering.\r\nFull test - ok.\r\n2172: NO PROBLEM.\r\n2476: new AT28C64B installed with updated DUMP DA-0P\r\n3165: short (non constant) between EEPROM pin\'s.\r\n3315: soldering the all IC\'s. 6264 - changed.\r\n3461: no problem.\r\n3643: no problem, changed LDO.\r\n3700: Soldering.\r\nFull test - ok.','32','','',8,3,1,11),(3703,3701,80450,NULL,NULL,NULL,21,390,98,'FIT-63293 2700R11BBAEZZZ','3032065','2024-09-16','2116: 1uF 450V\r\n3340: tested the HCNW4506 -ok, but replaced the new (U13, U14).\r\n3701: Test only. ok. 10uF 50V smd.','137','','',10,3,1,11),(3704,3702,12000,NULL,NULL,NULL,21,189,35,'46199001','1/S4403/38','2024-09-17','Capacitors, LCD 2*16.\r\nCleaning.','532#','','bin',10,2,5,11),(3705,3703,6000,NULL,NULL,NULL,21,607,98,'CM442-2D56/0','V6051E05G00','2024-09-17','Water inside.\r\nCleaning.','','','',5,2,1,11),(3706,3704,12000,NULL,NULL,NULL,21,685,143,'PCME QAL 181','53073','2024-09-18','Making circuit','130','','',30,20,9,11),(3707,3705,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','0039028','2024-09-18','Tested.','','','',4,3,1,11),(3708,3706,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049357699','2024-09-23','T2A, capacitors.','','','',6,1,1,11),(3709,3707,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','F004857','2024-09-23','Neon lamp -5.\r\n22M -8.','8','','',7,4,2,11),(3710,3708,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','F004779','2024-09-24','Neon lamp -10.\r\n22M -8.','8','','',7,4,2,11),(3711,3709,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','0024740','2024-09-24','22M -all.','8','','',7,4,2,11),(3712,3710,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','F007608','2024-09-25','Neon lamp -11.\r\n22M.','8','','',7,4,2,11),(3713,3711,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','F004789','2024-09-25','Neon lamp -11.\r\n22M, 470k (1_ch).','8','','',7,4,2,11),(3714,3712,2492,NULL,NULL,NULL,21,396,33,'2.250.1449','0017832','2024-09-26','762: Making new terminal,\r\nInstalling DC plug for PS 12V 2.08A\r\n1647: cleaning.\r\n3712: connector repairing.','','','',2,1,1,11),(3715,3713,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','3560','2024-09-30','Neon lamp -10.\r\n22M - all.','8','','',7,4,2,11),(3716,3714,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','31031036959','2024-10-01','Corrosion.\r\nTesting\r\n\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1','','','',9,2,1,11),(3717,3715,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','F004777','2024-10-01','Neon lamp -6.\r\n22M.','8','','',7,4,2,11),(3718,3716,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','31982707481','2024-10-01','Testing\r\n\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1','','','',9,2,1,11),(3719,3717,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','31982511213','2024-10-01','Testing\r\n\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1','','','',9,2,1,11),(3720,3718,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','F004199','2024-10-06','Neon lamp -5.\r\n22M -8.','8','','',7,4,2,11),(3721,3719,3200,NULL,NULL,NULL,21,498,43,'DCI-2','608883','2024-10-06','Capacitors, screws, soldering.\r\n\r\n0V => 4mA\r\n6V => 20mA','9','','',10,4,1,11),(3722,3720,1500,NULL,NULL,NULL,21,599,8,'MSR 020','M2-U3','2024-10-06','2921: 100uF 50V, 330uF 63V => 330uF 50V 125C\r\nRelay 24V -2.\r\nTr-r: 230V/24V/3W => 2.8W 24V (732-0395).\r\n3284: 2.8W 24V (732-0395).\r\n3502: 2.8W 24V (732-0395).\r\n3720: Tr-r: 230V/24V/3W => 2.8W 24V (732-0395).','','','',5,1,1,11),(3723,3721,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','JUL13/84','2024-10-07','Repairing the terminal\r\nTested.','','','',4,3,1,11),(3724,3722,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','31992807351','2024-10-08','4 relays was changed.\r\nTested.','','','',4,3,1,11),(3725,3723,18000,NULL,NULL,NULL,21,51,13,'30CX4G2N1','1291779','2024-10-14','1679: Corrosion.\r\nLarge Cooler 24V.\r\nShort in the input CHOKE, - replacing.\r\n3723: Pre DRV: R108 680k (DC BUS Meter)\r\nOP D-Type connector changed, corrosion on print side of the Pre-DRV.\r\n! Special parameters in CPU. Motor do not working \r\nwith base parameters. Must be changed.','7 548','','bin',30,20,1,11),(3726,3724,7600,NULL,NULL,NULL,21,161,17,'AS-H827-100','0041787','2024-10-22','Connector two side changed.\r\nCleaning.','','','',8,4,1,11),(3727,3725,7600,NULL,NULL,NULL,21,161,17,'AS-H827-103','0039994','2024-10-23','Cleaning.','','','',8,4,1,11),(3728,3726,4700,NULL,NULL,NULL,21,519,41,'3081','3726','2024-10-27','Cleaning.','','','',5,3,5,11),(3729,3727,7600,NULL,NULL,NULL,21,161,17,'AS-H827-100','0045126','2024-10-27','Repair the screws of D-SUB socket','','','',8,4,1,11),(3730,3728,12100,NULL,NULL,NULL,21,161,3,'AS-P810-000','309336','2024-10-27','Capacitors.\r\nTested.','123','','',15,4,1,11),(3731,3729,18200,NULL,NULL,NULL,21,161,135,'AS-J890-102','0029269','2024-10-27','Cleaning\r\nWithout test.','21','','',5,3,5,11),(3732,3730,2492,NULL,NULL,NULL,21,396,33,'Ladegerat 2.250.1005','24201','2024-10-29','3155: Capacitors\r\nSoldering the transformer pin,\r\ncleaning.\r\n3730: cleaning','','','',10,5,5,11),(3733,3731,2492,NULL,NULL,NULL,21,396,33,'Ladegerat','24155','2024-10-29','535: Soldering.\r\nCleaning.\r\n1793:Cleaning.1901: Cleaning\r\n2509: capacitors, cleaning\r\n3092: Cleaning\r\n3333: Cleaning\r\n3731: test ok.','','','',5,3,2,11),(3734,3732,3500,NULL,NULL,NULL,21,533,22,'140DDI35300','350162468270301121','2024-10-29','Corrosion on ASIC side.\r\nCleaning.','','','',6,2,5,11),(3735,3733,2600,NULL,NULL,NULL,21,2,1,'SIPART AT&S DSPTH NA','1165309','2024-10-29','Cleaning.','103','','',5,2,2,11),(3736,3734,1500,NULL,NULL,NULL,213,622,35,'GNP-1','14/0002','2024-11-03','Capacitors.\r\n4N33, 100k -2, 470k -2.','','','',8,4,1,11),(3737,3735,5000,NULL,NULL,NULL,21,117,67,'nc6120','CNU5391M4H','2024-11-05','PC for generator\r\n\r\n3V lithium','','','',4,3,1,11),(3738,3736,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','31980462510','2024-11-05','Corrosion.\r\nTesting\r\n\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1','','','',9,3,1,11),(3739,3737,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','31980462502','2024-11-05','Corrosion.\r\nTesting\r\n\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1','','','',9,3,1,11),(3740,3738,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','31980462487','2024-11-05','Corrosion, 10M.\r\nTesting\r\n\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1','','','',9,3,1,11),(3741,3739,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','31981102450','2024-11-05','Corrosion.\r\nTesting\r\n\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1','','','',9,3,1,11),(3742,3740,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','31980573087','2024-11-05','Corrosion.\r\nTesting\r\n\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1','','','',9,3,1,11),(3743,3741,2200,NULL,NULL,NULL,21,66,3,'A05B-2670-C202','ZFP239IA0022170','2024-11-06','Making the box for PS (robot).','','','',20,9,1,11),(3744,3742,4600,NULL,NULL,NULL,21,161,22,'AS-B825-016','0046263','2024-11-07','Testing\r\n\r\nConfiguration:\r\n\r\n/+24V 0V/ 16 15 14 13 12 11 10 9  /0V 0V/  8 7 6 5 4 3 2 1','','','',9,3,1,11),(3745,3743,6300,NULL,NULL,NULL,21,631,35,'LAE 1','009007','2024-11-07','24V relay 1CH/O -3\r\n470uF 40V => 470uF 35V\r\n470uF 16V','','','',4,3,1,11),(3746,3744,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','31041430366','2024-11-17','Tested.','','','',4,3,1,11),(3747,3745,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','F011636','2024-11-17','Tested.','','','',4,3,1,11),(3748,3746,1100,NULL,NULL,NULL,21,725,3,'GT-93020-0324','484367116/14','2024-11-17','24VAC input\r\n\r\n22uF 35V','','','',2,1,1,11),(3749,3747,4600,NULL,NULL,NULL,21,161,8,'AS-B814-108','0039066','2024-11-18','1 Relay was bad.\r\nTested.','','','',4,3,1,11),(3750,3748,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192177','2024-11-20','Capacitors. Corrosion!\r\nProgramming the new AT28CC64 with DUMP 0L','32','','',10,3,1,11),(3751,3749,345,NULL,NULL,NULL,21,731,90,'70222HI2','3749','2024-11-26','Zener 6V2, 220R','162','','',4,2,1,11),(3752,3750,2348,NULL,NULL,NULL,21,51,8,'CM221199 PC00273 F','273J21132464AS','2024-11-26','Bad DATA 24C02C (Atmel)\r\nProgramming','7 548#','','VACON',10,5,1,11),(3753,3751,2500,NULL,NULL,NULL,21,582,108,'HB1C4NEUH','WE032627015','2024-11-26','PFC faulted, flashed.\r\nInstalling the new one.','','','',2,1,1,11),(3754,3752,1500,NULL,NULL,NULL,213,622,35,'GNP-1','17/0381','2024-11-26','Flash on relay side\r\nCapacitors (was good)','','','',8,4,1,11),(3755,3753,2000,NULL,NULL,NULL,21,469,76,'PAXS0010','M2397A','2024-11-27','Capacitors.','40','','RedLion',8,3,5,11),(3756,3754,10497,NULL,NULL,NULL,21,329,19,'TPS820 480-230-230-5-8-D-S','WO18149429001','2024-11-27','No problem (bad contact in filter 4...20)','','','',5,3,5,11),(3757,3755,6500,NULL,NULL,NULL,21,672,98,'LFG10','568/99','2024-12-01','2271: Capacitors. 7812\r\n3755: test, ok. 5&6 input','','','',5,3,1,11),(3758,3756,2500,NULL,NULL,NULL,21,314,65,'MCVS','1942-8','2024-12-01','2165: Terminal repair.\r\nNew cover installed.\r\nTested.\r\nTerminal (+) broken. Replaced.','','','',8,1,1,11),(3759,3757,44196,NULL,NULL,NULL,21,161,67,'AS-J890-102','0030852','2024-12-02','Test CPU -ok.\r\n\r\nFor RUN CPU apply on pin 14\r\nof P1 low level (0V).\r\n\r\nhttps://www.ebay.com/itm/167060936494?_skw=as-j890&itmmeta=01JE3JHYF0SWVCG7NHHGCV06Y5&hash=item26e59b972e:g:pcIAAOSwdAxnJjTZ&itmprp=enc%3AAQAJAAAA0HoV3kP08IDx%2BKZ9MfhVJKlUBROUs7jam%2FuUnH%2BV9Nss0NiP%2B6bWZKPNUULK%2FZZczOvwt1Vvt%2FFd0iB6fpPlZSmdTXc31wdIes4rOs14xM5zNoFzG22zlUhpDtcaLm2Lh1zb1X4xIJlZJ9naXw%2Ff2QMmcZy1lJQa3qtrPM8CqRmhBbwOd9mhmkNAVosAVCsCUD1ppHc3iNPNqU8OnJu%2Ba6lEDAwhBsM%2BX0ki8%2Fb0oQpRCDC9kULdcU3NoPgL--MFh8v2lNZCJ06NpXlxbI4mG%2Fg%3D%7Ctkp%3ABk9SR8znx_LwZA','21','','',15,10,2,11),(3760,3758,4500,NULL,NULL,NULL,21,118,1,'SPCU 3D45-AA','295301','2024-12-08','Testing EPROM 27256 (076D).\r\nProgramming the new EEPROM\r\nRepairing the pin\'s of module SPAZ2410D','','','magnesium',8,4,1,11),(3761,3759,4500,NULL,NULL,NULL,21,118,1,'SPCU 3D45-AA','254629','2024-12-09','Bad EEPROM.\r\nProgramming the new EEPROM','','','magnesium',8,1,1,11),(3762,3760,25000,NULL,NULL,NULL,21,215,4,'07F5A1A-3E2F','305319983/2018287/2018/07/0010','2024-12-09','All capacitors are bad.\r\nCeramics module (DC Bus isolated meter) was tested 200Vin/1Vout','165','','PDF',40,20,1,11),(3763,3761,3230,NULL,NULL,NULL,21,118,8,'SPTU 240 R4','254737','2024-12-10','Electrican Laboratory.\r\nRelays, capacitors.','','','',7,3,1,11),(3764,3762,6700,NULL,NULL,NULL,21,524,104,'P3U 8317275','ARSH-0400','2024-12-11','Cleaning\r\nNo problem.','151','','',5,2,1,11),(3765,3763,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARJJ-0492','2024-12-19','No problem founded.\r\nTested','144','','',6,3,1,11),(3766,3764,2492,NULL,NULL,NULL,21,396,33,'Ladegerat 2.250.1005','30202','2024-12-19','3764: Cleaning.\r\nTest.','','','',5,3,2,11),(3770,3768,35000,NULL,NULL,NULL,21,702,50,'MSEL controller IXP-4N520-WA-M-P','A60766843','2024-12-29','2981: cooler MGT4024YB-O10\r\n3768: 10uF 50V short C803','','','',20,8,2,11),(3767,3765,6700,NULL,NULL,NULL,21,524,98,'P3U Polytron 7000 8317610','ARSJ-0454','2024-12-19','No problem founded.\r\nTested','144','','',6,3,1,11),(3768,3766,15000,NULL,NULL,NULL,21,671,35,'MT8070iH','110508564','2024-12-22','New LCD installed','','','',5,4,1,11),(3769,3767,6300,NULL,NULL,NULL,21,631,35,'LAE 1','000007','2024-12-22','3687: 24V relay 1CH/O -3\r\n470uF 40V => 680uF 35V\r\n470uF 16V\r\n100k.\r\nCleaning.\r\n3767: 16 Way Through Hole DIP Rotary Switch -3 (379-3812)\r\nCleaning the terminal.','','','',8,4,1,11),(3771,3769,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896515509','2024-12-29','2911: Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P\r\n3769: soldering','32','','',10,3,1,11),(3772,3770,6910,NULL,NULL,NULL,21,745,73,'VPA W4HOV70','80888337903200013','2025-01-02','Capacitors','','','',12,8,1,11),(3773,3771,6910,NULL,NULL,NULL,21,745,73,'VPA W4HOV70','80888337903200005','2025-01-02','Capacitors','','','',12,8,1,11),(3774,3772,15000,NULL,NULL,NULL,21,189,35,'1/S4403/38','46193001','2025-01-06','Capacitors:\r\n470Uf 50V, 470uF 25V, 47uF 25V -2.','532#','','bin',10,3,5,11),(3775,3773,15000,NULL,NULL,NULL,21,189,35,'1/94403/38','3773','2025-01-06','Make the new one','532#','','bin',10,3,5,11),(3776,3774,15000,NULL,NULL,NULL,21,189,35,'46199001','1461C18000000','2025-01-06','422:\r\n470uF 35V -2, 47uF 16V -2.\r\nLCD cable and conectors.\r\n_______________________\r\nKeyboard: \r\nPIN1: Up\r\nPIN2: Enter\r\nPIN3: Down\r\nPIN4: ESC\r\nPIN5: Gnd\r\n\r\n1305: 470uF 25V\r\n3774: Dridge was broken. Replacing the new OP.','532#','','bin',10,5,2,11),(3777,3775,2000,NULL,NULL,NULL,21,319,35,'LB 474-01-M0','6206','2025-01-07','23mA output insted 4...20mA\r\nMaking circuit.\r\nAO tested\r\nRadiation input not tested fully because the lack of conditions.','166','','',200,50,2,11),(3778,3776,2492,NULL,NULL,NULL,21,396,33,'Ladegerat 2.250.1005','30202','2025-01-08','3764: Cleaning.\r\nTest.\r\n3776: BC547 -2\r\nInt. print','167','','',8,4,2,11),(3779,3777,12000,NULL,NULL,NULL,21,118,18,'MF/E151374101005ER3321111','V/30535/3/1','2025-01-14','825: Capacitors.\r\n_________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\nDisplay: 10uF -2.\r\n1912: test\r\n3777: test','','','',10,6,1,11),(3780,3778,12000,NULL,NULL,NULL,21,118,18,'MF/E151374101005ER3321111','V/10384/13/1','2025-01-14','________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\nDisplay: 10uF -2.\r\n3778: test','','','',10,6,1,11),(3781,3779,12000,NULL,NULL,NULL,21,118,18,'MF/E101374101SS5ER3321111','V/10384/20/1','2025-01-14','1093: Capacitors.\r\n_________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\n\r\n1730: PLCC Socket.\r\n2744: test only\r\n2744: capacitors, cable repair, socket CPU cleaning','','','',10,6,1,11),(3782,3780,2500,NULL,NULL,NULL,21,582,108,'HB1C4NEUH','WE012632056','2025-01-15','PFC faulted, flashed.\r\nInstalling the new one.','','','',3,2,1,11),(3783,3781,2500,NULL,NULL,NULL,213,144,66,'LG-JCB01','2H11996A','2025-01-16','Making circuit.\r\nFull test','168','','',6,3,1,11),(3784,3782,14100,NULL,NULL,NULL,21,531,98,'3081','AIT52122/2','2025-01-16','277: Cleaning the corrosion on terminal.\r\nRepair screws. Test, - ok.\r\n334: cleaning. Test, - ok.\r\n392: changed all amplifiers, multiplexors and analog switches.\r\nCalibrated.\r\n502: new problem found: leakadge between input pin\'s.\r\n1063: new screws are installed.\r\n3782: Corrosion. Cleaning and calibration.','35','','',20,6,1,11),(3813,3811,14100,NULL,NULL,NULL,21,531,98,'3081','AIT52122/2','2025-03-12','277: Cleaning the corrosion on terminal.\r\nRepair screws. Test, - ok.\r\n334: cleaning. Test, - ok.\r\n392: changed all amplifiers, multiplexors and analog switches.\r\nCalibrated.\r\n502: new problem found: leakadge between input pin\'s.\r\n1063: new screws are installed.\r\n3782: Corrosion. Cleaning and calibration.\r\n3811: Cleaning and calibration.','35','','',20,6,1,11),(3785,3783,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0L FBM1','1895192153','2025-01-21','3470: Capacitors.\r\nProgramming the new AT28CC64 with DUMP 0L\r\n3783: no problem.','32','','',10,3,1,11),(3786,3784,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896505243','2025-01-21','2844: Capacitors.\r\nProgramming the new AT28HC64 with original DUMP from\r\nP0400DA-0P\r\n3784: test','32','','',10,3,1,11),(3787,3785,26324,NULL,NULL,NULL,21,51,13,'NXS01055A2H0SSVA1A2000000','12327672','2025-01-29','Fault_7\r\nPre-charge board: relay 24V -3, 2u2 305VAC  -4.\r\nTest.\r\n\r\nhttps://www.interautomatika.eu/en/vacon-nxp-drive-55kw-105a-380-500v-ip21.html','7 548#','','bin',20,10,1,11),(3788,3786,2500,NULL,NULL,NULL,21,314,65,'MCVS','1942-9','2025-02-02','2145: Terminal repair.\r\n2161: incorrect used.\r\nFlash on IC4: LM358\r\n3786: the terminals repairing. \r\n\r\nTested.','','','',5,3,1,11),(3789,3787,17300,NULL,NULL,NULL,21,390,98,'1700R12ABAEZZZ','3169930','2025-02-02','HCPL0501 (761-4457) - bad (feedback PWM PS).\r\n1uF 450V - was ok.\r\nU14: HCNW-4506 was ok\r\nC27 22uF 50V (was bad).\r\nL3: 4700uH.\r\nTested.','137','','',30,15,1,11),(3790,3788,3200,NULL,NULL,NULL,21,498,43,'RTI-2','608802','2025-02-04','Capacitors.\r\nWas calibrated later: \r\n0C     100R\r\n400C   247.09R','9','','',5,2,1,11),(3791,3789,5200,NULL,NULL,NULL,21,498,24,'AFM-16','2725907','2025-02-04','3329: Capacitors\r\n56R -2.\r\n3789: no problem founded (4051 -tested)','9','','',10,5,1,11),(3792,3790,17300,NULL,NULL,NULL,21,390,98,'FIT-99767 1700R12ABZEZZZ','3117482','2025-02-05','1918: 1uF 450V\r\nD2: 8S => SMD SOT23: ZD12V\r\nLT1017 (?)\r\n3076: U14 HCPL-4506 -bad.\r\nU13: also been replaced\r\n3790: Preventive maintenance: C27 (was bad), \r\nHCPL0501 /761-4457/ (was ok)','137','','',10,5,1,11),(3793,3791,17300,NULL,NULL,NULL,21,390,98,'2700R12BBZEXZX','3780857','2025-02-05','Preventive maintenance: \r\nC27 10uF 50V, 1uF 450V - was bad,\r\nU13, U14: also been replaced (was ok),\r\nU4: HCPL0501 /761-4457/ (was ok).','137','','',10,5,1,11),(3794,3792,17300,NULL,NULL,NULL,21,390,98,'FIT-99607 1700R12ABZEZZZ','3117216','2025-02-06','Was working.\r\n1uF 450V - changed later.\r\nU14 HCNW4506 - changed later.\r\nC27 10uF 50V - bad, - changed.\r\nU4 - HCPL0501 was ok, - changed.','137','','',10,6,1,11),(3795,3793,17300,NULL,NULL,NULL,21,390,98,'LFT12BAEZZZ','3206655','2025-02-06','Was working with good conditions.\r\n1uF 450V - was ok, - changed.\r\nU14 HCNW4506 - was ok, - changed.\r\nC27 10uF 50V - was ok, - changed.\r\nU4 - HCPL0501 was ok, - changed.','137','','',10,6,1,11),(3796,3794,9316,NULL,NULL,NULL,21,535,52,'P0400YE FBM4 REV-0S','SF02307529','2025-02-12','3383: Capacitors.\r\nProgramming the new AT28C64B rev 0S\r\n3794: no problem','32','','',15,5,5,11),(3797,3795,1200,NULL,NULL,NULL,21,423,3,'CM 90-PS-230AC/2x15DC/1','2943806','2025-02-12','Test.','','','',3,2,9,11),(3798,3796,80000,NULL,NULL,NULL,21,319,35,'LB 474-01-M0','6206','2025-02-12','23mA output insted 4...20mA\r\nMaking circuit.\r\nAO tested\r\nRadiation input not tested fully because the lack of conditions.\r\n3796: Working after changing the AD','166','','',200,50,2,11),(3799,3797,3210,NULL,NULL,NULL,213,654,3,'FPN5228A','850010','2025-02-19','2572: Price of new one: $904\r\n\r\nC26, C27: 560uF 200V\r\nCNY17-3 -3.\r\nR1, R86: 27k 2W => 56k 5W\r\nUC3842. \r\nC41: 1uF 50V, C23: 47uF 25V, C25: 56uF 35V, C19: 220uF 35V.\r\n220R 3W -3\r\nRegulation 13.75V to 14V\r\n3797: Capacitors. Fuse holder.','107','','',10,6,5,11),(3800,3798,3210,NULL,NULL,NULL,213,654,3,'FPN5228A','418247','2025-02-19','Price of new one: $904\r\n\r\nR1, R86: 27k 2W \r\nUC3842. \r\nC41: 1uF 50V, C23: 47uF 25V, C25: 56uF 35V, C19: 220uF 35V.\r\n220R 3W -3','107','','',10,6,5,11),(3801,3799,9316,NULL,NULL,NULL,21,535,24,'I/A Series P0400DA-0P FBM1','1896505251','2025-02-23','Board after global PS repairing. Used add transformer and DC-DC.\r\nReturning all to original configuration.\r\n\r\nPCB repairing. \r\nCapacitors.\r\nProgramming the new AT28HC64 with original updated DUMP from\r\nTest','32','','',15,8,5,11),(3802,3800,9316,NULL,NULL,NULL,21,535,52,'P0400YE-0P FBM 04','1896465232','2025-02-24','Capacitors\r\nEEPROM Programming rev 0P\r\n49R9.','32','','bin',20,5,1,11),(3803,3801,6700,NULL,NULL,NULL,21,116,98,'pH 2100/2','54268/1533223/MT','2025-02-25','2880:\r\nCleaning.\r\nCalibration (1100).\r\n3022: \r\nCleaning.\r\nCalibration (1100).\r\n3801: \r\nCleaning.\r\nCalibration (1100).','114','','',15,5,1,11),(3804,3802,45000,NULL,NULL,NULL,21,2,11,'SIMATIC IPC277E','VPK5950704 2018','2025-03-03','3640: Over heating CPU\r\nInserting the new thermal pad - resin.\r\nCleaning\r\n3802:  Over heating CPU\r\nInserting the new thermal pad - mica.\r\nCleaning','','','',10,5,1,11),(3805,3803,3200,NULL,NULL,NULL,21,498,43,'RTI-2','714496','2025-03-04','Capacitors.\r\nDIP-switches were is not correct position.\r\n\r\nCalibrated with hand make board: \r\n\r\n0C     100R\r\n400C   247.09R','9','','',8,4,5,11),(3806,3804,12000,NULL,NULL,NULL,21,118,18,'Kent-Taylor F/E','V/12031/2/1','2025-03-05','Another board installed (original, without repair later).\r\n \r\nCapacitors:\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.','','','',10,8,1,11),(3807,3805,6500,NULL,NULL,NULL,21,498,6,'FOM10','630573','2025-03-09','68: Full test - ok.\r\nFounded problem in fiber optics cable.\r\n3805: device ok.','9','','',7,1,2,11),(3808,3806,6970,NULL,NULL,NULL,21,434,3,'EWS600-24','CQT-821Z12-0014','2025-03-09','1025, 166\r\n3331: 3.9uF 400V, FAN, capacitors.\r\n3806: cleaning.','62','','',8,4,1,11),(3809,3807,3570,NULL,NULL,NULL,21,434,3,'EWS300-24','736Z31-0091W','2025-03-09','Flash on PCB:\r\nCleaning.\r\nF2.5A => T2A, ZD15V (KE15), 2SC2655, 2SK513.\r\nFull test.','62','','',10,6,2,11),(3810,3808,3570,NULL,NULL,NULL,21,175,3,'HWS600-24','5CT-931CA5-0179W0209','2025-03-11','2559: Hight power PS unit (24V 8.2A) with PFC. \r\nTotal corrosion, flash on PCB\r\nCleaaning.\r\nTest without lamp - ok.\r\n3808: cleaning, test.','','','',10,5,2,11),(3811,3809,12467,NULL,NULL,NULL,213,322,143,'VIS-8','218N3411','2025-03-11','22uF 35V - shorted.\r\n\r\nhttps://www.labgearusa.com/mrc-lab-vis-8s-intelligent-digital-viscometer/','','','',5,2,1,11),(3812,3810,15000,NULL,NULL,NULL,21,555,19,'10007 26/99','45143/04RS','2025-03-11','371: Lost phase alarm.\r\n220uF 35V -3,\r\n10uF 50V -8.\r\nInterrupted pin transformer (cool soldering).\r\n2895: Lost phase alarm.\r\n220uF 35V -3,\r\n10uF 50V -8.\r\nSFH617\r\n3810: soldering. Board 10009: 10uF -2, board 10010: C4 1uF.\r\nBoard was ok and stay working (tested in rectifier another)','141','','',25,10,1,11),(3814,3812,3570,NULL,NULL,NULL,21,434,3,'EWS300-24','706Z31-0088W','2025-03-12','801: 2.2uF 250V\r\nCleaning.\r\nFAN.\r\n1496: corrosion!\r\nUF4007 (snubber), fuse holders an fuses.\r\nCleaning.\r\n3812: corrosion! 2SC2655','62','','',20,7,2,11),(3815,3813,3000,NULL,NULL,NULL,21,206,58,'PM130EH PLUSE','801353','2025-03-17','Capacitors.\r\n47uF 25V => 47uF 35V, 4700uF 6.3V => 4700uF 10V','#583','','',9,3,2,11),(3816,3814,1500,NULL,NULL,NULL,21,175,3,'DPP240-24-1','150108-9601161501284','2025-03-18','Bridge 600V 6A => 1000V 10A','','','',10,5,1,11),(3817,3815,2500,NULL,NULL,NULL,21,746,3,'PLEY24HSZ-PU','3815','2025-03-19','Capacitors - ok.\r\nSoldering and cleaning.','','','',7,5,2,11),(3818,3816,9316,NULL,NULL,NULL,21,535,52,'P0400YE FBM4 REV-0P','1897046972','2025-03-20','595: Capacitors.\r\n2204: Problem with AT28C64B (KEB).\r\nEEPROM was changed, used original UPDATED DUMP 0P\r\nDuring time - restart, 6264 - was bad\r\n3816: unknown problem. LDO changed.','32','','',15,5,1,11),(3819,3817,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','31980573229','2025-03-23','Neon lamp -12.\r\n22M -12.','8','','',7,4,2,11),(3820,3818,7500,NULL,NULL,NULL,21,161,22,'AS-B809-016','31980573163','2025-03-23','Neon lamp -12.\r\n22M -12.','8','','',7,4,2,11),(3821,3819,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049704291','2025-03-24','3591: C10: 220uF 25V => 220uF 35V, 1000uF 25V => 35V.\r\nTest.\r\n3645: Varistor: RMS275V, T2A\r\nTest.\r\n3819: Varistor: RMS275V, T2A\r\nTest.','','','',3,1,1,11),(3822,3820,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049925891','2025-03-24','2983: Capacitors\r\n1M\r\nCNY17-3\r\n3682: Varistor 240V, Fuse T2A (original) to F2A.\r\n3820: Varistor 275V, T2A','','','',9,5,1,11),(3823,3821,6500,NULL,NULL,NULL,21,396,3,'SNT00 V1/5','0049315899','2025-03-24','3592: Varistor: RMS275V => RMS250V\r\nT2A, C10: 220uF 25V => 220uF 35V, CNY17-3.\r\nSoldering, cleaning.\r\nTest.\r\n3662: Varistor 275V, T2A (original) changed to F2A\r\n3821: Varistor 275V, T2A (original).','','','',6,3,1,11),(3824,3822,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3610','2025-03-24','3610: 100R, ZD10V 1W.\r\n3822: flashed. Replacing the board.','64','','',5,1,1,11),(3825,3823,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3823','2025-03-25','100R, 470nF 275V, 2N6075BG, \r\ncontactor coil interrupted from HV.','64','','',5,1,1,11),(3826,3824,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3824','2025-03-25','Was without board. Installed.  \r\nStay with bad contactor coil (interrupted from HV).','64','','',5,1,1,11),(3827,3825,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3825','2025-03-25','470nF 275V, 2N6075BG, \r\nStay with bad contactor coil (interrupted from HV).','64','','',5,1,1,11),(3828,3826,345,NULL,NULL,NULL,21,731,90,'CP4-EST','2826','2025-03-25','100R, 470nF 275V, 2N6075BG, \r\nStay with bad contactor coil (interrupted from HV).','64','','',5,1,1,11),(3829,3827,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3827','2025-03-25','100R, 22nF 275V, ZD10V','64','','',5,1,1,11),(3830,3828,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3828','2025-03-26','100R, ZD10V','64','','',5,1,1,11),(3831,3829,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3829','2025-03-26','100R, ZD10V','64','','',5,1,1,11),(3832,3830,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3609','2025-03-26','3609: 100R, ZD10V 1W, 22nF 630V\r\n3830: 100R','64','','',5,4,1,11),(3833,3831,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3613','2025-03-26','3613: 100R, ZD10V 1W, 22nF 630V\r\n3831: 100R','64','','',5,4,1,11),(3834,3832,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3427','2025-03-26','3427: 400VAC applied. 100R - flashed.\r\nClearing.\r\n3832: 100R','64','','',2,1,1,11),(3835,3833,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3607','2025-03-26','3607: 100R, ZD10V 1W.\r\n3833: 100R, 22nF 400V, ZD10V, 40106.','64','','',5,4,1,11),(3836,3834,1700,NULL,NULL,NULL,21,401,3,'MAP55-4001','7/99BJ','2025-03-26','999-202-214\r\n\r\n2180: 100uF 400V, 47uF 400V\r\nTrimmer 10k.\r\nRegulation  voltage output +5.3V\r\n2834: Regulation  voltage output +5.32V from 5.28V (with load).','19','','',7,4,1,11),(3837,3835,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3835','2025-03-26','100R, ZD10V 1W','64','','',5,1,1,11),(3838,3836,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3836','2025-03-26','100R, ZD10V 1W, 22nF 400V','64','','',5,1,1,11),(3839,3837,2492,NULL,NULL,NULL,21,396,33,'Ladegerat','24155','2025-03-27','535: Soldering.\r\nCleaning.\r\n1793:Cleaning.1901: Cleaning\r\n2509: capacitors, cleaning\r\n3092: Cleaning\r\n3333: Cleaning\r\n3731: test ok.\r\n3837: full test ok.','','','',7,4,2,11),(3840,3838,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3838','2025-03-30','100R, ZD10V 1W, D2: 1N4148, 40106','64','','',5,1,1,11),(3841,3839,2400,NULL,NULL,NULL,21,51,52,'OPTA1','253L 1243 1006 PS','2025-03-30','2828: 24C02 - flashed bad  (the new one programming)\r\nIDC-connectors and jumpers changed. .\r\nTesting.\r\n3839: Cleaning. Test.','7 548#','','VACON',10,2,1,11),(3842,3840,8800,NULL,NULL,NULL,21,51,13,'NXS00075A5H1SSVA1A2000000','12476347','2025-03-30','ALARM_13 (temperature)\r\n\r\n2FAN\'s was bad. \r\nFirst was changed (541-5306) and second was repaired. \r\nFull test.','7 548#','','',15,10,2,11),(3843,3841,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3841','2025-03-31','100R, ZD10V 1W, 22nF 400V','64','','',5,1,1,11),(3844,3842,6400,NULL,NULL,NULL,21,263,19,'PC11-F30B-0000','00006016','2025-04-01','2785: Int. 2 Coolers.\r\nCPU corrosion, cleaning.\r\n3842: SCR - ok\r\nTested 4-20mA, Phase control.','','SCR pdf','',10,5,1,11),(3845,3843,2200,NULL,NULL,NULL,21,555,19,'15002','60/10000','2025-04-01','Snubber board rectifier.\r\n47uF 350V => 47uF 450V\r\n1.5k 5W - interrupted => 3.3k 3W II 3.3k 3W.','141','','',5,3,1,11),(3846,3844,2200,NULL,NULL,NULL,21,555,19,'15002','2A2','2025-04-01','Snubber board rectifier.\r\n47uF 350V => 47uF 450V','141','','',5,3,1,11),(3847,3845,8000,NULL,NULL,NULL,21,555,19,'15008/10008','1U5K','2025-04-02','Capacitors, SFH617A  -6.\r\nFull test.','169','','',20,10,1,11),(3848,3846,8000,NULL,NULL,NULL,21,555,19,'15008/10008','1U4K','2025-04-02','Capacitors, SFH617A  -6.\r\nFull test.','169','','',20,10,1,11),(3849,3847,8000,NULL,NULL,NULL,21,555,19,'15008/10008','1U3K','2025-04-02','Capacitors, SFH617A  -6.\r\nFull test.','169','','',20,10,1,11),(3850,3848,8000,NULL,NULL,NULL,21,555,19,'15008/10008','1U2K','2025-04-02','Capacitors, SFH617A  -6.\r\nFull test.','169','','',20,10,1,11),(3851,3849,6400,NULL,NULL,NULL,21,263,19,'PC11-F30B-0000','00006018','2025-04-07','1373: Fire on input terminal.\r\nCooler 24V.\r\n2773: Cooler 24V, CPU -faulted because corrosion.\r\nCleaning.\r\n3849: Cooler 24V\r\nTest','27','SCR pdf','',20,7,1,11),(3852,3850,15000,NULL,NULL,NULL,21,555,19,'10007','26/99','2025-04-08','1953: LED_1  alarm.\r\nAll capacitors too bad.\r\nPC817.\r\nSoldering and cleaning.','141','','',15,5,1,11),(3853,3851,10497,NULL,NULL,NULL,21,329,19,'TPS170 480-230-230-5-8-S','WO24297063002','2025-04-08','Varistor shorted.','','','',5,3,5,11),(3854,3852,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3852','2025-04-08','100R','64','','',5,1,1,11),(3855,3853,345,NULL,NULL,NULL,21,731,90,'CP4-EST','3853','2025-04-08','100R, ZD10V 470nF, contactor','64','','',5,1,1,11),(3856,3854,6910,NULL,NULL,NULL,21,745,73,'VPA W4HOV70','75887955309100091','2025-04-10','1000uF 16V (broken)\r\nInstalled the radiators.','','','',12,2,1,11),(3857,3855,3200,NULL,NULL,NULL,21,498,43,'DCI-2','608889','2025-04-20','601: Capacitors.\r\nRepair  input connector.\r\nCalibrated:\r\n0V => 4mA\r\n6V => 20mA\r\n3855: calibration','9','','',5,3,1,11),(3858,3856,2000,NULL,NULL,NULL,21,498,43,'DCI-2','639217','2025-04-20','2130: Capacitors,\r\n0V => 4mA\r\n6V => 20mA\r\n3856: calibration','9','','',5,3,1,11),(3859,3857,2000,NULL,NULL,NULL,21,498,43,'DCI-2','639295','2025-04-20','Capacitors,\r\n0V => 4mA\r\n6V => 20mA\r\n3856: calibration','9','','',5,3,1,11),(3860,3858,2000,NULL,NULL,NULL,21,498,43,'DCI-2','639320','2025-04-20','Capacitors,\r\n0V => 4mA\r\n6V => 20mA\r\n3856: calibration','9','','',5,3,1,11),(3861,3859,12000,NULL,NULL,NULL,21,118,18,'Kent-Taylor F/E','V/10384/12/2','2025-04-20','209: capacitors\r\n3859: capacitors','','','',8,4,1,11),(3862,3860,2100,NULL,NULL,NULL,21,530,3,'140CPS11420','21043408344','2025-04-22','3300uF 6.3V (was ok) - changed.\r\n330uF 25V was bad.','','','',8,3,2,11),(3863,3861,12000,NULL,NULL,NULL,21,118,18,'MF/E151374101005ER3321111','V/30535/3/1','2025-04-22','825: Capacitors.\r\n_________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\nDisplay: 10uF -2.\r\n1912: test\r\n3861: testing with calibrator - ok.\r\n3777: test','','','',10,6,1,11),(3864,3862,12000,NULL,NULL,NULL,21,118,18,'MF/E200372101001ER3321111','V/12031/15/1','2025-04-22','1068: Capacitors.\r\n_________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\nPLCC socket and PAD\'s are broken,\r\nRepair it.\r\n3862: capacitors. Testing with calibrator - ok.','','','',10,6,1,11),(3865,3863,12000,NULL,NULL,NULL,21,118,18,'MF/E101342101SS5ER33214111','V/10384/1/1','2025-04-22','_________________________________________________________\r\nOriginal list: C803, 804: 220uF 63V. C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\n2277: C407: 100uF 35V.\r\nC408, CC410, C411: 47uF 25V. C409: 100uF 50V. C509, C412: 220uF 63V.\r\nC501+C503: 22uF 100V. C606: 2200uF 16V. C603, C605: 10uF 50V. C112, \r\nC113, C109: 10uF 50V.\r\n3863: testing with calibrator -ok.','','','',10,6,1,11),(3866,3864,4800,NULL,NULL,NULL,21,653,98,'OPTIPLUX 4100','A23007784','2025-04-23','Calibration.','','','',8,4,1,11),(3867,3865,345,NULL,NULL,NULL,21,731,90,'CP4 A','3865','2025-04-28','100R, ZD10V','64','','',5,1,1,11),(3868,3866,13000,NULL,NULL,NULL,213,367,48,'G216G48/40Wr-PCT','6359256','2025-05-04','FAN flow corrosion.\r\nInt. print.\r\n470k (was ok)\r\nUsed the new silicone paste for PFC and output voltage side PS\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. \r\nTested on real 230VAC (instead 115V-216V) after safe testing.','132','','',30,10,1,11),(3869,3867,13000,NULL,NULL,NULL,213,367,48,'G216G48/40Wr-PCT','06243406','2025-05-04','Device -ok.\r\n470k (was ok)\r\nUsed the new silicone paste for PFC and output voltage side PS\r\nWile test green and red LEDs flash shortly 10-20s\r\nafter this green stay on. \r\nTested on real 230VAC (instead 115V-216V) after safe testing.','132','','',30,4,1,11),(3870,3868,8600,NULL,NULL,NULL,213,181,19,'MBR5-40-60','0000499','2025-05-04','Capacitors, relay 24V.','','','',10,5,1,11),(3871,3869,8600,NULL,NULL,NULL,213,181,19,'MBR5-40-60','0000382','2025-05-04','Capacitors, relay 24V.','','','',10,5,1,11);
/*!40000 ALTER TABLE `issue` ENABLE KEYS */;
UNLOCK TABLES;

--
-- Table structure for table `manufacturer`
--

DROP TABLE IF EXISTS `manufacturer`;
/*!40101 SET @saved_cs_client     = @@character_set_client */;
/*!50503 SET character_set_client = utf8mb4 */;
CREATE TABLE `manufacturer` (
  `id` int NOT NULL AUTO_INCREMENT,
  `name` varchar(64) DEFAULT NULL,
  PRIMARY KEY (`id`),
  UNIQUE KEY `name` (`name`)
) ENGINE=MyISAM AUTO_INCREMENT=747 DEFAULT CHARSET=utf8mb3;
/*!40101 SET character_set_client = @saved_cs_client */;

--
-- Dumping data for table `manufacturer`
--

LOCK TABLES `manufacturer` WRITE;
/*!40000 ALTER TABLE `manufacturer` DISABLE KEYS */;
INSERT INTO `manufacturer` VALUES (1,'Wenf'),(2,'Siemens'),(3,'Milacron'),(4,'Ipco'),(5,'Mitsubishi'),(6,'el.ge. milano'),(7,'Nisan'),(8,'Elpack Packaging Systems'),(9,'Danfoss'),(10,'ECO'),(11,'TI'),(12,'Fecken-Kirfel'),(13,'Kuhnke'),(14,'Skalar Delft'),(15,'Eurotherm'),(16,'Sikora Industaieelektronik GMBH'),(17,'Kuniel'),(18,'Multivac'),(19,'Rosendahl'),(20,'Macon'),(21,'TDK'),(22,'Frenic'),(23,'Rotork'),(24,'Vector Corporation'),(25,'Gabelia Macchine'),(26,'Lenze'),(27,'Bosch'),(28,'Lenze (Polytype Prandi)'),(29,'Indramat'),(30,'Baumuller'),(31,'FMC'),(32,'Applikon'),(33,'Jost'),(34,'Galil Motion Controls'),(35,'Yaskawa'),(36,'Sepex'),(37,'Vicor'),(38,'L.A.E.B.'),(39,'AG'),(40,'Lerov Somer'),(41,'Taracord'),(42,'Bekum'),(43,'Ffb Electronics'),(44,'Lemo'),(45,'Cincinati Milacron'),(46,'JAW'),(47,'Uteco'),(48,'Csikeyboards Inc'),(49,'Nipsom'),(50,'Struckmeier'),(51,'Vacon'),(52,'Taralog'),(53,'Fuji'),(54,'Varian'),(55,'Tektronix'),(56,'C.M.S.'),(57,'Marcheluzzo'),(58,'Allen-Bradley'),(59,'Lower'),(60,'Unique Solutions'),(61,'Brabender'),(62,'Berger Lahr'),(63,'Intel'),(64,'K.A.N.'),(65,'Erhardt + Leimer'),(66,'Fanuc'),(67,'Fife'),(68,'Marquip'),(69,'AES'),(70,'Emmedi'),(71,'Gabella Macchine'),(72,'ASA'),(73,'Woodman'),(74,'Piovan + Starautomation'),(75,'Madovi'),(76,'BBC'),(77,'K-Tron'),(78,'DAH BAH Machinery Industrial'),(79,'Metalarm'),(80,'Westerstrand'),(81,'Lorenz Pan'),(82,'Hitech'),(83,'Elau'),(84,'Capchecker'),(85,'Atlas Copco'),(86,'Matsushita'),(87,'Sky Hight N.V.'),(88,'Futura Liv'),(89,'EPM'),(90,'Servomac'),(91,'Sinumerik'),(92,'IR'),(93,'Delcos'),(94,'Parvex'),(95,'Messer Grieshem'),(96,'SSD'),(97,'KME'),(98,'Nicon'),(99,'CAT'),(100,'Zumbach'),(101,'AEO'),(102,'Lauda'),(103,'Mozak'),(104,'Mero Electronic Equipment'),(105,'Benshaw'),(106,'Telsonic'),(107,'Mahr Federal Inc'),(108,'Tadiran'),(109,'Steuergerat'),(110,'EVG'),(111,'Condesa'),(112,'Precor'),(113,'Arcom'),(114,'Miwe'),(115,'I.A.G.'),(116,'Mettler Toledo'),(117,'HP'),(118,'ABB'),(119,'Maywood Instruments Limited'),(120,'Glassman'),(121,'Minarik'),(122,'Sanyo Denki'),(123,'Infranor'),(124,'HDT'),(125,'Lust'),(126,'Picodynac'),(127,'Safeline'),(128,'JA Elektronik'),(129,'Electronic Control'),(130,'Honeywell'),(131,'Berges'),(132,'Vectovar'),(133,'Thermox'),(134,'Toshiba'),(135,'Fireye'),(136,'Valotti Nino'),(137,'Krauss-Maffei'),(138,'Previero'),(139,'Spellman'),(140,'Intersoll-Rand'),(141,'Delem'),(142,'Beckoff'),(143,'Ohmart Corporation'),(144,'LG'),(145,'Omron'),(146,'CTD Ltd'),(147,'Climaveneta'),(148,'Moeller Klocker'),(149,'Ditron'),(150,'General Electric'),(151,'Measurex'),(152,'Eurofeedback'),(153,'Pillar'),(154,'Charis'),(155,'Accu-Count'),(156,'Morgana Rotary Numbering'),(157,'BOP Electronics'),(158,'Ashe Controls Ltd'),(159,'Moog'),(160,'Weniger'),(161,'Modicon AEG'),(162,'Scheda'),(163,'Elma'),(164,'Versamux'),(165,'Ansaldo'),(166,'RTA'),(167,'Suzuka Fuji Xerox'),(168,'Danielson'),(169,'Shekel Scales'),(170,'Movitrac'),(171,'Merck'),(172,'Parker'),(173,'Arucomp SK'),(174,'Jakobson'),(175,'TDK-Lambda'),(176,'Estel'),(177,'Simco'),(178,'Mer Group'),(179,'Gefran 2500'),(180,'Kone Elevators'),(181,'Rhewum'),(182,'Bunler'),(183,'Schurter'),(184,'Eagle Bst Pro Mark'),(185,'Unidrive'),(186,'Abus'),(187,'CompAir'),(188,'Kundig Control System'),(189,'Jesco Topax'),(190,'Flame-baka'),(191,'Meiko'),(192,'Jopaco'),(193,'VXL instrument'),(194,'Buehler'),(195,'ELB-Schliff'),(196,'Elgoelectric'),(197,'Verfilmt'),(198,'Weko Miller'),(199,'Nilpeter'),(200,'Nippon Otis'),(201,'SCS'),(202,'Krones'),(203,'ELO'),(204,'Dynatec'),(205,'Abrichter'),(206,'Satec'),(207,'Heidenhain'),(208,'Reikotronic Cmp'),(209,'Costec'),(210,'Burster'),(211,'Acopos'),(212,'Re Elettronica Industriale'),(213,'Elbex'),(214,'Jumo'),(215,'KEB'),(216,'Cryovac'),(217,'Yako'),(218,'Meldas'),(219,'Synrad'),(220,'Loher'),(221,'Concept'),(222,'Atlas Converting Equipment'),(223,'Multitron'),(224,'Bensheim'),(225,'Hitachi'),(226,'Lorch'),(227,'Wago'),(228,'B&R'),(229,'Harmo'),(230,'PAS'),(231,'Vero'),(232,'Vee-Arc Corp.'),(233,'Hawk BST'),(234,'REO'),(235,'Wright Machinery Co Ltd'),(236,'Delta'),(237,'Fox Board'),(238,'N.C.S Computer Italia'),(239,'Conveyor Drive Wiring'),(240,'FES'),(241,'Rexroth'),(242,'Toyo'),(243,'Cherokee International'),(244,'Sherman Treaters ltd'),(245,'Bowe'),(246,'Kruger Hirschmann pat group'),(247,'CCT'),(248,'CS CABUR'),(249,'Quad'),(250,'Ravas'),(251,'Machine Center Europe'),(252,'HDR'),(253,'Sarg'),(254,'Sussex Gauging Ltd'),(255,'Refu Elektronik'),(256,'SP'),(257,'Enerpro'),(258,'W+D'),(259,'Hydrovane'),(260,'Hitron'),(261,'Locamation'),(330,'Modicon Gould'),(263,'Watlow'),(264,'Deca'),(265,'Snijder Micro Systems'),(266,'FRER'),(267,'Auslosegerat'),(268,'Kriwan Industrie-Elektronik'),(269,'LASAG'),(270,'Markem'),(271,'COOLING TOWER CONTROLLER'),(272,'Shneider Electric  Modicon Quantum'),(273,'YFE'),(274,'Walchem'),(275,'Proton Product Ltd'),(276,'RNA'),(277,'Elobau'),(278,'COMEXI'),(279,'Keyo'),(280,'Texas instrument'),(281,'Dynisco'),(282,'RaycoWylie'),(283,'HedoN'),(284,'Westinghouse'),(285,'Koyo'),(286,'BM TECNOLOGIE INDUSTRIALI'),(287,'Kollmorgen'),(288,'Sew Eurodrive'),(289,'NAMFG'),(290,'Lp Loamation'),(291,'Binks Sames'),(292,'Kilian'),(293,'Control Techniques'),(294,'Otis Elevator'),(295,'IFM Electronic'),(296,'BEGsnc'),(297,'Pilz'),(298,'Tccco'),(299,'CCK'),(300,'Pulsar'),(301,'Telesis'),(302,'Eriez Metalarm'),(303,'Astec'),(304,'Murrelektronik'),(305,'Elan'),(306,'TAE'),(307,'Topax Jesco'),(308,'Bonfiglioli Vectron'),(309,'Saeco'),(310,'Grossen Bacher'),(311,'Hanwool System'),(312,'Baldor'),(313,'Gerald L. Greer Company'),(314,'Megatron'),(315,'Mannesmann Rexroth'),(316,'I.P.S. Sistemi programmabili'),(317,'Spektrum Controls'),(318,'JDM'),(319,'Berthold'),(320,'Keynce'),(321,'Woche'),(322,'M.R.C.'),(323,'Tadbik'),(324,'West Instruments'),(325,'Arburg'),(326,'Lambda'),(327,'Nipson'),(328,'Motronica'),(329,'Solcon'),(331,'R.I.M.'),(332,'Ritsch Rollfix Easy'),(333,'Cosel'),(334,'Leuze'),(335,'PS CAL'),(336,'Carl Zeiss'),(337,'JUMO GmbH'),(338,'Multicom'),(339,'Deutschmann Automation'),(340,'Weintek Labs'),(341,'Spsrtanics'),(342,'Yokogawa'),(343,'Leroy Somer'),(344,'Sharp'),(345,'COMSYS AG'),(346,'Diamond Systems Corp.'),(347,'Mirror'),(348,'Jean Lutz'),(349,'ppe'),(350,'JK'),(351,'Sames'),(352,'M.C.A.'),(353,'Epson'),(354,'Key tronic corporation'),(355,'Enilam'),(356,'UniOP'),(357,'GEFRAN'),(358,'Miniranger Pus'),(359,'Milltronics'),(360,'Vector Drive'),(361,'Enercon Industries Corporation'),(362,'Gossen-Metrawatt GMBH'),(363,'KMG Systems Limited'),(364,'Ahlbrandt System'),(365,'Pronomic'),(366,'Telemecanique'),(367,'Benning'),(368,'Perma-Drive'),(369,'Nada Electronics'),(370,'Ishida'),(371,'Kinco'),(372,'Lock Inspection Systems'),(373,'UPS-MAN'),(374,'KB Electronics, Inc.'),(375,'Open Date Ltd'),(376,'MAG'),(377,'Hioki'),(378,'TNA Packaging Systems'),(379,'SR Drives Ltd.'),(380,'Garvens'),(381,'NEC'),(382,'Tecnodue S.r.l.'),(383,'Philips'),(384,'Wolke'),(385,'Danieli Automation'),(386,'Festo'),(387,'Flash'),(388,'Eurotek'),(389,'CEIA'),(390,'Micro Motion Inc'),(391,'Skynet Electronic'),(392,'Elletre'),(393,'MR'),(394,'Robox'),(395,'REGLOPLAS'),(396,'NBB'),(397,'Apelco-Carel'),(398,'Mocon'),(399,'Schuntermann & Benninghoven'),(400,'Star Micronics Co. Ltd'),(401,'Power One'),(402,'Condair'),(403,'Sprint Electric'),(404,'Overback'),(405,'Karl Schermer'),(406,'Rigol'),(407,'Madah Com'),(408,'Drive Engineering Ltd.'),(409,'TDE'),(410,'Gould'),(411,'SCHOTT North America, Inc'),(412,'S.G.E.'),(413,'iCEM'),(414,'Artecnica'),(415,'Analog Devices'),(416,'Fatek'),(417,'Hakko'),(418,'Nordson'),(419,'YAM-Automation'),(420,'ElectroCraft'),(421,'Camille Bauer'),(422,'PolyScience'),(423,'Phoenix Contact'),(424,'Master System Control Inc.'),(425,'NUK'),(426,'Thies'),(427,'SSB'),(428,'Elettromeccanica Valotti'),(429,'SME elettronica'),(430,'Durma'),(431,'Antari'),(432,'Hottinger Baldwin Messtechnik'),(433,'Honda'),(434,'Nemic-Lambda'),(435,'Keba'),(436,'S.P.E. elettronica'),(437,'Haefely Trench'),(438,'Thermatool'),(439,'MAL'),(440,'Stromag'),(441,'Atlet'),(442,'Rectifier'),(443,'Schneider Automation S.A.'),(444,'Seewer Rondo'),(445,'Klockner Moeller'),(446,'HWH'),(447,'PDS Design Solutions Ltd'),(448,'SCM Group'),(449,'Prominent'),(450,'SGE'),(451,'Salmco Technik GMBH'),(452,'GE Sensing'),(453,'Elettronica Verona'),(454,'Magnetek'),(455,'A.M. LOCK Ltd Oldham England'),(456,'Matrix'),(457,'Bender'),(458,'Moti Kiduhim'),(459,'Ronan'),(460,'Unomat'),(461,'Woywod'),(462,'IMA'),(463,'Erni'),(464,'Teco Electric & Machinery Co., Ltd.'),(465,'Brankamp'),(466,'Lauer'),(467,'PSG Mannheim'),(468,'EME'),(469,'Red Lion Control'),(470,'Thermotron'),(471,'EL'),(472,'Selema'),(473,'Graphics West Micro Systems'),(474,'Kramer'),(475,'Nino\'s Enterprises, Inc.'),(476,'Powec'),(477,'Schindler'),(478,'Reliance Electric'),(479,'Denison Hydraulics'),(480,'Still'),(481,'Posidrive'),(482,'Socapel'),(483,'Calemard'),(484,'Phihong'),(485,'Electronic Systems'),(486,'Shimaden Co Ltd'),(487,'Contron Embedded Copmuters'),(488,'LS Industrial Systems'),(489,'Silma'),(490,'Clinton Instrument Company'),(491,'Bircher'),(492,'KliKlok-Woodman'),(493,'Martindale Electric Co. Ltd'),(494,'York'),(495,'Stentofon'),(496,'Positest'),(497,'DynAmp'),(498,'ConLAB'),(499,'Metrohm'),(500,'Gavish'),(501,'Masa Measurments Ltd'),(502,'Power Measurement Ltd'),(503,'Quarts'),(504,'Weidmuller'),(505,'Magnesium Laboratory'),(506,'TankRadar'),(507,'A&D CO., LTD'),(508,'Dimex'),(509,'Michshur ltd.'),(510,'Eliwell'),(511,'IME'),(512,'Pixsys'),(513,'Isotec'),(514,'Mescon'),(515,'TRENDnet'),(516,'Bartec'),(517,'Fltechnology Co., Ltd'),(518,'Nokeval'),(519,'Rosemount'),(520,'MRC'),(521,'ACD'),(522,'Minicom'),(523,'Veiwsonic'),(524,'Drager'),(525,'Schlenker elektronik'),(526,'Carlo Gavazzi'),(527,'Vulcan'),(528,'Streamlight'),(529,'Bentron Power'),(530,'Schneider Automation Inc'),(531,'Rosemount Analytical Inc'),(532,'Cameron'),(533,'Schneider Electric'),(534,'Megger'),(535,'Foxboro'),(536,'Rochester'),(537,'Iwaki'),(538,'Meitav-tec'),(539,'Fluke'),(540,'Trafo Guard'),(541,'Precisa Instruments Ltd.'),(542,'Asus'),(543,'Dini Argeo'),(544,'MSA'),(545,'Eilon Engineering'),(546,'Y.S.T'),(547,'Beamex'),(548,'Fisher-Rosemount Systems'),(549,'Sacal'),(550,'SEC'),(551,'Aten International Co Ltd'),(552,'Merlin Gerin'),(553,'lae ELECTRONIC'),(554,'Analogic'),(555,'Befeld'),(556,'Carrier'),(557,'Kemp'),(558,'Amiad Filtration Systems'),(559,'Testo Ltd'),(560,'Emerson'),(561,'Solartron Mobrey Limited'),(562,'MEILI'),(563,'Gilson'),(564,'Mobrey Ltd.'),(565,'AKO electronica'),(566,'Paul Brun AB'),(567,'Ledsgo'),(568,'C&TECH CO., LTD'),(569,'Rohrlux'),(570,'Giant Safety'),(571,'BIFA'),(572,'Schenck'),(573,'VEGAVIB'),(574,'FineTek'),(575,'PELCO'),(576,'PENTA Gruppo Kkt.'),(577,'BROOKS'),(578,'Dayna Vox'),(579,'ABB Limited'),(580,'Roal Electronics'),(581,'Belline'),(582,'Dialight'),(583,'Gamatronic'),(584,'Lumel'),(585,'Texas Industrial Peripherals'),(586,'Kensington'),(587,'Blasing Elektronik GmbH Essen'),(588,'EXEECO LTD. ENGLAND'),(589,'Lenovo'),(590,'Tecnologic'),(591,'Weigel Messgerate GmbH'),(592,'Legrand'),(593,'Cemarline'),(594,'FLT'),(595,'LMR'),(596,'ECA'),(597,'Zhongshan LED'),(598,'Gigasense AB'),(599,'Wiebrock'),(600,'KUK DONG'),(601,'TORNA TECH'),(602,'MTC'),(603,'Elmad'),(604,'Glacial Light'),(605,'Magnetrol'),(606,'Avron'),(607,'Endress & Hauser'),(608,'R. STAHL'),(609,'Riello'),(610,'Cutler-Hammer'),(611,'Trane'),(612,'cmc Instruments GmbH'),(613,'UNI-T'),(614,'MEG Mechanik GmbH'),(615,'GS'),(616,'I.T.E. Power Supply'),(617,'VITLAB'),(618,'SKF'),(619,'Nisco'),(620,'JODEWAY'),(621,'Baccara'),(622,'Psk Controllers Ltd'),(623,'PRIMAX'),(624,'BEKA associates'),(625,'Julabo'),(626,'Kamstruo'),(627,'Shimrad'),(628,'Fujicom'),(629,'Qualitrol Corp. Fairport N.Y'),(630,'Winsun'),(631,'STAHL Crane System'),(632,'LUMIN'),(633,'Pharmaceutical'),(634,'Jasco'),(635,'Sun microsystems'),(636,'Shenzhen Fujia Appliance Co.Ltd'),(637,'Tami 4'),(638,'Getriebebau NORD'),(639,'Unknown'),(640,'Poosled'),(641,'Scanvac'),(642,'EATON'),(643,'Tzubavision'),(644,'Sony'),(645,'BERSING TECHNOLOGY CO., LTD'),(646,'Sunkyo'),(647,'Conveyor Components'),(648,'Alre'),(649,'Jeil Media Co'),(650,'BALLUF'),(651,'IONPURE'),(652,'Powertron Elektronics Corp.'),(653,'Krohne'),(654,'Motorola'),(655,'Remote Control'),(656,'IFE'),(657,'Konzept Enerietechnik GmbH'),(658,'Solytech Enterprise Corporation'),(659,'Planet'),(660,'Anatel'),(661,'Enerdis'),(662,'VICONICS'),(663,'Citizen Scale'),(664,'ISCAN'),(665,'ELBA-electric GmbH'),(666,'Zika'),(667,'Cohen'),(668,'Biffi'),(669,'Kistler-Morse Corp.'),(670,'Converter Concepts inc.'),(671,'Weintek Labs, Inc'),(672,'ELWA'),(673,'Durag'),(674,'MACS'),(675,'UDS'),(676,'Eaton Hydraulics'),(677,'Static Control Systems STS'),(678,'3-K Elektrik'),(679,'Norsk Electro Optikk AS NEO'),(680,'CISCO'),(681,'Dizitech'),(682,'Power Electronics'),(683,'LaserGas-II'),(684,'INSTRON'),(685,'PCME Ltd'),(686,'Auma'),(687,'FROMM'),(688,'Raytek'),(689,'AMDAR'),(690,'HMS Industrial Networks AB'),(691,'Sectoriel'),(692,'Sun Yeh'),(693,'Vega'),(694,'Invertron'),(695,'TSI'),(696,'DOLD'),(697,'Mean Well'),(698,'Kav Electric Ltd.'),(699,'Molram'),(700,'Danbridge'),(701,'RGI industries'),(702,'IAI'),(703,'Dell'),(708,'BLE'),(709,'Syrelec'),(707,'IKA'),(710,'Sartorius'),(711,'Fizepr'),(712,'Programma'),(713,'KLEINTGES'),(714,'MASDAF'),(715,'Makita'),(716,'Sprecher+schun'),(717,'Electronic Design'),(718,'Dixell'),(719,'Matam'),(720,'Inter Anlork'),(721,'Protec'),(722,'Inter Alronk'),(723,'XP Power'),(724,'Weed Fiber-Optics'),(725,'GlobTek'),(726,'Biddle Instruments'),(727,'ICP DAS'),(728,'Billion'),(729,'Ariye ben ari'),(730,'Densei-Lambda'),(731,'OZ-ON'),(732,'DAKOTA ULTRACONICS'),(733,'RCS'),(734,'Allanson'),(735,'AeroAntenna Technology'),(736,'SALICRU'),(737,'Fizelektropribor'),(738,'CINCON ELECTRONICS'),(739,'Eljintek'),(740,'ZIVAN'),(741,'NOVA'),(742,'AFCON USA'),(743,'stpi'),(744,'movi THERM'),(745,'No brand'),(746,'ETA ELECTRIC IND CO');
/*!40000 ALTER TABLE `manufacturer` ENABLE KEYS */;
UNLOCK TABLES;

--
-- Table structure for table `priority`
--

DROP TABLE IF EXISTS `priority`;
/*!40101 SET @saved_cs_client     = @@character_set_client */;
/*!50503 SET character_set_client = utf8mb4 */;
CREATE TABLE `priority` (
  `id` int NOT NULL AUTO_INCREMENT,
  `name` varchar(16) DEFAULT NULL,
  PRIMARY KEY (`id`)
) ENGINE=MyISAM AUTO_INCREMENT=15 DEFAULT CHARSET=utf8mb3;
/*!40101 SET character_set_client = @saved_cs_client */;

--
-- Dumping data for table `priority`
--

LOCK TABLES `priority` WRITE;
/*!40000 ALTER TABLE `priority` DISABLE KEYS */;
INSERT INTO `priority` VALUES (1,'Highest'),(2,'High'),(5,'Medium'),(9,'Low');
/*!40000 ALTER TABLE `priority` ENABLE KEYS */;
UNLOCK TABLES;

--
-- Table structure for table `status`
--

DROP TABLE IF EXISTS `status`;
/*!40101 SET @saved_cs_client     = @@character_set_client */;
/*!50503 SET character_set_client = utf8mb4 */;
CREATE TABLE `status` (
  `id` int NOT NULL AUTO_INCREMENT,
  `name` varchar(16) DEFAULT NULL,
  PRIMARY KEY (`id`)
) ENGINE=MyISAM AUTO_INCREMENT=13 DEFAULT CHARSET=utf8mb3 ROW_FORMAT=DYNAMIC;
/*!40101 SET character_set_client = @saved_cs_client */;

--
-- Dumping data for table `status`
--

LOCK TABLES `status` WRITE;
/*!40000 ALTER TABLE `status` DISABLE KEYS */;
INSERT INTO `status` VALUES (2,'Not ready $'),(5,'Ready'),(8,'Warehouse'),(11,'Archive'),(1,'Not ready');
/*!40000 ALTER TABLE `status` ENABLE KEYS */;
UNLOCK TABLES;

--
-- Temporary view structure for view `view_issue`
--

DROP TABLE IF EXISTS `view_issue`;
