Total corrozion in all blocks.
Internal PS faulted.
100uF 35V  -4.
Repair and cleaning of LCD cable.
Read information about service mode.
Indication must be AUTO
Calibrator - resistor 320R / 500ppm
Alarm_1>100ppm
Alarm_2>130ppm